Advances in Terahertz Frequency Combs by Robert A. Stead MSci, Imperial College, London, 2004 A THESIS SUBMITTED IN PARTIAL FULFILLMENT OF THE REQUIREMENTS FOR THE DEGREE OF DOCTOR OF PHILOSOPHY in The Faculty of Graduate Studies (Physics) THE UNIVERSITY OF BRITISH COLUMBIA (Vancouver) October 2012 c Robert A. Stead 2012Abstract Following a review of the theory of terahertz generation via optical recti- cation in nonlinear crystals, a method for enhancing the nonlinear con- version e ciency of this process is proposed. A nonlinear crystal is placed at the intracavity focus of a passive optical resonator, which is seeded by an ytterbium-doped bre laser. Models of this arrangement indicate that an enhancement of the optical eld of several orders of magnitude is pos- sible. As the di erence frequency radiation produced through the optical recti cation process results from a mixing of the spectral components of the optical eld, one expects a corresponding increase in the terahertz eld. We present a design of optical resonator that compensates for the large group ve- locity dispersion of the nonlinear crystal. Our experimental results indicate that below bandgap absorption in the crystal severely limits the resulting enhancement of the optical eld, and hence the terahertz eld one would ex- pect from this nonlinear process. A scanning-delay terahertz time-domain spectrometer has been constructed, using a gallium phosphide guiding struc- ture to increase the interaction length of the optical and terahertz elds, thereby increasing the terahertz power produced. Our experiment demon- strates 20 dB signal to noise ratio over the spectral range of 0.5-1 THz. We propose a method for increasing the spectral resolution, whilst simul- taneously reducing the required data acquisition time of such a terahertz spectrometer, through the use of two femtosecond optical frequency combs. One of these elds drives the nonlinear optical recti cation process, whilst the second serves as a sampling local oscillator eld to probe the terahertz eld via electro-optic sampling in a second nonlinear crystal. By precisely controlling the relative pulse repetition rates of the two oscillators, we show that the full spectral content of the terahertz eld can be acquired at rf iiAbstract frequencies, and without the slow mechanical delay lines associated with conventional terahertz time-domain spectroscopy. Finally, we present ex- perimental e orts towards the demonstration of this technique, and show that, to be e ective, steps must be taken to increase the strength of the expected rf signal over that of the measurement noise oor. iiiPreface The initial proposal for the cavity-enhanced terahertz generation was de- vised by David J. Jones, with initial modelling and calculations performed by Jesse C. Petersen. I developed the modelling work to produce the results of section 3.3.3, and this led to the design of the optical cavity presented in section 3.3. The method for coupling the terahertz radiation out of the non- linear crystal was developed by Jesse C. Petersen. I was responsible for the experimental e ort and results presented in sections 3.4 through 3.9. The single-comb terahertz spectroscopy work presented in Chapter 4 was entirely my own e ort, and this was used as a precursor to the dual-comb work pre- sented later in Chapter 6. The initial concept for the heterodyne detection scheme presented in Chapter 5 was my own work, but bene ted greatly from discussions with David J. Jones and Arthur K. Mills. This formed the basis of work presented by David J. Jones at an invited talk of the 2010 March Meeting of the American Physical Society, entitled \Multi-frequency THz Heterodyne Spectroscopy Using Electro-optic Sampling." This work was then subsequently developed and forms the basis of discussions included in Chapter 5. I performed all modelling and calculations presented. This more detailed analysis has been accepted for publication as Robert A. Stead, Arthur K. Mills, and David J. Jones. \Method for High Resolution and Wide Band Spectroscopy in the Terahertz and Far-Infrared Region", Journal of the Optical Society of America B. The manuscript and discussions contained within were prepared in conjunction with the co-authors. Finally, the exper- imental e orts of Chapter 6 were performed by myself, save for the design and fabrication of the bandwidth-limited balanced photodetector presented in Appendix A, which was designed and built by Pavel Trochtchanovitch, to speci cations provided by David J. Jones. ivTable of Contents Abstract . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . ii Preface . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . iv Table of Contents . . . . . . . . . . . . . . . . . . . . . . . . . . . . v List of Tables . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . ix List of Figures . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . x Acknowledgements . . . . . . . . . . . . . . . . . . . . . . . . . . . xvi 1 Introduction . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 1 1.1 Motivation . . . . . . . . . . . . . . . . . . . . . . . . . . . . 1 1.1.1 Terahertz Spectroscopy . . . . . . . . . . . . . . . . . 1 1.1.2 The Terahertz Gap . . . . . . . . . . . . . . . . . . . 3 1.1.3 Frequency Comb Spectroscopy . . . . . . . . . . . . . 5 1.2 Outline of Thesis . . . . . . . . . . . . . . . . . . . . . . . . 5 2 Background . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8 2.1 Optical Recti cation . . . . . . . . . . . . . . . . . . . . . . . 8 2.1.1 Calculation of the Radiated Field . . . . . . . . . . . 13 2.2 Beam Polarisation and Crystal Orientations . . . . . . . . . 17 2.3 Phase-matching . . . . . . . . . . . . . . . . . . . . . . . . . 21 2.4 Lasers . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 25 2.4.1 Oscillator Design . . . . . . . . . . . . . . . . . . . . 25 vTable of Contents 2.4.2 Compensation of Pulse Chirp due to Dispersion of Fi- bre Components . . . . . . . . . . . . . . . . . . . . . 28 2.4.3 Pump Diode Line Noise . . . . . . . . . . . . . . . . . 33 3 Use of a Passive Enhancement Cavity for High E ciency Terahertz Generation . . . . . . . . . . . . . . . . . . . . . . . 37 3.1 Cavity Basics . . . . . . . . . . . . . . . . . . . . . . . . . . . 37 3.2 Ampli cation of a Femtosecond Optical Comb Using a Pas- sive Enhancement Cavity . . . . . . . . . . . . . . . . . . . . 42 3.3 Cavity Design . . . . . . . . . . . . . . . . . . . . . . . . . . 48 3.3.1 Mirror Choice and Dispersion Compensation . . . . . 48 3.3.2 Terahertz Output Coupling . . . . . . . . . . . . . . . 53 3.3.3 Cavity Performance Calculations . . . . . . . . . . . . 55 3.4 Cavity Locking . . . . . . . . . . . . . . . . . . . . . . . . . . 61 3.5 Spatial Mode-matching into the Optical Resonator . . . . . . 68 3.5.1 Gaussian Mode of the Optical Resonator . . . . . . . 68 3.5.2 Beam Pro ling . . . . . . . . . . . . . . . . . . . . . . 70 3.5.3 Mode-matching Solution . . . . . . . . . . . . . . . . 71 3.6 Characterisation of the Passive Enhancement Cavity - Results and Analysis . . . . . . . . . . . . . . . . . . . . . . . . . . . 72 3.7 Measurement of the Cavity Finesse . . . . . . . . . . . . . . 73 3.7.1 Results of Finesse Measurements . . . . . . . . . . . . 76 3.8 Enhancement Bandwidth of the Cavity . . . . . . . . . . . . 77 3.9 Linear Optical Loss of Gallium Phosphide . . . . . . . . . . 84 3.9.1 Loss Mechanisms in Gallium Phosphide . . . . . . . . 87 3.9.2 Characterisation of the Spectral Absorption of Gal- lium Phosphide . . . . . . . . . . . . . . . . . . . . . 90 3.10 Conclusions on the use of a Passive Enhancement Cavity for High E ciency Generation of Terahertz Radiation . . . . . . 92 4 A Single Comb Terahertz Time-domain Spectrometer with Detection Using Conventional Electro-optic Sampling . . 96 4.1 Introduction . . . . . . . . . . . . . . . . . . . . . . . . . . . 96 viTable of Contents 4.2 Experimental Arrangement . . . . . . . . . . . . . . . . . . . 97 4.3 Detection of the Terahertz Frequency Comb via Electro-optic Sampling . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 101 4.3.1 The Linear Electro-optic E ect . . . . . . . . . . . . 101 4.3.2 Comparison of Detection Schemes . . . . . . . . . . . 109 4.4 Results of Single Comb Detection Scheme . . . . . . . . . . . 112 4.4.1 Signal Scaling with Optical Probe Power . . . . . . . 115 4.5 Conclusions . . . . . . . . . . . . . . . . . . . . . . . . . . . . 115 5 Dual-comb Spectroscopy in the Terahertz and Far-IR Re- gion . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 119 5.1 Introduction . . . . . . . . . . . . . . . . . . . . . . . . . . . 119 5.2 Mapping of a Terahertz Frequency Comb to the Radio Fre- quency Spectral Region via Asynchronous Electro-optic Sam- pling . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123 5.3 E ect of Spectral Phase on Heterodyne Beat Signal . . . . . 135 5.4 E ect of Phase-matching on the Heterodyne Beat Signal . . 139 5.5 Bandwidth and Choice of Detuning . . . . . . . . . . . . . . 144 5.6 Signal to Noise Scaling . . . . . . . . . . . . . . . . . . . . . 146 6 Experimental Demonstration of Dual-comb Terahertz Spec- troscopy . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 148 6.1 Coupling of Detuned Local Oscillator Pulse . . . . . . . . . . 148 6.2 Controlling The Laser Repetition Rates . . . . . . . . . . . . 149 6.2.1 RF Locking Scheme . . . . . . . . . . . . . . . . . . . 149 6.2.2 Component Testing and Characterisation . . . . . . . 152 6.3 Cross Correlation of Laser Pulses to Verify Timing Jitter . . 156 6.4 Schemes for Detection of the Heterodyne Beat . . . . . . . . 161 6.4.1 High Gain Balanced Detector With High Extinction of Frequencies Greater than 80 MHz . . . . . . . . . . 161 6.4.2 RF Heterodyne Detection Using VCO Reference . . . 164 6.5 Results and Conclusions from Dual-comb Spectroscopy Mea- surement . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 169 viiTable of Contents 7 Final Conclusions . . . . . . . . . . . . . . . . . . . . . . . . . . 175 Bibliography . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 177 Appendices A Technical Information Regarding High Gain, Low-pass Fil- tered, Balanced Detector for Detection of the Heterodyne Beat in the Dual-comb Spectrometer Setup . . . . . . . . . 188 viiiList of Tables 3.1 Part numbers and speci cations for Layertec mirrors used in the design of the passive enhancement cavity. . . . . . . . . . 53 3.2 Properties of a Gaussian beam . . . . . . . . . . . . . . . . . 68 3.3 Optical enhancement, resonant bandwidths, and net cavity dispersions for various cavity con gurations used to verify the e ect of dispersion on the available enhancement. . . . . . 81 3.4 Thermal properties of gallium phosphide. . . . . . . . . . . . 87 ixList of Figures 1.1 Terahertz transitions of some simple molecules. . . . . . . . . 2 1.2 Major transitions of the water vapour molecule in the spectral region 0.5 to 1.5 THz. . . . . . . . . . . . . . . . . . . . . . . 3 2.1 Time and domain descriptions of a mode-locked laser oscillator. 11 2.2 Decomposition of material response to pulsed radiation into sum and di erence frequency terms. . . . . . . . . . . . . . . 12 2.3 DFG for the case of a frequency comb input. . . . . . . . . . 13 2.4 Growth of the di erence frequency eld, E2(z) with propaga- tion distance for the case of perfect phase-matching, k = 0. 17 2.5 Coordinate system used for deriving the optimum optical beam orientation for terahertz generation. . . . . . . . . . . . 19 2.6 Relative nonlinear polarisation as a function of incidence an- gle, , and azimuthal angle, . . . . . . . . . . . . . . . . . . 20 2.7 Dimensions, cuts, and orientation of gallium phosphide crys- tal for enhancement cavity work. . . . . . . . . . . . . . . . . 21 2.8 Variation of refractive index throughout the optical and ter- ahertz spectral ranges, along with an experimental t of the e ective refractive index for optical wavelengths, as per equa- tion 2.33 . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 23 2.9 Coherence length for terahertz frequency generation via op- tical recti cation in gallium phosphide, as predicted by equa- tion 2.34. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 24 2.10 Design of the two mode-locked ytterbium bre lasers. . . . . 27 xList of Figures (a) Laser oscillator design used to provide the high-power (following two subsequent ampli cation stages) pulses for both the cavity work and the single-comb, terahertz time-domain spectroscopy work described later. . . . . . 27 (b) Laser oscillator design used to provide the local oscilla- tor pulses for the work described in Chapter 6. . . . . . 27 2.11 Con guration of the di raction grating pulse compressor used for dispersion compensation. . . . . . . . . . . . . . . . . . . . 31 2.12 Typical autocorrelation trace of the laser output pulse of the pump laser following compression. . . . . . . . . . . . . . . . 32 2.13 Comparison of modulation sidebands on the optical pulse train repetition rate of 104.3 MHz following the high-power ampli er stage with the controller in its original con gura- tion, and following modi cations to allow the regulation of the transistor control voltage with an external linear dc supply. 34 2.14 Comparison of noise on the electrical current used to drive the Apollo Instruments pump diode, as measured across a shunt resistor. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 35 2.15 Schematic of the Apollo Instruments laser diode driver, model no. D-560. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 36 3.1 A simple two-mirror optical resonator. . . . . . . . . . . . . . 39 3.2 Resonant modes of an optical resonator. . . . . . . . . . . . . 41 3.3 Varying overlap of the resonant modes of the passive enhance- ment cavity with the longitudinal modes of the mode-locked laser used as the seed. . . . . . . . . . . . . . . . . . . . . . . 45 3.4 Resonance map of a dispersion-free enhancement cavity. . . . 47 3.5 Resonance map for a passive enhancement cavity exhibiting a large net group velocity dispersion. . . . . . . . . . . . . . . 49 3.6 Experimental arrangement for the passive enhancement cavity. 50 3.7 Refractive Index and dispersion of gallium phosphide as pre- dicted by the Sellmeier equation. . . . . . . . . . . . . . . . . 52 3.8 Crystal geometry for output coupling of the terahertz radiation. 54 xiList of Figures 3.9 Optical design for output coupling of terahertz radiation for a gallium phosphide crystal at Brewster incidence. . . . . . . 56 3.10 E ective enhancement of the range of optical frequencies ef- fectively coupled into the enhancement cavity for various in- put couplers. . . . . . . . . . . . . . . . . . . . . . . . . . . . 59 3.11 Calculated residual GDD of the cavity components for the cavity con guration shown in gure 3.6. . . . . . . . . . . . . 60 3.12 Estimated broadening of a 200 fs input optical input pulse within a passive enhancement cavity of the same design of that used in gure 3.10. . . . . . . . . . . . . . . . . . . . . . 60 3.13 Circulating pulse energy enhancement and duration as a func- tion of input coupler re ectivity for the cavity con guration of 3.12. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 61 3.14 The re ection signal measured from the enhancement cavity near a position of resonance. . . . . . . . . . . . . . . . . . . . 63 3.15 Calculated error signal for a passive enhancement cavity hav- ing a nesse of F = 2 105 . . . . . . . . . . . . . . . . . . . 66 3.16 Beam waist (radius) pro le following a 250 mm focal length converging lens. . . . . . . . . . . . . . . . . . . . . . . . . . . 71 3.17 Schematic of optics used for mode-matching into the passive enhancement cavity. . . . . . . . . . . . . . . . . . . . . . . . 73 3.18 Ring down measurement for the passive enhancement cavity without the gallium phosphide crystal, the cavity having a predicted nesse of â€™960. . . . . . . . . . . . . . . . . . . . . 77 3.19 Ring down measurement for the passive enhancement cavity without the gallium phosphide crystal, the cavity having a predicted nesse of â€™1800. . . . . . . . . . . . . . . . . . . . . 78 3.20 Intracavity enhancement for varying amounts of residual GDD. 83 3.21 Optical spectra supported by the passive enhancement cavity with the gallium phosphide crystal placed at the intracavity focus. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 84 xiiList of Figures 3.22 Variation of the voltage bias applied to the piezoelectric ac- tuator (PZT) as a function of time following locking of the enhancement cavity. . . . . . . . . . . . . . . . . . . . . . . . 86 3.23 E ect of parasitic loss on the optimal enhancement factor that can be obtained from a passive enhancement cavity. . . . 88 3.24 Average power transmitted through the gallium phosphide crystal as a function of incident average power. . . . . . . . . 89 3.25 Spectral absorption of gallium phosphide. . . . . . . . . . . . 91 3.26 Maximum possible enhancement factor provided by an impedance- matched enhancement cavity containing a nonlinear crystal having absorption coe cient, = 0:5 cm 1 and varying length. 94 4.1 Experimental arrangement of single-comb terahertz time-domain spectrometer . . . . . . . . . . . . . . . . . . . . . . . . . . . 100 4.2 Beam propagation in the gallium phosphide waveguide struc- ture. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 101 4.3 Crystal geometry for the electro-optic detection scheme. . . . 105 4.4 Deformation of the two-dimensional refractive index ellipsoid in the (110) crystal plane. . . . . . . . . . . . . . . . . . . . . 107 4.5 Time-domain electro-optic sampling (EOS) traces of the gen- erated terahertz eld . . . . . . . . . . . . . . . . . . . . . . 113 (a) Electro-optic sampling trace with single channel detector113 (b) Electro-optic sampling trace acquired using balanced detection scheme. . . . . . . . . . . . . . . . . . . . . . 113 4.6 Resulting time-domain EOS traces for varying average powers of the optical probe pulse. . . . . . . . . . . . . . . . . . . . . 116 4.7 Scaling of the peak photocurrent observed from the single channel electro-optic sampling apparatus as a function of av- erage power of the optical pulse. . . . . . . . . . . . . . . . . 117 5.1 Overview of the multi-frequency heterodyne detection scheme for the dual-comb terahertz spectrometer. . . . . . . . . . . . 122 xiiiList of Figures 5.2 Schematic of a terahertz spectrometer where the pump fem- tosecond frequency comb (FFC) and local oscillator (LO FFC) frequency combs di er in pulse repetition rate by an amount frep. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 124 5.3 Frequency components [produced through mixing of the ter- ahertz frequency comb with the detuned LO comb. . . . . . . 132 5.4 E ect of various spectral phase contributions on the expected signal magnitude predicted by equation 5.23. . . . . . . . . . 140 5.5 Variation of phase and amplitude of A-EOS sidebands as a function of optical wavelength for a ZnTe detection crystal of 0.4 mm length. . . . . . . . . . . . . . . . . . . . . . . . . . . 142 5.6 Variation of phase and amplitude of A-EOS sidebands as a function of optical wavelength for a ZnTe detection crystal of 0.8 mm length. . . . . . . . . . . . . . . . . . . . . . . . . . . 142 5.7 Variation of phase and amplitude of A-EOS sidebands as a function of optical wavelength for a ZnTe detection crystal of 1.6 mm length. . . . . . . . . . . . . . . . . . . . . . . . . . . 143 5.8 Calculated heterodyne beat signal for a THz frequency of 0.85 THz, as a function of nonlinear crystal length for a ZnTe detection crystal, and an LO FFC of 7 nm bandwidth, centred at 1030 nm. . . . . . . . . . . . . . . . . . . . . . . . . . . . . 144 5.9 Map of the expected bandwidth performance one can reason- ably expect from the dual-comb terahertz spectrometer sys- tem for given constraints in oscillator design and acquisition methods. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 145 6.1 Experimental arrangement of the dual-comb terahertz spec- trometer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 150 6.2 Schematic of the rf locking scheme used to control the pulse repetition rates of the two lasers. . . . . . . . . . . . . . . . . 151 6.3 Arrangement for measuring of the noise oor of the available low frequency mixers. . . . . . . . . . . . . . . . . . . . . . . 153 xivList of Figures 6.4 Measured noise oors for the three available low frequency mixers. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 154 6.5 Spectrum of the residual noise on the error signal for the range of high frequency mixers available. . . . . . . . . . . . . . . . 155 6.6 Pulse trains from two lasers, with pulse repetition rates frep;1 and frep;2 = frep;1 + frep. . . . . . . . . . . . . . . . . . . . 157 6.7 Experimental arrangement for the cross-correlation measure- ment. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 158 6.8 Cross-correlation trace of the femtosecond optical combs in -BBO with the repetition rate detuning set to 400 Hz. . . . 159 6.9 RF spectral output for the cross-correlation measurement de- picted in gure 6.7. . . . . . . . . . . . . . . . . . . . . . . . . 160 6.10 Output of the custom-designed balanced detector for a typical balanced input with additional external ampli cation. . . . . 162 6.11 Noise spectrum for the high-speed balanced detector used for detection of the heterodyne beat signal. . . . . . . . . . . . . 163 6.12 Schematic of the rf circuitry used to provide a reference signal at the expected rf heterodyne beat frequency. . . . . . . . . . 166 6.13 Circuit schematic for the active lter used to lock the volt- age control oscillator (VCO) to the expected frequency of the heterodyne beat signal. . . . . . . . . . . . . . . . . . . . . . 167 6.14 Measured residual noise level on the VCO control signal for locking of the VCO to the detuning and a reference oscillator. 168 A.1 Circuit schematic of the custom-designed, bandwidth-limited balanced photodetector. . . . . . . . . . . . . . . . . . . . . . 189 xvAcknowledgements I would like to rst thank Professor David Jones for taking me into his research group and providing me with the opportunity to learn from him over the course of my studies. I am sure many other graduate students would be envious of having a supervisor who spends so much time in the lab, and who makes himself so accessible for discussions regarding ideas and problems. In addition, I owe a great deal of thanks to Arthur Mills, whose seem- ingly endless time, patience and enthusiasm were invaluable throughout my time at UBC. Art never failed to o er insightful input, regardless of the sub- ject material! My gratitude also goes to Jie Jang, another Post-doc whom I had the pleasure of working with in the initial part of my research at UBC. Jieâ€™s magic touch when it came to lasers never ceased to amaze me. Thank you also to Jesse Petersen, who completed his Ph.D at Simon Fraser Univer- sity shortly after I began my research, and who provided many interesting technical discussions and ideas regarding the cavity enhanced terahertz gen- eration work. Thanks too to Professor Steve Dodge, who supervised Jesse and invited me to visit their lab to see their terahertz spectrometer system in action. I am also grateful to Mario Beaudoin who kindly o ered the use of his spectrometer setup for measurement of the band-gap edge pro le of gallium phosphide, and who provided useful input on the topic of absorption mechanisms. Of course, my time at UBC was made all the more enjoyable thanks to the company of the other students I had the good fortune to work with. TJ Hammond, whose experiences and frustrations always served as an indicator of what I could look forward to, as well as encouraging me that there is, indeed, light at the end of the tunnel! Thank you also to Matt Lam, with xviAcknowledgements whom I shared many interesting discussions, and had the pleasure of working alongside during many teaching assignments. I would like to thank the many project students and undergraduate students I had the opportunity to engage with during my time at UBC. Your enthusiasm and thirst for knowledge were a constant source of inspiration, and I sincerely hope that I was able to inspire in return, as well as o er counsel on research and life as a postgraduate student. Finally, thank you to my friends and family for your patience, under- standing and encouragement. I consider myself incredibly lucky to have had the support I have enjoyed regardless of the path I choose to pursue. xviiChapter 1 Introduction 1.1 Motivation 1.1.1 Terahertz Spectroscopy Optical spectroscopy is now a well established tool for the characterisation of many chemical substances. Indeed, there exists many spectroscopic tech- niques, spanning the electromagnetic spectrum from microwave frequencies to extreme ultraviolet and x-ray wavelengths. Recently, there has been much interest in developing spectroscopic tools in the far infra-red (IR) range and at longer wavelengths, in what is commonly referred to as the â€˜terahertz gapâ€™, so called due to the in-availability of sources and detectors in this frequency range. Whilst the exact de nition of the terahertz gap is often disputed, it is commonly used to refer to frequencies between 1 and 100 THz, or wave- lengths between 300 and 3 m. It is of considerable interest to develop spectroscopic tools in this range, as many molecules exhibit characteris- tic transitions at terahertz wavelengths. Indeed, lighter molecules and, in particular, those containing carbon-oxygen bonds [1], as well as ammonia and water vapour [2], have been found to have transitions corresponding to rotational and vibrational excited states in the terahertz range. Further, the scaling of the strength of these spectral features as the square or even the cube of the frequency makes the increased sensitivity of terahertz spec- troscopy over more established microwave spectroscopy a draw unto itself. The unique nature of the spectroscopic signatures of many molecules in the terahertz range has led to the moniker of the â€˜ ngerprint rangeâ€™ when refer- ring to these wavelengths. Terahertz spectroscopy has been shown to be an excellent method for distinguishing molecular structures of otherwise similar 11.1. Motivation molecules. See, for example, [3]. 10 20 30 40 50 60 70 80 10âˆ’5 10âˆ’4 10âˆ’3 10âˆ’2 10âˆ’1 100 101 Absorptio nI nte nsit y(c mâˆ’ 1 /m olecul ec mâˆ’ 2 @ 296K ) Frequency (THz) 30002000100030 300020001000 Wavenumber (cm )-1 Wavelength (Î¼m) 30.3 10 5 330 Figure 1.1: Terahertz transitions of some simple molecules [4]. This capability has led to work towards devices capable of remotely identifying and characterising gas leaks and potentially dangerous or noxious gas leaks and plumes, [5], as well as considerable interest in the use of terahertz radiation for detection and characterisation of various drugs and pharmaceutical agents [6], explosives [7],[8] , DNA and other materials [9], [10]. Of particular interest to this project is the spectral range 0.5-1.5 THz. Water vapour has some particularly strong molecular transitions in this range, at 1.16 and 1.11 THz. These correspond to rotational molecular tran- sitions [4]. There is currently great interest in the absorption characteristics of water vapour due to its role in atmospheric radiation budgets. Indeed, some studies have attributed up to 50% of the Earthâ€™s greenhouse e ect to 21.1. Motivation absorption of THz and far IR radiation [11]. Further, there is also great interest in the use of THz sensing methods for terrain mapping, veri cation of atmospheric models, and meteorological predictions [12], [13]. In order to develop these tools, a detailed understanding of terahertz propagation in humid air is necessary. See, for example, recent measurements conducted by Grishkowsky et al. [14] using a scanning Fourier transform terahertz spec- trometer from 0.2 to 2 THz. Whilst these measurements represent state of the art accuracy, we note for later discussions that the authors state limita- tions due to the thermal and mechanical stability of the apparatus, as well as long data acquisition times of around three minutes. 0.5 0.6 0.7 0.8 0.9 1 1.1 1.2 1.3 1.4 10âˆ’3 10âˆ’2 10âˆ’1 100 Frequency/THz Absorptio nI ntensit y(c mâˆ’ 1 /molecul ec mâˆ’ 2 @ 296K ) Figure 1.2: Major transitions of the water vapour molecule in the spectral region 0.5 to 1.5 THz. Fitted from data in [4] 1.1.2 The Terahertz Gap There remains much debate about how best to generate radiation in this wavelength range, at the intersection of spectral regions typically dominated by electronic generation methods, such as Gunn oscillators and Schottky 31.1. Motivation diodes on the low frequency side, and optical methods, such as semicon- ductor diodes on the high frequency side. Whilst e orts have been made to extend the capabilities of electronic, radio frequency (rf) methods into the terahertz range, di culties typically arise due to the limited carrier transit-times, and the simple fact that many of the conventional compo- nents used show resistive losses that give rise to high frequency roll-o at terahertz frequencies [5]. Instead, optical approaches have been developed in order to access this di cult wavelength range. Of these, quantum cas- cade lasers have shown promise in generating high powers. In these devices, photons are emitted by electrons making transitions not across the bandgap, as in conventional semiconductor sources, which are typically limited to fre- quencies above 15 THz due to the in-availability of materials with smaller bandgaps [15], but between conduction subbands of periodic structures, syn- thetically grown to give the desired gain characteristics. Whilst quantum cascade lasers have been demonstrated capable of producing up to 250 mW pulsed power [16], operation in the terahertz range usually requires cryogenic cooling due to the scaling of free carrier absorption with the radiation wave- length, as well as practical di culties in building the quantum well structure [17]. Electro-optic antennas, where a high intensity optical pulse is incident on a semiconductor material, \strained" by a large electrical bias, have also been used, both for the generation and detection of terahertz radiation [18]. Bartels et al. [19] demonstrated a THz time domain spectrometer system using this technique, with a generated THz power of 40 W. However, at higher powers, these techniques su er from saturation e ects of the excited carriers. Likewise, the THz bandwidth is limited by the relaxation lifetime of the carriers to around 5 THz [20]. An alternative scheme is that of optical recti cation, where an ultrashort, and therefore broadband, optical pulse is focused onto a nonlinear crystal, and di erence frequency generation occurs within the bandwidth of the optical pulse. Because this technique is not lim- ited by saturation e ects, the advent of high power, femtosecond pulsed bre lasers has made optical recti cation an exciting method for the generation of terahertz radiation for spectroscopy [21]. 41.2. Outline of Thesis 1.1.3 Frequency Comb Spectroscopy It has been shown (see, for example, [22]) that the output of a mode-locked laser can be described in the frequency domain by a comb of evenly spaced frequency elements, with each element being separated from its neighbouring elements by the laser pulse repetition rate. Diddams et al. [23] showed that this frequency comb can be used for high resolution optical spectroscopy. Indeed, these authors were able to record spectra of transitions in iodine vapour with a resolution of â€™330 MHz, or â€™ 5 10 4 nm at 633 nm. Since this work, the bene ts of femtosecond comb spectroscopy have been ex- tended into the near-IR spectral region [24],[25] but have yet to be extended to the terahertz region due to the lack of suitable frequency comb sources. Such high spectral resolution is particularly desirable at terahertz frequen- cies, due to the abundance of rotational molecular transitions in this fre- quency range, as shown in gure 1.2 for water vapour. Further, the study of vibrational transitions, with their long lifetimes and correspondingly narrow spectral widths would bene t greatly from high spectral resolution at tera- hertz frequencies, aiding electron dipole moment investigations in molecules such as ThO (895 cm-1) [26] and YbF (501 cm-1) [27]. In this work, we use optical recti cation to produce, via di erence fre- quency generation within a broad spectral bandwidth, a terahertz frequency comb. Further, we show that the power of such a frequency comb may be increased for a given driving optical power using resonant enhancement of the optical eld. In addition, we show that the resulting comb nature of the terahertz eld allows novel detection schemes capable of extending the methods of frequency comb spectroscopy to the terahertz spectral range. 1.2 Outline of Thesis In order to increase the signal to noise performance of a terahertz frequency comb spectrometer, it is desirable that the incident comb be of high power so as to allow the detection of low intensity absorption features. In this thesis, we begin in Chapter 2 by reviewing the basic principles underlying 51.2. Outline of Thesis terahertz generation via optical recti cation. Further, we extend this theory to the case where the optical recti cation process is driven by a mode-locked optical frequency comb. In Chapter 3 we consider a method of generating a high-power terahertz frequency comb by increasing the power in the driving optical frequency comb. The enhancement of the optical eld is achieved through the use of a passive optical resonator, similar in operating principal to those used to increase the e ciency of other nonlinear optical processes, namely high harmonic generation [28], [29]. The theory of operation of such an enhancement cavity is reviewed before studying the design of the cavity in detail. In particular, the e ect of various orders of phase delay on the optical enhancement within the cavity are considered. To this end, we present a scheme for controlling the average resonant frequency of the enhancement cavity, along with methods of compensating for the large group velocity dispersion of the nonlinear crystal used to generate the terahertz eld. Performance estimates of the enhancement cavity are made before reviewing experimental measurements of the optical enhancement achieved. Factors limiting the optical enhancement are discussed. In Chapter 4 we demonstrate a terahertz time-domain spectrometer us- ing free space electro-optic sampling and a swept mechanical delay to vary the overlap of the terahertz eld with a sampling optical probe eld in a nonlinear detection crystal. In order to enhance the nonlinear conversion e ciency of the terahertz generation process, we use a gallium phosphide waveguide structure to con ne the strongly diverging terahertz eld to the interaction region of this long crystal. Whilst our experimental arrangement demonstrates good signal to noise over the spectral range of 0.5-1 THz, data acquisition is time consuming due to the need to vary the length of the op- tical path of the probe eld using the mechanical delay, Further, the trans- lating delay line leads to instabilities in alignment, which gives rise to drifts in the time-domain signal. Chapter 5 outlines in detail a method for high resolution terahertz comb spectroscopy with high speed data acquisition. Instead of a mechanical de- lay line, the proposed scheme makes use of a heterodyne detection scheme whereby detection of the terahertz frequency comb is performed using an 61.2. Outline of Thesis optical frequency comb di ering slightly in pulse repetition rate to that of the terahertz comb. In this sense, we refer to this method as dual-comb terahertz spectroscopy. We show, with a fully frequency domain analysis, that frequency resolution limited only by the relative stability of the laser pulse repetition rates can be achieved. At the same time, the technique is shown to o er data acquisition rates far greater than those of conventional swept delay terahertz time domain spectrometers. Finally, we discuss ex- perimental progress towards implementing this detection scheme in Chapter 6. A method of controlling the repetition rate detuning of the terahertz and sampling optical combs is presented and characterised. Having determined the central frequency and bandwidth of the terahertz eld in the work of Chapter 4, several methods of measuring the terahertz frequency comb at the rf heterodyne beat frequency calculated in Chapter 5 are presented and discussed. 7Chapter 2 Background 2.1 Optical Recti cation It is well known that the response of a dielectric material to a strong monochromatic optical eld can be divided into a linear and a nonlinear portion. More speci cally, when driven by a monochromatic, plane wave of the form E(t) = E0e i!t + c.c.; (2.1) the material will exhibit a polarisation given by P (t) = "0 (1)E(t) + (2)E(t)E(t) + (3)E(t)E(t)E(t) + ::: (2.2) where (n) is the nth order nonlinear susceptibility of the medium. Note that, for clarity, we consider only scalar elds and neglect any spatial depen- dence of the eld for the present discussion. In principle, the above relations apply to each vector component of the eld and resulting polarisation, in which case the susceptibilities become tensors. See, for example, [30]. The rst term in equation 2.2 is the usual linear response, whilst the second and third constitute the second and third order responses respectively. Consider- ing in more detail the second order response to the electric eld of equation 2.1, we obtain a second order nonlinear response given by P (2)(t) = ( (2)E20e 2i!t + c.c.) + 2 (2)E0E 0 : (2.3) The rst of these terms is the commonly recognised sum frequency term, whilst the E0E 0 term represents a contribution at zero frequency, and de- scribes optical recti cation. It is this latter term that interests us. Whilst 82.1. Optical Recti cation the above is true for the monochromatic eld of equation 2.1, for pulsed radiation with a broadband spectral pro le this dc polarisation follows the temporal envelope of the incident optical pulse as shown in gure 2.2, an e ect rst experimentally detected by Bass et al. in 1962, [31]. To analyse the e ect of this optical recti cation process on the output pulse train of a mode-locked oscillator, we consider the latter in more detail. For a mode-locked oscillator, the laser output can be described in the time domain as a train of pulses, of separation rep = 1 frep , where frep is the pulse repetition rate and the duration of each pulse, , is assumed to be much less than rep. The electric eld is of the form of a high frequency carrier wave at the central optical frequency !c = 2 fc, modulated by an envelope function which is periodic in rep, as shown in gure 2.1. Further, the maxima of the high frequency carrier wave are o set from the maxima of the envelope by a phase ce, known as the carrier-envelope phase. Following the methodology of [22], the electric eld of the pulse train is given by E(t) = 1X n= 1 E(t n rep)e i(!ct+n( ce !c rep)+ 0); (2.4) where the in nite limits on n imply that the pulse train contains an in nite number of pulses. In this expression, ce = n ce+ 0 is the carrier-envelope phase of the nth pulse, ce being the pulse-to-pulse carrier-envelope phase shift. E(t n rep) is the temporal envelope function which, as required, is periodic in rep and has a maximum for t n rep = 0. Taking the Fourier transform of this expression, and rearranging, we obtain [22] ~E(!) = X n ei(n( ce !c rep)+ 0)e i(n! !c) rep ~E(! !c) =ei 0 ~E(! !c) X M ( ce ! rep 2 M); (2.5) where M runs over the frequency bandwidth of the optical pulse. The delta 92.1. Optical Recti cation function in the nal term describes a comb of frequency elements at !M = 2M rep ce rep : (2.6) Or, equivalently, fM = Mfrep + f0 (2.7) where f0 = cefrep 2 is the o set of the lowest frequency from zero (dc) frequency. The laser spectrum therefore has the form of a comb of evenly- spaced frequencies of spacing frep with o set from dc by an amount f0, determined by the pulse-to-pulse phase-shift, ce, as illustrated in gure 2.1. The time domain expression for the electric eld, equation 2.4, can therefore be expressed as a sum of these spectral frequency components as E(t) = X M A(!M )e 2 i(Mfrep+f0)t (2.8) where A(!M ) is the complex spectral amplitude of the spectral line at fre- quency !M . In equation 2.8 the sum over M again runs over the entire bandwidth of the optical spectrum. Having derived an expression for the output of a mode-locked oscillator as a function of time, we now return to our consideration of the nonlin- ear polarisation of equation 2.3. Substituting equation 2.8 into the optical recti cation term of equation 2.3 gives P (2)DFG(t) = 1X M;N=0 N<M (2)(!M !N ;!M ; !N )A(!M )A (!N ) e 2 i(Mfrep+f0)te2 i(Nfrep+f0)t: (2.9) The limits on the sum now arise from assuming there is no contribution to a polarisation at negative frequency, and we assume the same conditions on M and N in the expression to follow. Simplifying, and using the fact that 102.1. Optical Recti cation Figure 2.1: The laser pulse train from a mode-locked oscillator can be de- scribed in either the time domain (top) or the frequency domain (bottom). In the time domain, the carrier-envelope phase evolves by an amount be- tween pulses. In the frequency domain, the output of the laser is equivalent to a discrete comb of frequencies, each separated by the laser pulse repetition rate, frep and o set from zero frequency by an amount f0 = frep 2 . 112.1. Optical Recti cation A(!) is real: P (2)DFG(t) = X M;N (2)(!M !N ;!M ; !N )A(!M )A (!N ) e 2 i(M N)frept: (2.10) Equation 2.10 shows that every frequency component in the input optical Figure 2.2: Decomposition of material response to pulsed radiation into sum and di erence frequency terms. The di erence frequency term gives rise to a dc polarisation that will rise and fall with the pulse envelope. pulse mixes with every other frequency component, producing a comb of di erence frequencies at fTHz = X M;N (M N)frep: (2.11) We note that this new comb necessarily has a zero carrier-envelope o set frequency. It is also important to note that, whilst the input frequencies are all entirely in the optical range of frequencies, that is, around 290 THz for optical wavelengths of 1030 nm, the di erence frequencies will be centred at dc, and extend out to the bandwidth of the original optical pulse, as shown 122.1. Optical Recti cation Figure 2.3: DFG for a frequency comb gives rise to a second frequency comb, centred at dc and with a necessarily zero o set frequency. in gure 2.3. 2.1.1 Calculation of the Radiated Field In order to determine the power radiated at each of the di erence frequencies as a result of the nonlinear polarisation of equation 2.10, we treat each frequency component of the driving optical eld as a continuous wave (CW). We then compute the magnitude of the eld radiated at each of the resulting di erence frequencies using the nonlinear wave equation: r2E = 0 @2E @t2 + 0 @2 @t2 PNL; (2.12) where E is the sum of the three eld components involved. That is, the mixing components of the driving optical eld, and the resulting radiated eld due to the nonlinear polarisation component PNL. Away from a res- onance, Kleinman symmetry dictates that the second order nonlinear sus- ceptibility, (2)(!M !N ;!M ; !N ) is independent of frequency. We will see that terahertz generation via optical recti cation often relies on some fre- quencies being near an optical resonance so as to satisfy the condition of phase-matching. Regardless, this approximation is adequate for the pur- poses of this discussion. In the case of the nonlinear susceptibility be- ing independent of frequency, we use the e ective nonlinear coe cient, 132.1. Optical Recti cation deff = 12 (2)(!M !N ;!M ; !N ). Further, third and higher order contri- butions to the nonlinear polarisation are neglected, such that PNL(t) â€™ P (2)(t) = 2deffE(t)E (t). Finally note that, in the semiconductor crys- tals considered in this work, deff will vary in magnitude depending on the polarisation of the elds, due to the rotational asymmetry of the crystal lattice. We consider the generation of a single di erence frequency, !2, generated from the nonlinear interaction between two input optical frequencies at !1 and !3, with !2 = !3 !1. Further, in order to account for the growth of the di erence frequency eld with propagation, we include the dependence of the electric eld with propagation direction, such that, for the three elds, the eld at frequency !i is described by E!i(t; z) = Eie i(!it kiz) + c.c. (2.13) where ki is the magnitude of the wavevector ~ki of the eld E!i , and Ei is the complex envelope. By virtue of the fact that E in 2.12 is the sum of the three elds in 2.13, the nonlinear polarisation serves to couple the eld amplitudes and it is necessary to nd all of the sum and di erence frequency components of the nonlinear polarisation, such that we may solve the wave equation. The nonlinear polarisation at the di erence frequency !3 !1 is given by P!3 !1NL (t; z) = deffE3E 1e i[(!3 !1)t (k3 k1)z] + c.c. (2.14) Similarly for the other frequency components P!3 !2NL (t; z) = deffE3E 2e i[(!3 !2)t (k3 k2)z] + c.c. (2.15) P!1+!2NL (t; z) = deffE3E 2e i[(!1+!2)t (k1+k2)z] + c.c. (2.16) It is these nonlinear polarisations that serve as dipole sources for the radi- 142.1. Optical Recti cation ated di erence frequency signal. In order to obtain the set of coupled wave equations, we substitute these expressions into 2.12 and make the slowly varying amplitude approximation, d2 dz2 EN << kN d dz EN : (2.17) That is to say that the amplitude of any one eld varies slowly on the scale of one wavelength. The nal result yields d dz E1 = i2 !21 c2 1 k1 deffE3E 2e i(k3 k2 k1)z (2.18) d dz E2 = i2 !22 c2 1 k2 deffE 1E3e i(k3 k2 k1)z (2.19) (2.20) We now make the assumption of perfect phase-matching such that k = k3 k2 k1 = 0 which allows us to drop the exponential dependence. We also make the important assumption of an undepleted pump for the eld E!3 . That is, that deff is small such that the eld amplitude E3 remains unchanged. This allows us to de ne the constants K1 = 2 i !21 c2 1 k1 deffE3 K2 = 2 i !22 c2 1 k2 deffE3; (2.21) such that equation 2.18 and 2.19 can be written d dz E1 = K1E 2 (2.22) d dz E2 = K2E 1 (2.23) : (2.24) 152.1. Optical Recti cation Di erentiating and substituting yields d2E1 dz2 = K1K 2E1; (2.25) which has the solution E1(z) = C sinh(Kz) +D cosh(Kz); (2.26) where C and D are constants to be determined by boundary conditions, and K = p K1K 2 . Substituting this into the expression for E2 yields E2(z) = K K 1 [C cosh(Kz) +D sinh(Kz)]: (2.27) Applying the boundary condition that the di erence frequency eld, E2(z = 0) = 0 at the crystal entrance face, so that C = 0. The nal solution for the di erence frequency eld is then E2(z) = i E3 jE3j E1(0) r k2n1 k1n2 sinh(Kz); (2.28) which can be squared to yield the intensity of the di erence frequency eld I2 = [jE1(0)j 2k2n1 k1n2 sinh2(Kz)]: (2.29) For perfect phase-matching, the intensity of the di erence frequency eld therefore grows exponentially with propagation length, as shown in gure 2.4. The topic of phase-matching will be reserved for section 2.3. For a single pass con guration, it is therefore desirable to use as long a nonlinear crystal as possible. However, typically the condition of perfect phase-matching can only be satis ed for narrow terahertz bandwidths. For broadband generation of terahertz radiation it will be necessary to reduce the crystal length so as to not exceed the propagation length over which the optical and di erence frequency elds remain in phase. Further, optimal use of such a nonlinear crystal in an optical resonator requires certain other crystal properties to 162.2. Beam Polarisation and Crystal Orientations Amplitud e z Figure 2.4: Growth of the di erence frequency eld, E2(z) with propagation distance for the case of perfect phase-matching, k = 0. be accounted for, namely, the material dispersion and the symmetry of the crystal lattice. These issues form the basis of the following discussions. 2.2 Beam Polarisation and Crystal Orientations In order to determine the optimum beam polarisations and crystal orienta- tion so as to maximise the e ciency of terahertz generation, it is necessary to account for the vector nature of the elds in equations 2.14-2.16. We use the notation ~E!N to denote the time-dependent eld at !N . Further, the nonlinear crystals used are in general not isotropic, such that the ef- fective nonlinear optical coe cient, deff is a tensor that depends on the crystal structure. However, for cubic crystals as per gallium phosphide and zinc telluride, all elements of the nonlinear optical coe cient must vanish, 172.2. Beam Polarisation and Crystal Orientations except for d14 = d25 = d36 = d, giving P!2x P !2 y P !2 z = 0 B @ 0 0 0 d 0 0 0 0 0 0 d 0 0 0 0 0 0 d 1 C A 0 B B B B B B B B B @ E!1x E !3 x E!1y E !3 y E!1z E !3 z E!1y E !3 z + E !3 z E !1 y E!1x E !3 z + E !3 x E !1 z E!1x E !3 y + E !3 x E !1 y 1 C C C C C C C C C A ; (2.30) For the case of optical recti cation of a broadband optical pulse from a single laser source, the input elds ~E!1 and ~E!3 are clearly parallel, ie. E!1i = E !3 i = Ei. The matrix expression above then simpli es to P!2x P !2 y P !2 z = 2d 0 B @ EyEz ExEz ExEy 1 C A : (2.31) In order to determine the optimum beam polarisation for terahertz gen- eration, we de ne the coordinate system of gure 2.5. The optical pulse is incident on a [110] gallium phosphide crystal [32] at an angle of incidence, , and with an azimuth angle . The electric eld is tangentially polarised, so as to minimise re ections from the crystal surface at Brewsterâ€™s angle of incidence, calculated from Sellmeier data in [33] to be 73 . Transforming into the frame of the crystal, calculating the polarisation according to equa- tion 2.31, and transforming back into the frame of the input eld allows the resulting nonlinear polarisation to be expressed as sagittal and tangential components P!2 / 1 2 sin ((3 cos 2 + 1) cos2 + 2 sin2 ) 3 cos cos (sin2 cos2 sin2 ) : (2.32) In gure 2.6 the relative magnitude of the nonlinear polarisation is plotted 182.2. Beam Polarisation and Crystal Orientations Figure 2.5: Coordinate system used for deriving the optimum optical beam orientation for terahertz generation. for varying and , along with the transects through this plot for normal incidence, ( = o) and at Brewster incidence, where refraction leads to = 17:8 . It is seen that, at Brewster incidence, the amplitude of the nonlinear polarisation is â€™25% down on the maximum that can be obtained at normal incidence. Whilst further increases in e ciency could therefore be achieved by using a custom-grown crystal where the [110] crystal axis is at Brewsterâ€™s angle to the surface normal, we instead use here a more readily-available [110] crystal for reasons of economics. Further, we note that, in order to maximise the terahertz generation e ciency, the optimal azimuthal angle is 57 to the [001] crystal axis. Mounting the crystal in a conventional rotation mount allows this to be tuned experimentally. For the cavity experiment, where we couple the generated terahertz radiation out of the polished end face, the edges are cut as in gure 2.7. 192.2. Beam Polarisation and Crystal Orientations ï¿½(degrees ) 0 45 90 0 0.2 0.4 0.6 0.8 0 45 90 135 1800 0.2 0.4 0.6 0.8 1 ï¿½ (degrees) Relativ eN LP olarisati on ï¿½ = 0 ï¿½ = 17.8 ï¿½ Figure 2.6: Relative nonlinear polarisation as a function of incidence angle, , and azimuthal angle, , as de ned in gure 2.5. Cuts are shown for normal incidence, = 0, and for incidence at Brewsterâ€™s angle, which leads to a propagation direction in the crystal such that = 17:8 . 202.3. Phase-matching Figure 2.7: Dimensions, cuts, and orientation of gallium phosphide crystal for enhancement cavity work. 2.3 Phase-matching The nonlinear polarisation produced by the driving optical eld will propa- gate with the envelope of the driving optical pulse, that is, with the optical group velocity, vgopt. In order that this generated polarisation remain in phase with the radiated terahertz eld, we seek to match this optical group velocity to the phase velocity of the terahertz eld, vpTHz. The distance over which the constructive interference between the radiated terahertz eld and the nonlinear polarisation due to the driving optical eld is maintained is the coherence length of the interaction. In contrast to other nonlinear opti- cal processes, where one seeks to match the phase velocities of the driving and generated radiation, optical recti cation requires the matching of the optical group velocity to the terahertz phase velocity. In order to determine the coherence length, we follow the method of [34] in de ning an e ective optical refractive index, or optical group index: n(group)opt = nopt opt dnopt d opt (2.33) 212.3. Phase-matching where nopt is the conventional optical refractive index of the material in which the radiation propagates, and opt is the wavelength of the optical radiation. The coherence length for the nonlinear optical process can then be written Lc = c !THzjn (group) opt nTHzj : (2.34) Many nonlinear materials used for terahertz generation exhibit large dis- persion at terahertz frequencies owing to a number of absorption features. For example, gallium phosphide exhibits a phonon resonance around 10 THz, and therefore nTHz is signi cantly greater than the optical refractive index. However, the absorption edge at 2.25 eV ( 540 THz, 550 nm) gives rise to substantial dispersion at optical wavelengths, such that dnoptd is signi cantly less than zero. Careful tuning of the optical wavelength can then give rise to excellent phase-matching at around 1 m , where the e ective optical refractive index is equal to the terahertz refractive index, as shown in gure 2.8. To emphasise this point, the coherence length for a range of terahertz frequencies and optical wavelengths, as predicted by equation 2.34 in gal- lium phosphide at room temperature is plotted in gure 2.9. Further, cuts through this contour plot are given for the optical wavelength of 1030 nm produced by an ytterbium bre laser system, and at a terahertz frequency of 1 THz. The former cut shows that, under such operating conditions, long coherence lengths of up to several millimetres can be generated. In addition, the coherence length as a function of driving optical wavelength is plotted for a terahertz frequency of 1 THz. It is seen that the optimal optical wavelength falls conveniently near that of an ytterbium bre laser system. Driving a gallium phosphide nonlinear crystal with the 1030 nm output of a mode-locked ytterbium-doped glass bre laser therefore repre- sents a very attractive method for the generation of terahertz radiation via optical recti cation. 222.3. Phase-matching Figure 2.8: Variation of refractive index throughout the optical and terahertz spectral ranges from data in [33], along with an experimental t of the e ective refractive index for optical wavelengths, as per equation 2.33. Note that, due to the resonance at 2.25 eV, tuning of the optical wavelength allows the e ective refractive index to be matched to the terahertz refractive index, therefore satisfying the phase-matching condition. 232.3. Phase-matching 0.0 0.5 1.0 1.5 2.0 2.5 3.0 400 600 800 1000 1200 1400 0.0 0.5 1.0 1.5 2.0 2.5 3.00 5 10 15 20 400 600 800 1000 1200 1400 012345 Figure 2.9: Coherence length for terahertz frequency generation via optical recti cation in gallium phosphide, as predicted by equation 2.34. Cuts are shown for optical wavelengths of 1030 nm, and for terahertz frequencies of 1 THz. 242.4. Lasers 2.4 Lasers 2.4.1 Oscillator Design Due to the quadratic dependence of the generated nonlinear polarisation on the input electric eld predicted by equation 2.3, it is desirable to use short pulses of intense optical radiation to drive the process, rather than a constant amplitude, continuous wave (CW) beam. Further, a broad optical spectrum is necessary to generate di erence frequency components in the terahertz frequency range. By far the most common method of generating short, broadband optical pulses is to use mode-locked lasers. In order to generate the short pulses required for this experiment, mode- locked ytterbium bre lasers of the type described in [35] are used. An overview of each laser is given in gure 2.10. The rst, identical in design to that of [35], is mode-locked by virtue of the nonlinear polarisation in the optical bre components. (SMF Corning Hi-1060). High peak intensity pulses generated in a length of ytterbium-doped gain bre (Cor-Activ DC- 170-40-YB-2) exhibit a stronger nonlinear polarisation rotation (NPR) than the low peak intensity CW case. This polarisation rotation is converted into an amplitude modulation at the repetition rate of the pulse in a section of bulk polarisation optics arranged as in gure 2.10a) [36]. The combined e ect of the NPR and polarisation optics is therefore analogous to that of a saturable absorber, such that intense pulses experience reduced intracavity losses compared to those of the continuous wave (CW) case. In order to give stable mode-locked operation, the bandwidth, and therefore the mini- mum pulse duration, is limited by spectrally ltering the intracavity pulses using a 10 nm bandwidth birefringent lter centred at 1030 nm. Further, the constant normal dispersion of the laser, combined with gain saturation, means that the intracavity pulse undergoes strong spectral breathing in a single cavity roundtrip. Further details are given in reference [35]. The second laser was modi ed from a cavity con guration initially assembled by Simon Bret, and relies on the same physical principles as the laser discussed previously, but the components are arranged such that a bulk high re ec- tivity mirror can be incorporated into the cavity. This mirror can then be 252.4. Lasers mounted on a piezzo-electric actuator and used to make small scale, but high speed adjustments to the cavity length, and hence the pulse repetition rate. For both laser designs, a portion of the intracavity power is coupled out of the cavity using a 90/10% bre coupler (FS). A bre-coupled wavelength- division multiplexer (WDM) is used to further isolate the pump radiation of subsequent ampli cation stages from the laser cavity. For each laser, 10% of the intracavity power is coupled out of the cavity to provide an output power of approximately 12 mW. A further 10% of this output power is picked o using bre splitters for the purpose of triggering the rf electronics used in the dual-comb experiment. This fraction of the laser output serves only as a diagnostic port in the case of the cavity experiment. The outputs of each laser are ampli ed in bre coupled, forwards-pumped ytterbium-doped bre ampli ers, henceforth referred to as pre-ampli ers so as to distinguish this stage from the later power-ampli cation stage. Each preampli er gives an average power output of about 180 mW. In the case of the rst laser, the output of the pre-ampli er stage is passed through a large aperture optical isolator (IsoWave part no:I-106-T5-L), and coupled into a 1.5m length of ytterbium-doped photonic crystal bre (PCF) for the nal ampli cation stage, henceforth the power ampli er. The PCF is backwards pumped with a 40 W 976 nm pump diode (Apollo Instruments part no:F20- 976-1) which has had the drive circuitry modi ed to reduce the amplitude of line harmonics on the high current supply (see section 2.4.3). Due to the fact that none of the bre components used in this apparatus are polarisation maintaining (PM) bres, the output of any bre stage is, in general, elliptically polarised. In order to obtain linear polarisation it is necessary to use a quarter wave plate-half wave plate pair to provide ad- justment of an arbitrary elliptical polarisation to a given linear polarisation state, as required for use with optical isolators, di raction gratings, e cient coupling into the enhancement cavity, and optimum terahertz generation in electro-optic crystals. 262.4. Lasers (a) Laser oscillator design used to provide the high-power (following two subse- quent ampli cation stages) pulses for both the cavity work and the single-comb, terahertz time-domain spectroscopy work described later. (b) Laser oscillator design used to provide the local oscillator pulses for the work described in Chapter 6. Figure 2.10: Design of the two mode-locked ytterbium bre lasers. The operation of each laser is described in the text. Parts labelled are: quarter wave plate (QWP), half wave plate (HWP), interference lter (IF), piezo- electric actuator (PZT), polarisation beam splitter (PBS), 90/10% bre cou- pler (FS), and bre-coupled wavelength-division multiplexer (WDM). 272.4. Lasers 2.4.2 Compensation of Pulse Chirp due to Dispersion of Fibre Components Due to dispersive e ects in the bre components used in the oscillator and ampli er stages, the output pulses are highly chirped. The dependence of the material refractive index on radiation wavelength means that, within the broad spectral amplitude of the pulse, each spectral component will prop- agate with a slightly di erent phase velocity. This is the e ect of material dispersion. In addition, the con nement of the electric eld within the di- electric medium of the bre, and the resulting dependence of the degree of eld con nement on the wavelength of the radiation leads to waveguide dis- persion. We neglect the e ect of dispersion between transverse modes here as the bre components used only allow propagation of a single transverse mode at our wavelengths. The combined e ects of waveguide and material dispersion are encompassed in the mode propagation constant (!) and pro- duce a frequency-dependent phase delay on the optical pulse. Writing the pulse as the product of a carrier wave at the central optical frequency !c, and an envelope function describing the pulse propagation, a pulse having a continuous spectral amplitude is described as E(z = 0; t) = ei!ct Z ~E( )ei td ; (2.35) where !c represents the optical carrier frequency, and ~E( ) is the Fourier transform of the pulse envelope. Propagation of this pulse over a length z of a dispersive medium writes a frequency-dependent phase onto the eld, such that E(z; t) = ei!ct Z ~E( )ei( t (!c+ )z)d (2.36) where (!c+ ) is the mode propagation constant at frequency ! = !c+ . It is common to expand near the optical carrier frequency in a Taylor series thus: (!c + ) = (!c) + d d! !c + 1 2 d2 d!2 !c 2 + 1 6 d3 d!3 3 + : : : : (2.37) 282.4. Lasers It can be shown that d d! = 1 vg where vg is the group velocity, and that therefore the group velocity dispersion is given by d d! 1 vg = d2 d!2 : (2.38) It is common to de ne the group velocity dispersion parameter, D, to char- acterise the optical bre in which the pulse propagates by D = 2 c v2g 2 dvg d! = 2 c 2 d2 d!2 ; (2.39) The frequency dependent phase-shift in equation 2.36 gives rise to a temporal broadening of the pulse and therefore a lowering of the peak electric eld intensity. As optical recti cation is a nonlinear e ect whose e ciency will increase with the amplitude of the driving electric eld, it is highly desirable to use the shortest pulses possible. To second order, this is the case when the quantity d2 (!) d!2 = 2 c 2 d2 d!2 z = zD (2.40) is minimised. d 2 (!) d!2 is termed the group delay dispersion (GDD). d3 (!) d!3 , as given by the fourth term in equation 2.37, is the third order dispersion (TOD) delay. In order to correct for the dispersive e ect of the bre components and re-compress the optical pulses, an optical pulse compressor employing a di raction grating pair is used for each laser. The principal of operation of such a device was rst outlined by Treacy [37]. Fork et al. [38] showed that the amount of GDD induced by propagating an optical pulse through a system of di raction gratings, con gured as in gure 2.11, is given by d2 g d!2 = 3lg c2d2 " 1 d sin 2 # 32 (2.41) 292.4. Lasers where lg is the separation of the gratings, is the angle of incidence and d is the groove spacing for the gratings. In order to estimate the required separation of the grating compressor, we start from manufacturers data for the single mode bre used in the laser system. In the case of the lasers used here, the total length of bre is 15 m, and the GVD is given as 27 ps2/km. The gratings used are transmission di raction gratings with a ruling of 1200 lines/mm, or d â€™ 830 nm, and an optimum angle of incidence of 41.7 . From equation 2.41 the approximate required separation is therefore 4 cm. Note that this estimate neglects third order dispersion (TOD) of both the grating compressor and bre components. Treacy et al. showed that, for a typical grating compressor, the sign of the TOD term is positive, as opposed to the negative GVD term. In contrast, in typical single mode bres, both the TOD and GVD terms are positive. The use of a grating compressor to compensate for the GVD of the bre components therefore increases the amount of TOD in the total optical system, which leads to the temporal duration of the resulting pulse being greater than the predicted transform limit [39]. In practice lg was set to the predicted value of 4 cm, and then adjusted experimentally to give optimum pulse compression. The optimum grating separation was found to be 37 mm, and the resulting pulse compression is shown in gure 2.12, together with ts to Gaussian and sech2 pulse shapes of 240 fs full width at half maximum durations. Note that, for the observed 7 nm spectral bandwidth, one expects a temporal duration of 160 fs at the Fourier transform limit. The fact that the minimum pulse width attainable is 240 fs is attributed to TOD, arising both in the bre components and in the grating compressor. Whilst methods have been demonstrated that produce an output pulse closer to the transform limit by minimising the e ects of both GVD and TOD, [40], the additional components required signi cantly reduce the nal pulse energy. For this work, therefore, we choose not to employ such techniques. 302.4. Lasers Figure 2.11: Con guration of the di raction grating pulse compressor used for dispersion compensation. lg is the separation of the two component grat- ings along the grating normal, is the angle of incidence, and is determined by the blaze angle of the grating. The transect MMâ€™ shows the line of sym- metry of the system. In practice, the beam is re ected back through the same pair of di raction gratings using a metal mirror, thereby compensating for the spatial chirp across the beam pro le arising from a single pass. 312.4. Lasers Figure 2.12: Typical autocorrelation trace of the laser output pulse (blue line) of the pump laser following compression, together with ts to Gaus- sian (solid black line) and sech2 (dashed line) pulse shapes, both of 240 fs full width at half maximum. Inset: Comparison of compressed and uncom- pressed pulse widths. 322.4. Lasers 2.4.3 Pump Diode Line Noise During the experimental work, it was noticed that the optical output of the pump diode used for the power ampli er stage displayed modulations at harmonics of the 60 Hz line frequency, and that these modulations were written onto the pulse train of the mode-locked oscillator at the output of the high-power ampli cation stage, as indicated in gure 2.13. This was measured using a fast photodiode and rf ampli er circuit positioned behind the dichroic mirror used to couple the seed pulse into the input collimator of the PCF at the power ampli er stage. The origins of these modulations were eventually traced to the linear voltage supply used to provide a control signal to a power transistor, internal to the diode controller provided by Apollo instruments, a schematic of which is shown in gure 2.15 with the linear voltage supply in question indicated by Vcontrol. Whilst exact speci cations of the controller were not made available, it was found that deriving this control voltage from a known clean linear dc supply lessened the amplitude of the line harmonics on the controller output current ( gure 2.14) with a corresponding reduction in noise on the optical output. The ultimate bene t of driving the laser in this way can be seen from gure 2.13 to be a reduction in the amplitude of the line harmonics out to 360 Hz of approximately 5- 10 dB. In addition to replacing the control voltage with that from a clean, linear supply, the e ect of replacing the xed 12V, 60A supply (labelled Vdrive in gure 2.15) was examined, but it was found that this gave no further bene t over the data shown in gure 2.13. 332.4. Lasers âˆ’450 âˆ’360 âˆ’270 âˆ’180 âˆ’90 0 90 180 270 360âˆ’160 âˆ’140 âˆ’120 âˆ’100 âˆ’80 âˆ’60 âˆ’40 Frequency offset from approximate repetition rate of 104.3MHz [Hz] Power Level [dBm ] Internal Drive Circuit External linear DC Supply Figure 2.13: Comparison of modulation sidebands on the optical pulse train repetition rate of 104.3 MHz following the high-power ampli er stage with the controller in its original con guration, and following modi cations to allow the regulation of the transistor control voltage with an external linear dc supply. 342.4. Lasers 0 50 100 150 200 250 300 350 400âˆ’160 âˆ’140 âˆ’120 âˆ’100 âˆ’80 âˆ’60 âˆ’40 Frequency (Hz) Pow er S pe ctr al De nsi ty (d Bm /Hz) With External Driving Voltage Original Configuration Figure 2.14: Comparison of noise on the electrical current used to drive the Apollo Instruments pump diode, as measured across the shunt resistor, R shown in gure 2.15. Of particular note are the complete elimination of modulations at 180 Hz, and the substantial reduction of harmonics at 300 Hz. 352.4. Lasers Figure 2.15: Schematic of the Apollo Instruments laser diode driver, model no. D-560. The basic principal of operation is the use of a variable dc voltage, Vcontrol to regulate the current owing through a power transistor. The laser diode itself is connected in series with a xed 12V, 60A supply, Vdrive, and with the collector emitter terminals of the control transistor. It should be noted that actual schematics were not made available, and that this gure serves only to document the process involved in reducing the line harmonics on the optical output of the diode. It was found that replacing the internal supply used to provide Vcontrol reduced the appearance of line harmonics on the laser output, and that no further improvement was gained by replacing the xed 60A supply. 36Chapter 3 Use of a Passive Enhancement Cavity for High E ciency Terahertz Generation 3.1 Cavity Basics Equation 2.3 predicts that the second order nonlinear polarisation increases quadratically with the intensity of the driving electric eld, as is the case for all second order nonlinear optical processes. It is therefore desirable to increase the pump intensity as much as possible so as to increase the polarisation intensity, and therefore the intensity of the radiated di erence frequency eld. A number of options exist for doing this. For an input eld provided by a bre laser, it is possible to increase the laser pulse en- ergy using single pass bre ampli er stages, but at higher pulse energies the variation of the bre refractive index with optical intensity will lead to self phase modulation. The resulting new spectral components generated in this process lead to frequency chirping of the optical pulse, which cannot be simply compensated for at the laser output as was shown in section 2.4.2 for the case of GVD. This therefore increases the minimum pulse duration, and hence reduces the peak intensity, with which it is possible to drive the nonlinear crystal, reducing the potential gain in nonlinear conversion e - ciency. Alternatively, regenerative ampli ers can be used. These however typically operate on a â€˜pulse-pickingâ€™ scheme, whereby one in â€™ 105 seed 373.1. Cavity Basics pulses are ampli ed, thereby sacri cing repetition rate for peak power. Fur- ther, by reducing the pulse repetition rate in such a manner, the spacing of the frequency comb elements is reduced, resulting in di culties resolving individual frequency elements. A third option to increase the amplitude of the driving laser eld is to use a passive enhancement cavity in which the laser pulse is resonantly enhanced. This is particularly appealing for enhancement of nonlinear optical processes due to the relatively low single-pass conversion e ciency implied by the low values of deff . Essentially, in the enhancement cavity, the pump radiation not converted in the rst pass of the optical eld through the nonlinear crystal is \recycled", increasing the e ciency of conversion. In order to understand the enhancement process in more detail, consider a monochromatic plane wave incident on an optical resonator consisting of a pair of highly re ecting mirrors, as shown in gure 3.1. The incident eld is written Einc and that portion coupled into the resonator is denoted Ein. After one roundtrip, the eld is E1, after two roundtrips, E2, and so on. The total eld at the input mirror is then the sum of these elds: ET = Ein + E1 + E2 + E3::: (3.1) Upon travelling one roundtrip of the cavity, the wave undergoes a phase change = kd + M where d is the optical path length of the cavity, and M is the total roundtrip phase-shift due to re ection at the cavity mirrors. We can therefore de ne a complex roundtrip attenuation factor h = jrje i ; (3.2) where r is the roundtrip re ectivity factor for the eld that accounts for re ectivity losses at the mirrors, as well as any losses due to propagation through the cavity itself. This complex roundtrip attenuation factor then 383.1. Cavity Basics Figure 3.1: A simple two-mirror optical resonator. relates each successive term in equation 3.1. Ie: ET = Ein + hE0 + h 2Ein + h 3Ein + ::: = Ein(1 + h+ h 2 + h3 + :::) = Ein 1 h (3.3) The intensity then varies as the square of this electric eld: I / jEj2 = jEinj2 j1 hj2 = jEinj2 j1 jrje i j2 : (3.4) Using trigonometric identities, we rewrite this as I = Iin 1 + jrj2 2jrj cos ; (3.5) where Iin = c" 2 jEinj2 is the intensity transmitted by the input mirror. It is 393.1. Cavity Basics common to characterise the resonator in terms of its nesse, F = p jrj 1 jrj ; (3.6) to give I = Imax 1 + (2F ) 2 sin2( 2 ) : (3.7) Here, Imax is the maximum intracavity intensity one can hope to maintain given the cavity is tuned to perfect resonance, where sin( 2 ) = 0: (3.8) Imax is therefore given by Imax = Iin (1 jrj)2 = Iinc 1 Ric (1 p RicR)2 ; (3.9) where Iinc is the input intensity and R is the cumulative product of the more commonly known power re ectivity magnitude of all the mirrors except the mirror upon which the input beam is incident, known as the input coupler, with power re ectivity Ric. Ie. R = Y i Ri where Ri is the power re ectivity of the ith mirror, and the product runs over the i mirrors in the cavity. We note that it is straightforward to include cavity parasitic losses due to scattering etc in the factor R thus: R = (1 Lcav) Y i Ri (3.10) where Lcav is the roundtrip loss factor. The nesse of the cavity can be rewritten in terms of the power re ectivities as F = 4 p RRic 1 p RRic (3.11) A high nesse cavity is therefore produced when the roundtrip optical loss is low. Furthermore, the peak intensity, Imax is produced when on res- 403.1. Cavity Basics Figure 3.2: Resonant modes of an optical resonator. The cavity shows maxima for roundtrip phases of = 0; 2 ; 4 ::: onance, that is to say that the roundtrip phase, = 0; 2 ; 4 :::. The cavity therefore shows maxima in intracavity intensity (and therefore transmission via the non-unity re ectivity of any of the resonator mirrors) for roundtrip phase changes of integer multiples of 2 . When tuned to resonance, the cavity therefore gives an ideal ampli cation factor of Imax Iin = 1 Ric (1 p RRic)2 (3.12) which, for given R 1, has a maximum Imax Iinc = 1 1 R â€™ F at Ric = R: (3.13) Therefore, in order to obtain the maximum intracavity intensity, the losses at the input coupler should be equal to the roundtrip losses due to all other components in the cavity. Under these conditions, the cavity is said to be impedance-matched. 413.2. Ampli cation of a Femtosecond Optical Comb Using a Passive Enhancement Cavity 3.2 Ampli cation of a Femtosecond Optical Comb Using a Passive Enhancement Cavity Section 3.1 focussed on how a passive enhancement cavity can be used to increase the amplitude of an electromagnetic eld at a single laser frequency. For the purposes of this work, we wish to optimise the e ciency of the optical recti cation process described by equation 2.3, where the amplitude of the nonlinear polarisation scales with the square of the driving optical eld. It is therefore desirable to use short pulses to increase the e ciency of the optical recti cation process. We therefore seek to use a passive optical resonator to simultaneously enhance the frequency content of the broad optical bandwidth produced by the mode-locked laser described in section 2.4. In section 2.4, a mode-locked oscillator was shown to have an output that can be described in the frequency domain as a broad spectrum of evenly spaced laser comb lines. In order to calculate the ampli cation of such a broadband output, we consider the ampli cation of each of the â€™ 105 laser modes within the cavity. Recall equation 3.7 describing the enhancement of a monochromatic eld inside a passive enhancement cavity. I = Imax 1 + (2F ) 2 sin2( 2 ) : (3.14) Further, recall that the roundtrip phase of the enhancement cavity was writ- ten as = 2kd+ M , where where d is the optical path length of the cavity, and M is the sum of the phase-shifts occurring due to re ection at each of the cavity mirrors over a single roundtrip. For the following discussion, we generalise this by including the frequency dependence of explicitly, such that ! (!) thereby allowing for the e ects of dispersion as the pulse propagates within the cavity. Further, we include in an additional term X(!) corresponding to the frequency dependent phase delay due to transmission through a dispersive optic (in our case, this will be the non- linear crystal) placed within the cavity. The resonance condition therefore 423.2. Ampli cation of a Femtosecond Optical Comb Using a Passive Enhancement Cavity becomes (!) = k(!)d+ M (!) + X(!) = 2m ; (3.15) where M (!) accounts for the frequency dependence of the phase delay due to re ection of the eld at the cavity mirrors. The dispersion due to the air within the cavity is included in the dependence of the wavevector, k(!) on frequency. For the case of zero group velocity dispersion, where M and X do not depend on frequency, and the wavevector k(!) varies linearly with frequency such that k(!) = k(!c) + (! !c) 1 vg , we can apply the methods of section 2.4.2 to obtain 1 (!) = k(!)d+ M + X = d k(!c) + (! !c) 1 vg + M + X : (3.16) The resonant modes of the enhancement cavity therefore satisfy !m = 2vgm d + vg d dk(!c) + d!c vg M X : (3.17) In this dispersion-free case, the resonant spectrum of the cavity therefore has the form of an evenly spaced comb of frequencies, spaced from zero frequency by an amount given by the second term of equation 3.17. We note that, in the absence of dispersion, the phase and group velocities are equal such that vg = c, and equation 3.17 predicts a spacing of the cavity modes in frequency of FSR = 2 c d ; (3.18) which is the familiar cavity free spectral range. Recall now the frequencies of a mode-locked oscillator, as described by 1In section 2.4.2 we use the mode propagation constant, (!) to describe the evolution of the phase with propagation distance in the bre components of the laser system. In freespace, where it is not necessary to include the added e ect of waveguide dispersion, we use the wavevector, k(!). 433.2. Ampli cation of a Femtosecond Optical Comb Using a Passive Enhancement Cavity equation 2.6, !M = 2M rep ce rep = M!rep + !0: (3.19) which shows that the output of a mode-locked laser has a similar form of a comb of evenly spaced frequencies, spaced from zero frequency by the laser carrier-envelope o set frequency !0 = cefrep. The angular pulse repetition frequency is simply !rep = 2 frep. In the dispersion-free case of equation 3.17, it is therefore possible to obtain complete overlap of the resonant modes of the enhancement cavity with the frequency comb elements of the laser, as in gure 3.2a), such that each and every laser mode is enhanced within the cavity by an amount determined by equation 3.9. Doing so requires control of both the mode spacing and mode o set of the frequencies being ampli ed [28]. In order to study the behaviour of the cavity for the case where the modes do not overlap with those of the seed laser completely, we de ne the o set, p, of the cavity modes with respect to the laser modes thus: m = M + p = !M !0 !rep + p (3.20) Substituting this expression into equation 3.15 yields the cavity length for which each Mth laser mode is resonant d(!M ) = 2 !M !0 !rep + p M (!M ) X(!M )) 1 k(!M ) = 2 c n(!M ) 1 !rep 1 !M !0 !rep p ( M (!M ) + X(!M ))c !Mn(!M ) (3.21) Again, consider a simple case where the laser carrier-envelope o set fre- quency, !0 = 0, and the phase contributions of M and X are zero. Fur- 443.2. Ampli cation of a Femtosecond Optical Comb Using a Passive Enhancement Cavity Figure 3.3: Varying overlap of the resonant modes of the passive enhance- ment cavity (in blue) with the longitudinal modes of the mode-locked laser used as the seed (dashed lines). a) Both the mode spacing and o set from zero frequency are matched, giving complete overlap, and therefore optimal enhancement of every frequency emitted by the laser. In this case, every mode of the laser is enhanced by a factor determined by equation 3.9. b) The repetition rate of the cavity, frc is greater than that of the laser, frl giving only complete overlap for certain modes. Only these modes circulate within the cavity, with the remainder being re ected at the input coupler. The power circulating within the cavity is therefore reduced with respect to that of a). c) The cavity shows dispersion, as indicated by the frequency- dependent spacing of the cavity modes. Whilst it is possible to vary the degree of overlap by varying the mode spacing and o set, the even spacing of the laser modes is such that complete overlap is not possible for this case. d) The o set frequency of the cavity f0c is greater than that of the laser, f0l. Again, mode overlap, and hence enhancement is reduced. 453.2. Ampli cation of a Femtosecond Optical Comb Using a Passive Enhancement Cavity ther, assume the Mth mode of the laser is resonant with the mth mode of the cavity, such that p = 0. In this case, we obtain, d(!M ) = 2 c !rep : (3.22) We can therefore satisfy the resonance condition for every single laser mode simultaneously, providing FSR = !rep: (3.23) In the more general case where the cavity displays dispersion, the cavity mode spacing varies with frequency, as shown in gure 3.3c). In this case, it is not possible to obtain complete overlap of the laser and cavity modes, even with independent control of the mode spacing and o set frequencies. In this case, only those modes of the cavity corresponding to output modes of the laser will be ampli ed (the remaining modes of the cavity are not excited), reducing the power circulating within the cavity. In order to better understand the e ect of dispersion on the enhancement of the broadband optical eld, we use the method developed in [41], and consider the power circulating within the enhancement cavity when seeded with the eld of a mode-locked oscillator having the form of equation 2.5. Figure 3.4 shows the resonance map as a function of cavity length and optical wavelength. It is seen that, for p 6= 0, when the cavity length is varied away from that satisfying equation 3.23, the enhancement is reduced by virtue of laser modes at the edges of the pulse bandwidth not satisfying equation 3.15. In contrast, when the cavity is on resonance such that p = 0, the entire bandwidth of the seed laser resonates within the cavity. In this case, every single laser mode (and therefore the total optical power) is ampli ed according to equation 3.9. In the presence of cavity dispersion, the length at which each laser mode exhibits maximum enhancement varies with frequency, as predicted by equa- tion 3.21. For example, if the cavity exhibits a net roundtrip phase delay that is quadratic in frequency, as for the case of group velocity dispersion, one observes a â€˜bendingâ€™ of the resonance map as shown in gure 3.5. As a 463.2. Ampli cation of a Femtosecond Optical Comb Using a Passive Enhancement Cavity Wavelengt h(nm ) 1010 1015 1020 1025 1030 1035 1040 1045 1050 Mode Offset, p 0 1 2 3-1-2-3 0 0.5 1.0 Normalised field enhancement factor Figure 3.4: Main: Resonance map of a dispersion-free enhancement cavity seeded with a mode-locked laser with central wavelength 0 =1030 nm, optical bandwidth =7 nm, and repetition rate frep =100 MHz. The nesse of the cavity is â€™ 100 to better show the resonance features. Bottom: Intensity enhancement as a function of cavity length, showing the decreased performance of the cavity for p 6=0. Right: Intensity spectrum circulating within the cavity for p =0, where the cavity length = cfrep . 473.3. Cavity Design result of this spectral phase-shift, the number of laser modes that are reso- nant within the cavity at any given length is reduced, leading to a reduction in the power circulating within the cavity. Further, the spectral narrowing that occurs leads to a temporal broadening of the pulse, further reducing the peak power. This is undesirable in our case due to the quadratic depen- dence of the expected amplitude of the terahertz eld on the electric eld amplitude. We will see later that the nonlinear crystals typically used for generation of terahertz radiation via optical recti cation typically have large dispersion. For example, the group velocity dispersion of gallium phosphide is â€™1250 fs 2 mm . In order to compensate for this large dispersion, we use Gires- Tournois inerferometric mirrors with a speci ed group delay, such that the roundtrip group delay of the cavity is minimised. We model the expected performance of this cavity in the following section. 3.3 Cavity Design 3.3.1 Mirror Choice and Dispersion Compensation The passive enhancement cavity design studied here is portrayed in gure 3.6. We use a ring cavity design consisting of six mirrors in total. Section 3.2 showed that, in order that the maximum possible optical bandwidth resonates in the enhancement cavity it is necessary that the res- onance condition, equation 3.9, is satis ed for each spectral component of the laser output with which the cavity is seeded. Further, in the frequency domain this can be pictured as ensuring that the longitudinal modes of the enhancement cavity overlap with those of the seed laser. Equation 3.23 predicts that this requires that the free spectral range of the enhancement cavity, FSR is equal to the pulse repetition rate of the laser, !rep. For this work, a pump laser with a pulse repetition rate of nominally 104 MHZ was used, such that the required optical path length in air of the enhancement cavity is 2.88 metres. The optical cavity consists of a pair of curved mirrors, an input cou- pler, and several at â€˜foldâ€™ mirrors. The curved mirrors serve two purposes. 483.3. Cavity Design Wavelengt h(nm ) 1010 1015 1020 1025 1030 1035 1040 1045 1050 Cavity Spectrum at Max Power Input Spectrum Mode Offset, p 0 1 2 3-1-2-3 0 0.5 1.0 Normalised field enhancement factor Figure 3.5: Main: Resonance map for a passive enhancement cavity seeded with an identical mode-locked oscillator to that of gure 3.4. The cavity has the same nesse, but now exhibits a net group delay of 104 fs2. Right: Due to this mode bending, the maximum intracavity power is obtained for a cavity length that is slightly detuned from the resonance length for the central wavelength. Further, wavelengths at the edge of the laser bandwidth are not resonant, leading to a narrowing of the optical bandwidth circulating in the cavity. Bottom: There is an accompanying distortion of the intensity supported in the cavity as a function of cavity length, as previously reported in [41]. 493.3. Cavity Design ++ Figure 3.6: Experimental arrangement for the passive enhancement cavity. Layertec part numbers are given in italics. The cavity is a folded six mir- ror arrangement, with the input eld incident on the low-re ectivity face of a Layertec input coupler (IC) which is changed to vary the cavity nesse. Mirror and IC speci cations are given in table 3.1. The nonlinear gallium phosphide crystal is placed at the focus of two concave mirrors, having ra- dius of curvature Ri as shown, so as to further increase the e ciency of the nonlinear process. The optical path length, as measured in air, between the two curved mirrors is l1 along the path including the IC, and l2 along the path including the nonlinear crystal. In order to maintain the reso- nance condition required by equation 3.9 a spectral portion of the power re ected by the optical cavity is monitored on a fast photodetector, and an appropriate locking scheme used to control the position of one of the cav- ity mirrors (ATF) using a piezzoelectric actuator (PZT). This mirror has 7.75 mm diameter and 3 mm thickness so as to minimise the load on the PZT. The mirror itself has a speci ed power re ectivity of 99.99% at normal incidence, and negligible GDD. 503.3. Cavity Design Firstly, to increase cavity stability over that of a planar mirror resonator and, secondly, to create an intracavity focus at which to place the nonlinear crystal, where the incident optical eld intensity is increased, enhancing the e ciency of the di erence frequency generation process. Various input cou- plers, as listed in table 3.1, were used in order to yield the required cavity nesse. An arbitrary number of at mirrors can be used to yield the required beam path and dispersion characteristics, although it should be noted that the small losses from re ection at each mirror are not inconsiderable over many cavity roundtrips. In order to minimise re ection losses at the entry face, the nonlinear crystal is placed at Brewsterâ€™s angle to the horizontally polarised incident optical beam. For gallium phosphide, this is calculated to be 72.2 for the optical wavelength of 1030 nm. In selecting the curved mirrors and their separation, one must satisfy the resonator stability condition of equation 3.50. For the ring cavity considered here, we obtain 2 1 R1R2 [(R1 +R2)(l1 + l2) + l1l2] 0; (3.24) where l1 and l2 are the cavity lengths between the intracavity curved mir- rors, as shown in gure 3.6, and R1 and R2 are the radii of curvature of these curved mirrors. The resonator used in this work uses two identical convex mirrors with radius of curvature 750 mm, such that a beam waist of 190 m is formed halfway between the two curved mirrors. This spot size was chosen so as to maximise the e ciency of terahertz radiation generation in accordance with the ndings of Zhang et al. [42] who found experimen- tally that terahertz generation was maximised when the radius of the optical spot size was comparable to the terahertz wavelength. For a frequency of 2 THz, a spot radius of 150 m is therefore optimal. Numerical simulations of the Gaussian beam parameters (see section 3.5.1) for a cavity resonator produced the cavity design in gure 3.6. Recall the conclusion of section 3.2 that, for a pulsed laser, the condition of resonance requires that the roundtrip phase of the cavity must satisfy the resonance condition, equation 3.8 for the entire pulse bandwidth. If 513.3. Cavity Design 600 800 1000 1200 1400 1600 1800 2000 3.00 3.10 3.20 3.30 3.40 3.50 500 1000 1500 2000 2500 Wavelength (nm) Re fra ctive Inde x Dispe rsio n (fs2 mm -1 ) Figure 3.7: Refractive Index (in blue) and dispersion (in red) of gallium phosphide as predicted by the Sellmeier equation and data in [33]. there is no dispersion in the cavity, that is, the roundtrip phase of the cavity is the same for all optical frequencies, this is relatively easy to achieve with careful control of the cavity length. However, if there exist dispersive elements within the cavity, = (!), and only those frequencies for which the resonance condition is satis ed will interfere constructively within the cavity, reducing the overall enhancement. Gallium phosphide, by virtue of the absorption edge at 2.25 eV (550 nm), exhibits strong dispersion near 1030 nm, as shown in gure 3.7. In this work, we compensate for this large material dispersion by using dispersive Gires- Tournois inerferometric (GTI) mirrors from Layertec corperation. GTI mir- rors tune the design of the dielectric stack so as to impart a controlled amount of group delay dispersion on the re ected optical pulse. It is there- fore necessary to establish the required crystal length prior to specifying the mirrors. Section 2.3 established that, for an optical pump of 1030 nm, gallium phosphide demonstrates a long coherence length of a few millime- tres. So as not to excessively limit the terahertz bandwidth, a thickness of 2.1 mm is used in this work. Because the crystal is to be mounted at Brew- 523.3. Cavity Design sterâ€™s angle, the actual physical thickness is 2:1 sin B = 2 mm. The amount of intracavity group velocity dispersion (GVD) for the 2.1 mm optical path length is then 2625 fs2. Possible GTI mirrors for dispersion compensation are listed in table 3.1 along with the pertinent speci cations. We see that, with the cavity con guration as shown in gure 3.6, it is possible to correct for the GVD of gallium phosphide to within 50 fs2. Layertec Description Power Re ectivity GVD part no. @ 1030 nm fs2/mm 103236 -75cm ROC GTI >99.85% -900 100 105200 Planar GTI >99.9% -800 100 102091 Planar GTI >99.9% <20 103143 5% IC 95 0.75% <20 103141 1.6%IC 98.4-0.3, 0.1% <20 102316 0.4%IC 99.6 0.1% <20 108251 0.8%IC 99.2-0.1, 0.4% <20 101147 0.5%IC 99.5 0.3% N/A Table 3.1: Part numbers and speci cations for Layertec mirrors used in the design of the passive enhancement cavity. Group velocity dispersion (GVD) is given for the dispersive mirrors over the range 1030 5 nm. Source: Layertec GmbH 3.3.2 Terahertz Output Coupling The refractive index of gallium phosphide at a frequency of 1 THz is 3.24 [33]. This means that the critical angle for the terahertz radiation at the gallium phosphide-air boundary upon exiting the crystal is 18.2 . Care- ful consideration is therefore needed in order to increase the e ciency with which the terahertz radiation is coupled out of the nonlinear crystal so as to avoid high losses due to total internal re ection of the terahertz wave at the crystal boundary. A proposed scheme for doing this is shown in gure 3.8. Geometrical considerations show that, following total internal re ection at the crystal face through which the optical beam exits, (henceforth the â€˜backâ€™ face) the terahertz radiation propagates at 90 to the incident optical beam. 533.3. Cavity Design Figure 3.8: Crystal geometry for output coupling of the terahertz radiation. The end of the gallium phosphide crystal is cut at the Brewster angle to ensure the terahertz radiation, internally re ected at the back crystal face, is normally incident on the cut surface. In this way, output coupling e ciencies are maximised. In order to maximise the output coupling e ciency of the terahertz radia- tion, the end of the gallium phosphide crystal was polished at the Brewster angle of 72.2 , such that the terahertz radiation is normally incident on the crystal boundary, as shown in gure 3.8. A further consequence of the high refractive index of gallium phosphide in the terahertz is that the radiation will strongly diverge upon exiting the crystal. In order to collimate this output beam, a silicon lens is used [43], together with an o -axis parabolic mirror. Ray tracing models performed by Jesse Petersen [44] have shown that a silicon hypohemisphere of diameter 7.0 mm and radius of curvature 4.0 mm will give aplanatic imaging and move the virtual focus back to 17 mm behind the lens, reducing the maximum THz divergence half-angle to about 7 . This is calculated for THz radiation emitted halfway through the gallium phosphide crystal along the optical path. A 25.4 mm focal length, 90 o -axis parabolic mirror is used to collect and collimate the radiation. Ray tracing has shown that an o -axis mirror of 25.4 mm diameter is su cient to collect the output radiation, and the compact size makes the physical arrangement within the cavity more 543.3. Cavity Design straightforward. The resulting ray traces are shown in gure 3.9 3.3.3 Cavity Performance Calculations In section 3.2 it was shown that, for a cavity exhibiting a net roundtrip dispersion, such that there is a frequency dependence on the net roundtrip phase delay, there is a reduction in the spectral bandwidth that is able to resonate in the cavity. This leads to a reduction in the intracavity power. In the following section, we study in detail the optical characteristics of the cavity design of gure 3.6, and include explicitly the dispersion of the intracavity gallium phosphide crystal in which the terahertz radiation is generated. The modelling work performed builds on that carried out by Jesse Pe- tersen [44], in which the enhancement cavity is seeded with an unchirped, Gaussian input pulse of the form Einc(t) = s 2 r 2 ln 2 exp 2 ln 2 t2 2 ; (3.25) where the temporal power full width at half maximum (FWHM) is , and oscillations at the carrier frequency are neglected. Einc(t) therefore describes the envelope of the incident pulse. Equation 3.25 normalises Einc(t) such that Z 1 1 jEinc(t)j 2dt = 1; (3.26) thus the average power can be described in terms of the pulse energy, Epulse by P (t) = EpulsejEinc(t)j2: (3.27) Taking the Fourier transform of this input pulse yields the spectrum ~Einc( ) = r 2 p ln 2 exp 2 2 8 ln 2 (3.28) where is the frequency o set from the central frequency of the pulse, = ! !c due to the neglecting of the optical carrier frequency in 3.25. 553.3. Cavity Design THz frequency (THz) Figure 3.9: Optical design for output coupling of terahertz radiation for a gallium phosphide crystal at Brewster incidence. The silicon lens is a hypohemisphere of diameter 7 mm and radius of curvature 4 mm. The parabolic mirror for collimating the collected radiation is JANOS optic part no. A8037-101, with a 1" diameter and 1" focal length (2" e ective focal length). Source [44]. 563.3. Cavity Design The e ect of dispersion is incorporated by rewriting equation 3.15 in the form (!) = k(!)d+ M (!) + X(!) = 0 + 0 0 + ( ) = 2m : (3.29) In this way, the constant part of the spectral phase shift accumulated from one cavity roundtrip is included in 0 0 = (!c) = k(!c)d+ M (!c) + X(!c): (3.30) Similarly, the part of the spectral phase that varies linearly with frequency is included in 0 0 0 0 = d vg + d M d! + d X d! : (3.31) Finally, terms of higher order in are grouped together in the term (!), (!) = 2 d2 d!2 + 3 d3 d!3 + : : : : (3.32) The e ects of GVD, TOD, and higher orders of dispersion are therefore all included in the nal term (!). Equation 3.17, where we now include dispersion, becomes !m = c d+ c 00 (2 m 0 + !c 0 0 (!m !c)); (3.33) any variation in the mode spacing of the cavity as a function of optical frequency is therefore attributed to (!). The cavity is assumed to be tuned to resonance such that, for the laser modes near the centre of the output spectrum (ie: where = 0), equation 3.23 is satis ed (The method used to control this is discussed in the following section). Therefore, for these frequencies, the mode o set, p is zero. In this case, these modes are perfectly coupled into the cavity and we expect maximum enhancement according to equation 3.9. Further, we assume that, close to line centre the cavity is dispersion compensated such that (!) = 0. 573.3. Cavity Design Away from the central optical frequency we calculate the circulating complex amplitude spectrum by de ning a complex cavity roundtrip transmissivity h( ) = RRic exp(i ( )) (3.34) and, following the methodology of equation 3.3, the intracavity circulating spectrum is calculated thus: ~Ecirc( ) = p 1 Ric ~Einc( )(1 + h( ) + h( ) 2 + :::) = p 1 Ric 1 h( ) ~Einc( ): (3.35) (!) is calculated from ts to dispersion data provided by the mirror man- ufacturer, and Sellmeier data for the nonlinear crystal in [33]. Substitution of equation 3.28 into 3.35 then allows the calculation of intracavity pulse energy and duration, and hence a measure of the enhancement in terahertz generation. Modelling Results The rst gure of merit we use to characterise the performance of the en- hancement cavity is the enhancement bandwidth. Fitting the dispersion and re ectivity data for the mirrors used from Layertec GmBH and applying equation 3.35 we see that optical enhancements of 3000 times are possible for optical frequency bandwidths of around 2 THz. Further, from the plot in gure 3.10 we see that, whilst increasing the re ectivity of the input coupler may move the cavity nesse closer to that of the optimal impedance-matched case suggested by equation 3.9, the increased photon lifetime within the cavity increases the e ect of small dispersion mismatches, leading to spec- tral narrowing and loss of optical bandwidth. Increasing the cavity nesse therefore reduces the bandwidth over which dispersion can be e ectively compensated. This is undesirable in our case as the THz bandwidth relies on a large optical bandwidth within which to generate di erence frequency components. We therefore expect a trade-o between greater ampli cation 583.3. Cavity Design - 3 - 2 - 1 0 1 2 3 0 500 1000 1500 2000 2500 3000 3500 Î© THz Enh ancem ent Figure 3.10: E ective enhancement of the range of optical frequencies ef- fectively coupled into the enhancement cavity for various input couplers. The total material dispersion is 2620 fs2 calculated for 2.14 mm of gallium phosphide. This is compensated for using mirrors from Layertec GmBH of two of part number 103236 and one of 105200, giving a total group velocity delay of 2600 fs2 at the centre wavelength of 1030 nm. The net roundtrip GDD pro le for the cavity is plotted in gure 3.11 of fewer optical modes, as in the case of a high nesse cavity, or reduced ampli cation of a broader bandwidth, achieved by spoiling the nesse with a lower re ectivity input coupler. In order to determine the optimal cavity con guration, the circulating spectrum is Fourier transformed into the time domain and tted to Gaussian pulse in order to determine the resulting pulse duration. The resulting pulse broadening for the mirrors speci ed in gure 3.6 is shown for a 200 fs input pulse in gure 3.12. It is seen that the circulating pulse has approximately twice the duration of the input pulse, owing to the residual GDD of the cavity components, plotted in gure 3.11 as a function of optical frequency . In order to determine the optimum input coupler, a compromise will be made between ampli cation of a given frequency, and the total optical bandwidth. In order to study this, we refer to gure 3.13. It is seen that the pulse broadening remains modest, that is, below 20% for the 200 fs input 593.3. Cavity Design -3 -2 -1 0 1 2 3-100 0 100 200 300 W HTHzL Ne tCavit y GD D Hfs 2 L Figure 3.11: Calculated residual GDD of the cavity components for the cavity con guration shown in gure 3.6, and with an optical path length in the gallium phosphide crystal of 2.14 mm. Small negative phase delays are seen for < 0, where the total GDD is close to zero. As increases above 0 a growing positive delay is calculated. - 1000 - 500 0 500 10000.0 0.2 0.4 0.6 0.8 1.0 Re lat ive Inten sity Time (fs) Input pulse Fit to circulating pulse Circulating pulse Figure 3.12: Estimated broadening of a 200 fs input optical input pulse within a passive enhancement cavity of the same design of that used in gure 3.10. The circulating pulse is tted to a Gaussian pulse shape of full width half maximum 350 fs, shown by the dashed red line. 603.4. Cavity Locking 0.990 0.992 0.994 0.996 0.998 1.000 0 500 1000 1500 2000 2500 200 250 300 350 Pul se Ener gy En ha nc em ent Cir cul atin g Pul se Dur atio n fs Pulse energy enhancement Circulating pulse duration () Figure 3.13: Circulating pulse energy enhancement and duration as a func- tion of input coupler re ectivity for the cavity con guration of 3.12. pulse, for input couplers of up to 99.8% re ectivity. Whilst this will come with a trade-o in enhancement, the lower cavity nesse will also increase the cavity linewidth, (indeed, it is exactly this broadening of the cavity linewidth that increases the optical bandwidth within the cavity) which, in turn, will lessen the demands on locking the cavity mode spacing with that of the laser. An estimate of the ability of the feedback loop to lock a narrow en- hancement cavity resonance to a comb element of the mode-locked laser can only be obtained with a detailed model of the electronic components used. Rather than model this numerically, we therefore choose to optimise the input coupling of the cavity experimentally. 3.4 Cavity Locking The need for precise control of the longitudinal mode spacing of the passive enhancement cavity has been discussed previously. It was shown that the the mode spacing of both the laser and enhancement cavity are determined by 613.4. Cavity Locking the respective o set frequencies (!0 for the laser) and repetition rates (!rep for the laser). Further, due to the dependence of the frequency of the laser modes on the repetition time, as per equation 2.6, the laser frequencies will exhibit instabilities due to environmental factors such as temperature uctu- ations and acoustic vibrations. It is therefore necessary to devise a method of stabilising the cavity modes with respect to the laser modes, such that the maximum amount of laser power is resonant in the cavity at all times. In order to ensure complete mode overlap, one must have independent con- trol of both the relative o set frequencies, and the relative repetition rates. Whilst experiments achieving this have been demonstrated [28], it requires two high-speed servo loops and considerable amount of experimental com- plexity. Instead, we lock the average frequency of the enhancement cavity to that of the laser. In this way, we ensure the maximum possible power in the cavity for a given o set frequency mismatch and cavity dispersion, as shown in the projections along wavelength of the resonance map picture of gure 3.5. We make use of a locking scheme developed by Hall et al. [45], originally developed for stabilizing the frequency of a laser by locking it to a passive reference cavity. In our adaptation of the technique, the length of the passive enhancement cavity is modulated at â€™ 800 kHz, hitherto referred to as the dither frequency, by driving a piezzo-electric actuator (PZT) onto which one of the cavity mirrors is xed, as shown in gure 3.6. This frequency is chosen to be high enough so as to exceed any acoustic vibrations, but low enough such that the PZT still has some small response. The signal re ected by the enhancement cavity is spectrally resolved and a portion of the optical bandwidth focussed onto a fast photodiode. The bandwidth and center wavelength of the measured signal were tuned to give the maximum intracavity power. The measured signal is compared to a sample of the electronic signal used to drive the intracavity PZT, henceforth the local oscillator (LO) signal. For any spectral component of the measured optical signal, when the cavity is tuned to the exact resonance of equation 3.8, one observes a minimum in the re ection signal measured by the photodiode. Any deviation from this resonance condition will produce an increase in the 623.4. Cavity Locking Figure 3.14: The re ection signal measured from the enhancement cavity near a position of resonance. The phase of the re ection signal with respect to a modulation in cavity length provides the direction in which one must adjust the cavity length in order to restore the resonance condition. eld re ected by the cavity at this particular wavelength. The minimum re ected power occurs when the greatest amount of power is supported in the cavity. The locking scheme discussed here provides the necessary feedback in order to determine which direction to tune the cavity length so as to restore the resonance condition of equation 3.8. By modulating the cavity, the required sign of cavity length change is obtained by determining whether the re ected power measures in phase or 180 out of phase with the dither signal, as shown in gure 3.14. The experimental arrangement is shown in gure 3.6. To understand the locking process in more detail, we write the electric eld incident on the cavity as a monochromatic eld of the form of equation 2.1 Einc(t) = E0e i!t + c.c. (3.36) 633.4. Cavity Locking Whilst in the original technique, demonstrated by Hall et al. the length of the laser cavity (and therefore the phase of the input eld) is modulated at the dither frequency, we have found modulation of the length of the enhancement cavity to give analogous results. Further, for the purposes of this discussion, we consider the modulation in frequency arising due to the translating mirror rather than the phase change directly. When modulated at a dither frequency dit with a modulation depth B the modulated input eld is given by Einc(t) = E0e i(!t+B sin dit t) â€™ E0(1 + iB sin t)e i!t = E0 1 + B 2 (ei dit t e i dit t) e i!t = E0e i!t + E0B 2 ei(!+ dit )t e i(!+ dit )t (3.37) We therefore have the familiar result that a monochromatic eld at opti- cal frequency !, modulated at a frequency dit actually consists of three elds. One at the original optical frequency !, and two optical sidebands at frequencies ! dit . The eld re ected by the cavity is given by Eref = ET p Rh Einc = Einc p R e i 1 1 p RRice i ! = EincF (!); (3.38) where F (!) is the cavity re ection coe cient. The remaining constants were de ned in section 3.1, from which we recall ET is the total eld resonant within the cavity. The frequency dependence arises due to the fact that the roundtrip phase, = d!c + M , where M is the phase change due to re ection at the cavity mirrors. Combining equations 3.37 and 3.38, we 643.4. Cavity Locking obtain Eref = E0F (!)e i!t + E0B 2 F (! + dit)e i(!+ dit )t E0B 2 F (! dit)e i(! dit )t: (3.39) The power measured by the photodetector due to this re ected eld is given by Pref = jEref j 2 = E20 jF (!)j 2 + E20B 2 2 jF (! + dit)j 2 + E20B 2 2 jF (! dit)j 2 + E20B 2 F (!)F (! + dit)e i dit t + F (!)F (! + dit)e i dit t F (!)F (! dit)e i dit t F (!)F (! dit)e i dit t + terms in 2 dit t: (3.40) Providing the modulation depth is su ciently small that B2 E0 we may neglect the terms in 2 dit t on the grounds they are small in amplitude. Further, the use of a rf mixer allows the rejection of the non time-varying dc terms such that only the terms having a frequency equal to that of the local oscillator (LO) signal are measured. The measured dc signal output at the mixer can then be rewritten S = E20B [Re fF (!)F (! + dit)g cos dit t+ Im fF (!)F (! + dit)g sin dit t Re fF (!)F (! dit)g cos dit t Im fF (!)F (! dit)g sin dit t] : (3.41) We set the phase of the LO signal with respect to the eld modulation such that the imaginary term is measured. In this case, the detected error signal is S = E20B [Im fF (!)F (! + dit) F (!)F (! dit)g] : (3.42) The resulting error signal is calculated and plotted in gure 3.15. 653.4. Cavity Locking -2000 -1000 0 1000 2000 -1.0 -0.5 0.0 0.5 1.0 Laser Frequency Detuning from Resonance HkHzL Amplitud e Figure 3.15: Calculated error signal for a passive enhancement cavity having a nesse of F = 2 105 as a function of laser frequency detuning from the exact resonance condition of equation 3.8. In this example, the dither frequency, dit is 800 kHz and the optical wavelength is 1030 nm. 663.4. Cavity Locking If the nesse of the cavity is su ciently high, and the dither frequency su ciently large such that the sidebands fall at frequencies for which F (! dit) â€™ 1 then only the eld at the optical frequency is coupled into the cavity. The power measured at the photodetector, as per equation 3.40, becomes â€™ Pref â€™ E 2 0 jF (!)j2 + B2 2 + BE0 2 (2iF (!) sin dit t 2iF (!) sin dit t) = E20 jF (!)j2 + B2 2 2BE0 Im fF (!) sin dit tg : (3.43) Again, the dc term is rejected by the rf mixer and we measure S / sin dit t. On resonance, the phase of the eld in the cavity at the input coupler is = 2m where m is some integer. If the cavity is shifted away from resonance by some fraction, l, of the optical wavelength, we obtain â€™ 2m + 4 L : (3.44) We can then re-write the re ection coe cient of the cavity as F ( L) â€™ p R 1 p RRic 4 i L â€™ i 4F L; (3.45) such that S â€™ 8BE0 F L (3.46) Under these conditions, S therefore provides a control signal that varies linearly with the detuning of the cavity from resonance. This control signal is used in a servo loop to control the piezzo-mounted mirror, as shown in gure 3.6. In this way, the average resonance frequency of the enhancement cavity tracks that of the mode-locked laser with which it is seeded, ensuring that the maximum possible power resonates inside the cavity. 673.5. Spatial Mode-matching into the Optical Resonator w(z) = w0 s 1 + z z0 2 Beam radius at position z R(z) = z 1 + z0 z 2 Beam wavefront radius of curvature at position z (z) = tan 1 z z0 Guoy phase-shift w0 = r z0 Beam waist 2z0 = 2 w20 Depth of focus Table 3.2: Properties of a Gaussian beam 3.5 Spatial Mode-matching into the Optical Resonator 3.5.1 Gaussian Mode of the Optical Resonator It is well known (see, for example, [46]) that the electric eld circulating in an optical resonator has the solution of a Gaussian beam E0(~r) = A w0 w(z) exp x2 + y2 w2(z) exp ik x2 + y2 2R(z) exp[ikz] exp[i (z)]; (3.47) where A is the amplitude of the electric eld in the polarisation direction, w(z) is the beam waist at position z. This leads to the de nition of the Gaussian beam paramters in table 3.2. It is common to further simplify equation 3.47 by de ning the q-parameter, q(z) of the Gaussian beam 1 q(z) = 1 R(z) + i w2(z) : (3.48) 683.5. Spatial Mode-matching into the Optical Resonator It is this q-parameter, having units of metres, that we may propagate through a system of optical components using the commonly used ABCD matrix transfer method, in which the e ect of each optical component is de- scribed by its speci c ABCD matrix, (see, for example, [46]) and the ABCD matrix for the entire optical system, in this case, the enhancement cavity, is given by the matrix product of the ABCD matrices of its components. For a given system of optical components, the q-parameter transforms as q2 = Aq1 +B Cq1 +D ; (3.49) where the values of A, B, C, and D are determined by the speci c optical components in the cavity. By requiring that the cavity is stable, that is, that the Gaussian beam parameters of the eld contained within the cavity are the same for N round trips as for N+1 round trips, we require that qf = qi. Further, we require that the determinant of the cavity ABCD matrix be unity. Equation 3.49 therefore gives rise to the familiar resonator stability condition, 1 2 jA+Dj 1: (3.50) It is this condition that determines the appropriate mirror radius of curva- ture for a given cavity length. Having established the conditions for a stable resonator, the parameters in table 3.2 can be used to determine the position of the intracavity focal planes, as well as the beam spot sizes at these focal points. Further, the beam q-parameter can be calculated for any point within the enhancement cavity, in particular, we seek to evaluate the q-parameter at the cavity side of the input coupler, such that the q-parameter of the input beam may be manipulated to match it, therefore maximizing the power from the input beam that is coupled into the Gaussian mode of the enhancement cavity. This will be the subject of section 3.5.2. 693.5. Spatial Mode-matching into the Optical Resonator 3.5.2 Beam Pro ling In order to determine the appropriate mode-matching solution for the en- hancement cavity work, and the optimum focussing arrangement for the single pass terahertz generation and electro-optic sampling experiments, it was rst necessary to determine the Gaussian beam parameters of the eld input to the experiment. As the same laser output was used for the case of the enhancement cavity, and for the terahertz generation in the dual-comb spectrometer work, it is su cient to determine the beam parameters at the output of the compression stage, from which appropriate solutions can be found for each case. An additional set of beam parameters were determined for the laser used as the local oscillator in the dual-comb spectrometer ex- periments. In order to determine the Gaussian beam parameters, a scanning knife-edge technique [47] was used to determine the beam diameter at sev- eral points through the focal region following a lens of known focal length. The clip levels required to block 10% and 90% of the beam power at various points along the optical axis are measured. Following the analysis in [47] for a TEM00 Gaussian beam, the beam waist is then calculated according to 2w0 = 4Dc 2:563 : (3.51) Figure 3.16 shows the result of such a knife edge measurement, where a 250 mm converging lens is used to focus the beam down to a spot. We de ne the plane in which this lens is positioned as the compressor output plane. It is seen that the beam waist produced at the focus of this lens is 120 m , and that the Rayleigh range, that is, the range at which the on-axis intensity falls to half that at the focus, is found to be 20 mm. The half-angle divergence of the beam is found to be 4.5 mrads. We note that this gives an M2 value, the ratio of the waist-diameter to divergence product to that of the ideal Gaussian beam, of 1.7, which is a typical value for a bre laser. Applying the relations in table 3.2 and nding the q-parameter at the focus to be purely imaginary, 1 q(focus) = i w20 , we then propagate this q parameter back through to the entry face of the lens using equation 3.48 to 703.5. Spatial Mode-matching into the Optical Resonator 50 100 150 200 2500 100 200 300 400 500 600 Distance from Lens [mm] Beam Radius [Âµm ] Figure 3.16: Beam waist pro le following a 250 mm focal length converging lens, as measured using the knife-edge beam-clipping technique as described in [47]. nd q(compressor output) = 0:17 0:04i metres: (3.52) This is used as a baseline q-parameter for calculating the required focus- ing optics, both for mode-matching into the enhancement cavity, and for creating the optimum spot size in the gallium phosphide waveguide for the dual-comb spectrometer work. 3.5.3 Mode-matching Solution In the case of the cavity depicted in gure 3.6, the required q-parameter at the input coupler is found by rst determining the q-parameter at either of the curved mirrors, by recognising that the wavefront radius of curvature must match that of the mirrors, and that the focal points must be halfway between the two curved mirrors in each case. The same method as before is 713.6. Characterisation of the Passive Enhancement Cavity - Results and Analysis then used to propagate this q-parameter to the input coupler. One obtains q(cavity input) = 0:52 + 0:59i: (3.53) The task of mode-matching therefore becomes to nd some system of optics, having transfer matrix A B C D such that q(cavity input) = Aq(compressor output) +B Cq(compressor output) +D : (3.54) There are, of course, an in nite number of solutions to this problem given lack of restraint on components available and space, although it should be noted that it is desirable to minimise transmission losses by reducing the number of components as much as possible. For the case here, a pair of lenses of focal lengths 1050 mm and 750 mm were positioned 2.18 and 2.04 metres before the cavity input coupler respectively. The total optical path length between the compressor output plane and input coupler is set to 2.5 metres. Both the lenses used were anti-re ection coated (Newport coating reference AR.16) at 1030 nm to minimise re ection losses. A schematic of this arrangement is shown in gure 3.17. 3.6 Characterisation of the Passive Enhancement Cavity - Results and Analysis It is clear that enhancement of the optical pulse within the cavity relies on the precise control of a number of factors, namely the cavity nesse, the roundtrip dispersion of the cavity, the mode-matching of the input beam into the stable Gaussian mode of the resonator, and the satisfaction of the resonance condition, equation 3.9. In order to determine satisfactory per- formance of the measures taken to control these factors, it is desirable to be able to measure them independently. In accordance with equation 3.13, the enhancement factor of the cavity is de ned as Enhancement = Pcav Pin (3.55) 723.7. Measurement of the Cavity Finesse Figure 3.17: Schematic of optics used for mode-matching into the passive enhancement cavity. Some cavity mirrors omitted for clarity. where Pcav is the optical power circulating within the cavity, and Pin is the power incident on the input face of the input coupler. It is possible to measure the former using the optical power transmitted by any one of the cavity mirrors due to the non-unity re ectivity, and measurements of the transmissivity of the mirror performed without the cavity. Both the cavity measurement and mirror calibration is performed with a conventional optical power meter, which provides only a measure of the optical power integrated over the entire optical bandwidth, and lacks sensitivity to small changes in the mirror re ectivity. The following section discusses a number of methods used to analyse the cavity in more detail. 3.7 Measurement of the Cavity Finesse The expected enhancement factor of the cavity depends critically on the cavity nesse, as derived in equation 3.6. The nesse itself is determined by the re ectivity of the cavity mirrors. An accurate measurement of the 733.7. Measurement of the Cavity Finesse nesse, however, is di cult to attain by simply measuring the mirror re- ectivities individually and calculating, as per equation 3.6, as the nesse, and therefore the enhancement factor, changes dramatically with only small changes in mirror re ectivity. Instead, we use a method outlined in [48] to measure the nesse directly. The length of the enhancement cavity is rapidly swept using the intracav- ity piezoelectric actuator, such that the cavity is e ectively swept through a resonance, allowing light into the cavity, and then rapidly o resonance, preventing further input of light. As the cavity is swept, the transmission through one of the cavity mirrors is monitored with a fast photodiode, where one observes the gradual ringing down of the pulse as it circulates within the cavity. Consider further the analysis of [48], in which the cavity length is increased by translating the intracavity mirror at velocity v, such that the length at time t follows L(t) = L0 + vt: (3.56) Following the same methodology of equation 3.3, the eld transmitted by the cavity as a function of time is the in nite sum of the partial elds En(t), where En(t) has made (n 1) roundtrips of the cavity. The total eld transmitted is therefore given by Eout(t) = Tout 1X n=1 En(t) = Tout 1X n=1 EinTR n 1ei n(t); (3.57) where n(t) is the phase shift of the partial eld En(t), and Tout is the eld transmissivity of the cavity mirror through which the leak is measured. Accounting for the Doppler shift in frequency of 1 2v c at each re ection 743.7. Measurement of the Cavity Finesse at the moving mirror, n(t) can be written n(t) = 1(t) + ! c 1 v c (2d0 + vt) + ! c 1 v c 2 (2d0 + 3vt) + ! c 1 v c 3 (2d0 + 5vt) : : : + ! c 1 v c n 1 (2d0 + (2n 3)vt): (3.58) Under the assumptions that v c, and vt L0 such that we can neglect terms in v c 2 , we can rewrite this expression as n(t) = (2n 1) 1(t) n(n 1) v FSR ; (3.59) where 1(t) = 2 L(t) , v = 2v , and FSR = c L0 is the cavity free spectral range. For a high nesse cavity, the photon lifetime, = F 2 FSR becomes large, such that, on average, each pulse completes many roundtrips of the cavity, and the discrete sum of equation 3.57 can be closely approximated by the continuous integral. Taking the square of the nal result, so as to nd the measured quantity of the intensity, yields Iout(t) / e t j erfc[ (t)]j2; (3.60) where erfc[ (t)] is the complimentary error function, and (t) is a complex function of the cavity nesse, free spectral range, and sweep parameters [48]: (t) = 1 i 2 p 2 FSR v 1 2 F 1 + i 2 p 2 " FSR v 1 2 2 1(t) FSR v 1 2 # : (3.61) The decay of the swept cavity is therefore proportional to the decay of the static cavity, multiplied by the j erfc[ (t)]j2 term, which manifests itself as a 753.7. Measurement of the Cavity Finesse ringing e ect as the cavity is swept through the resonance point. By tting the decay of the cavity, one can therefore determine the photon lifetime, , from which it is a trivial matter to determine the nesse. The rst and second maxima of equation 3.60 occur with a temporal separation t = p 5 1 p 2 FSR : (3.62) In order that the maxima are spaced such that a reliable t to the cavity decay can be obtained, it is desirable that this time is less than the decay time , which requires the cavity mirror be swept with a velocity, v, that satis es v 15c F2 L0 : (3.63) 3.7.1 Results of Finesse Measurements The cavity was con gured as in gure 3.6, with a 0.5% input coupler, and without the nonlinear crystal. It should be noted that, as the cavity is de- signed to be dispersion compensated with the nonlinear crystal, the â€˜emptyâ€™ cavity has a total average group delay dispersion of -2650 fs2. Therefore, in order to avoid distortion of the ring down data due to di erent wavelengths arriving with a di erent roundtrip phase-shift, the measurement was per- formed with a continuous wave laser at 1064 nm. Figure 3.18 shows the result of measuring the nesse via such a â€˜ringdownâ€™ method as described above. For the cavity con guration shown, one expects a nesse of 970, as predicted by equation 3.6. The exponential t gives a nesse of 930, in good agreement with the predicted value, demonstrating that, for the seed wavelength of 1064 nm, the mirror re ectivities are as speci ed. It should be noted that the rst maxima shown in gure 3.18 is excluded from the exponential t, as it typically displays a level higher than that predicted. This is likely due to imperfect mode-matching of the input beam, meaning that some of the optical power escapes the cavity in the 1 s taken for the swept cavity to hit the second resonance. For comparison, the results of a similar measurement where the input coupler has been replaced with one 763.8. Enhancement Bandwidth of the Cavity âˆ’5 0 5 10 0 0.5 1 Time (Âµs) Relativ eI nte ns ity Experimental Data Exponential Fit Figure 3.18: Ring down measurement for the passive enhancement cavity without the gallium phosphide crystal. An exponential t is shown for a photon lifetime of 1.4 s, or a nesse of 935. This corresponds well with the expected cavity nesse of 960 for the six mirror cavity with each mirror having R = 0:9997 and Ric = 0:995, or a transmissivity of 0.5%. For this data, the cavity was swept with mirror velocity of 3 mm/s, in accordance with equation 3.63. for which Ric = 0:998 or transmissivity of 0.2%, is given in gure 3.19. In this case, the predicted nesse is 1800, in good agreement with the 1730 measured. 3.8 Enhancement Bandwidth of the Cavity For the cavity nesse indicated in gure 3.18, one expects an optical build- up factor of 470, as predicted by equation 3.13. It was possible to obtain an intracavity intensity of 290 mW for an input power of 1.2 mW, indicating 773.8. Enhancement Bandwidth of the Cavity âˆ’5 0 5 10 0 0.5 1 Time (Âµs) Relativ eI nte ns ity Experimental Data Exponential Fit Figure 3.19: Plot of data as for gure 3.18 where the input coupler has been changed to a transmissivity of 0.2%, giving a predicted nesse of 1800. The exponential tted is for a photon lifetime of 2.7 s, giving a measured nesse of 1730. 783.8. Enhancement Bandwidth of the Cavity an ampli cation of 240, about half of the theoretical maximum predicted by equation 3.6 with the speci ed mirror re ectivities. This shortcoming in performance is attributed to imperfect mode-matching of the input beam. Indeed, only a single lens was used to give an approximate solution to the mode-matching problem for this CW test laser. Due to the discussions set forth in section 3.2, it is not reasonable to expect similar levels of enhance- ment for the broadband optical pulse of a mode-locked oscillator for this cavity con guration, due to the large net dispersion of the cavity in the absence of the nonlinear crystal. However, placing the nonlinear crystal at the cavity focus as in gure 3.6, and seeding the cavity with the broadband optical pulses described in section 2.4 resulted in minimal build-up, in spite of the predicted net cavity dispersion being close to zero. In order to study the e ect of dispersion on the cavity enhancement fac- tor for the broadband mode-locked oscillator, and to verify that the GTI mirrors had su cient group delay to compensate for that due to the mate- rial dispersion of the nonlinear gallium phosphide crystal, the crystal was removed, and SF11 glass substrates of various thickness were placed at the Brewsterâ€™s angle of 60 within the cavity. The total dispersion of the en- hancement cavity was then varied according to the optical path length in SF11 glass per roundtrip of the cavity. As was shown in section 3.10, for a cavity with non-zero roundtrip group delay dispersion, the optimum build- up of a broadband pulse is achieved for a cavity con guration that is slightly â€˜overcoupledâ€™. That is, that the transmissivity of the input coupler is greater than that suggested by the impedance-matching condition of equation 3.9 alone. For the present work, the transmissivity of the input coupler was increased to 1%, thereby allowing su cient enhancement of the eld within the cavity such that a leak through one of the high re ectivity cavity mirrors could be used to measure the stored spectral bandwidth within the cavity, and estimate the intracavity build-up by measurement of the power trans- mitted through the cavity mirror. In this case, the nesse of the cavity is reduced to 500, and the predicted optimum enhancement is 300. The amount of SF11 used and the resulting nominal cavity dispersion based on the Sellmeier coe cients given in [49] are given in table 3.3. Note 793.8. Enhancement Bandwidth of the Cavity that the 17.7 mm thick piece of SF11 glass was chosen speci cally to have the same group delay as 2.1 mm of gallium phosphide at Brewster incidence. In order to illustrate the e ect of the cavity dispersion on the obtained enhancement factor, we include in table 3.3 the 10 dB bandwidth of the spec- tra resonant inside the cavity upon locking of the cavity average frequency to that of the mode-locked laser used to seed it. It is seen that higher en- hancement factors are produced for greater resonant spectral bandwidths, as predicted by section 3.2. Close to zero nominal cavity group delay disper- sion, the spectrum stored in the enhancement cavity is close in bandwidth to that of the input spectrum. As the cavity dispersion is increased, either positively or negatively, the dependence of roundtrip phase delay on wave- length leads to spectral narrowing, and a loss of intracavity power. In gure 3.20, we plot the measured enhancement against the estimated net group delay of the cavity, along with the measured 10 dB resonant spectral band- width. For comparison, the enhancement and 10 dB bandwidths predicted by the resonance map calculations of section 3.2 are included. It is seen that the measured enhancement decreases with increasing net cavity group delay in a manner similar to that calculated. That the measured values fall below those calculated for non-zero cavity group delays may be attributed to the neglecting of third and higher order dispersion in the calculations. Figure 3.20 also shows the strong correlation between the resonant spectral bandwidth and the enhancement, the measured resonant spectral bandwidth being in excellent agreement to the calculated values. 803.8 . Enhanceme nt Bandwidt h of th e C avi ty Table 3.3: Optical enhancement, resonant bandwidths, and net cavity dispersions for various cavity con gurations used to verify the e ect of dispersion on the available enhancement. The enhancement and bandwidth achieved with the â€˜emptyâ€™ cavity (where the beam propagates only in air) are also given for reference. The seed spectrum incident on the cavity has a 10 dB bandwidth of 13.8 nm. Note that the theoretical maximum enhancement for this cavity con guration, with the 1% input coupler, is â€™ 300. The absorption of SF11 glass at 1030 nm [49], combined with slight errors in mode-matching, are likely responsible for the shortfall in performance, even for cavity con gurations having minimal roundtrip dispersion. Thickness of Optical Path Length Net. Cavity GDD Measured 10 dB Bandwidth SF11 Glass (mm) in SF11 at at 1030 nm (fs2) Enhancement Factor of Resonant Brewsterâ€™s Angle (mm) Factor Spectrum (nm) 0 0.00 -2625.00 10.5 4.50 14.0 16.05 -550.70 18.9 6.32 16.8 19.33 -126.92 26.5 10.4 17.7 20.32 0.96 45.5 13.6 31.6 36.38 2075.26 9.5 5.63 Seed - - - 13.8 813.8. Enhancement Bandwidth of the Cavity Figure 3.21 compares the optical spectrum supported within the en- hancement cavity to the input spectrum, where now the gallium phosphide crystal is placed at the intracavity focus. Minimal spectral narrowing is seen when the dispersion of the gallium phosphide contributes to the net group delay dispersion of the cavity. Indeed, in this case, the 10 dB bandwidth of the supported spectrum is 13.2 nm, as opposed to the 13.8 nm of the input spectrum. The lack of spectral narrowing observed in the case where the nonlinear crystal is placed at the cavity focus suggests that the dispersive GTI mirrors are su ciently compensating for the material dispersion of the crystal, as was the intent. However, for this cavity con guration, the optical power enhancement was reduced to around a factor of 6. In the absence of sig- ni cant spectral narrowing, this is attributed to a reduced cavity nesse. Indeed, attempts to measure the nesse via the method outlined in section 3.7 proved unsuccessful, due to the di culty in driving the cavity mirror with a velocity that satis es equation 3.63 for this reduced nesse. Recall the conclusions of section 3.2 and, in particular, gure 3.10, where it was seen that, for given non-zero net cavity dispersion, increasing the cav- ity nesse results in a narrower range of optical bandwidths being supported in the cavity. Indeed, the resonance map calculations for a cavity showing a 7% roundtrip loss produce a 10 dB resonant spectral bandwidth that varies with cavity group delay as in the black line of gure 3.20, which shows the reduced sensitivity of the resonant bandwidth to cavity group delay under such conditions. The observation of minimal spectral narrowing for the case of a cavity of such low nesse as gure 3.21 should therefore not be taken as proof that the cavity is su ciently dispersion-compensated to expect sig- ni cant ampli cation of large optical bandwidths at higher nesses. Whilst methods of precisely determining the cavity dispersion have been demon- strated [41, 50, 51], these require the cavity to have low roundtrip loss or, equivalently, high nesse, in order to make an accurate measurement of the dispersion. It is therefore not practical to make a measurement of the resid- ual dispersion of the enhancement cavity with the nonlinear crystal in place. We therefore devote the remainder of this section to a discussion of e orts to 823.8. Enhancement Bandwidth of the Cavity 0.2 0.4 0.6 0.8 1 Enhancem en tRelativ et oPea k âˆ’2500 âˆ’2000 âˆ’1500 âˆ’1000 âˆ’500 0 500 1000 1500 2000 25002 4 6 8 10 12 14 16 Net Cavity Group Delay (fs2) 10d BRes onan tBand widt h(nm ) Figure 3.20: Top: Power enhancement of 7 nm full width at half maximum (13.8 nm 10 dB) bandwidth optical eld as a function of net group delay dispersion. Blue line: As predicted by resonance map calculations for a passive enhancement cavity having a nesse of 960. Points: Measured en- hancements for the cavity con gurations of table 3.3. Note that all data is scaled relative to the peak enhancement at zero GDD for each respective data set to account for shortfalls in enhancement due to mode-matching of the seed optical eld. Further shortfalls of the measured values with respect to the predicted value at non zero group delays are likely due to the omission of higher order dispersive delays from the simulation. Bottom: Red: Mea- sured and predicted 10 dB resonant spectral bandwidths for a cavity having a nesse of 960. Black: Predicted 10 dB resonant spectral bandwidth for a cavity having a nesse of 100. 833.9. Linear Optical Loss of Gallium Phosphide 1010 1015 1020 1025 1030 1035 1040 1045 1050âˆ’35 âˆ’30 âˆ’25 âˆ’20 âˆ’15 âˆ’10 âˆ’5 0 Wavelength (nm) Relativ eLev el Belo w Pea k(dBm ) Input Cavity bandwidth with GaP Figure 3.21: Optical spectra supported by the passive enhancement cavity with the gallium phosphide crystal placed at the intracavity focus, compared to the input spectrum. Note that the former is displaced downwards by 10 dB for clarity. determine the source of the reduced cavity nesse, due to loss mechanisms within the nonlinear crystal. 3.9 Linear Optical Loss of Gallium Phosphide The previous section demonstrated that, in the absence of the nonlinear crystal, the enhancement cavity has both high nesse and large dispersion. In installing the nonlinear crystal, the nesse of the cavity is signi cantly reduced, resulting in minimal enhancement of the optical pulses. That it is the nesse of the cavity limiting the build up in this case, and not the disper- sion, can be seen in gure 3.21, which shows minimal bandwidth narrowing of the broadband optical pulses used to seed the enhancement cavity. Having established that, with the nonlinear crystal installed, the disper- sion of the cavity is su ciently minimised such that the cavity is providing 843.9. Linear Optical Loss of Gallium Phosphide equal enhancement of the entire optical bandwidth and, further, that the cavity exhibits high nesse in the absence of the nonlinear crystal, oneâ€™s attention turns to the nonlinear crystal itself. Upon locking the enhance- ment cavity with the gallium phosphide placed at the focus, it was observed that the control loop would continuously increase the voltage at the piezo- electric actuator, thereby e ectively reducing the length of the cavity. This behaviour is indicative as to why the gallium phosphide reduced the nesse and enhancement of the cavity. The gallium phosphide exhibits mid-gap, linear absorption of the 1030 nm light inside the cavity, thereby reducing the nesse as per equation 3.11. This absorption leads to thermal expansion of the crystal, which in turn was responsible for the feedback loop applying a continuous ramp to the PZT as the control loop works to keep the cavity on- resonance. In the following section, we determine the voltage ramp applied by the servo to the piezoelectric actuator, and make certain approximations that allow the absorption coe cient, , of the crystal to be estimated. As was the case for section 3.8, in order to obtain su cient optical power in the enhancement cavity such that a leak through one of the cavity high re ectivity mirrors could be measured, the transmission of the input coupler was further increased to 5% (Layertec Part no. 103143). Otherwise, mirrors were as shown in gure 3.6. It is useful to note that, in the absence of any absorption from the nonlinear crystal, the cavity would have a modest nesse of 120. The crystal was placed slightly away from the cavity focus at a point where the beam waist was measured using the technique described in section 3.5.2 as 400 m . When the cavity was locked, the power measured at a leak through one of the high re ectivity mirrors was 400 W. With the speci ed mirror re ectivity of 99.97%, found to be in good agreement with a single pass measurement of the transmission, this implies an intracavity power of 1.3W. A plot of the voltage ramp at a monitor terminal at the servo output is shown in gure 3.22. This monitor port gives 1 20 VPZT , where VPZT is the applied PZT bias. The data shows a clear asymptotic increase in the voltage required to maintain the resonance condition as a function of time, as one expects for the case where the crystal undergoes thermal expansion due to 853.9. Linear Optical Loss of Gallium Phosphide âˆ’2 âˆ’1 0 1 2 3 4 5 61 2 3 4 5 6 7 time (s) PZ TBias/2 0(V ) PZT Voltage Bias/20 V(t) = Vf + (Vi âˆ’ Vf)e âˆ’bt Linear fit âˆ’0 0.17 0.34 0.51 0.68 Expansio no fGa P( m) Î¼ Figure 3.22: Variation of the voltage bias applied to the piezoelectric ac- tuator (PZT) as a function of time following locking of the enhancement cavity. The data is shifted such that the cavity is locked at time = 0 s. An exponential t is shown with Vf = 10:5 V, Vi = 3 V, and b = 0:1155 s 1, as well as a linear t to the data at early times with V (t) = 0:8335t+ 2:995 V. absorption of optical power. The PZT used is model number HPSt 150/6 2-2 from Piezomechanik GmbH. It has a full de ection of 2 m at a maximum 150 V applied bias. Assuming that the length change as a function of applied voltage is linear, using the previously calculated value for the refractive index of gallium phosphide at 1030 nm of 3.11, and using the thermal properties of gallium phosphide listed in table 3.4, one calculates a power absorption of 30 mW, or about 3% given the intracavity power calculated earlier. We note that this is very much a lower limit on the absorption, as all modes of heat transfer out of the interaction region are neglected. For ex- ample, thermal conduction out of the area of the beam waist will mean that an area larger than the beam is heated, which would increase the absorption 863.9. Linear Optical Loss of Gallium Phosphide Table 3.4: Thermal properties of gallium phosphide, according to [52]. Property Value at 300K Units Coe cient of linear thermal expansion 4.65 10-6 C 1 Speci c heat capacity 0.43 Jg 1 C 1 Mass density 4.14 gcm 3 estimate. Radiation of heat to the surroundings will have a similar e ect. Still, an absorption of 3% means that, even in the case of perfect locking and zero dispersion, equation 3.13 predicts a maximum ampli cation factor of 30 times, as opposed to the 700 times if the crystal loss were negligible. The e ect of such an additional parasitic loss on the optimal build-up factor is plotted in gure 3.23. It is clear, therefore, that the optical loss within the gallium phosphide is detrimental to cavity performance. In order to determine if it might be possible to overcome this di culty, we consider possible mechanisms contributing to this loss. 3.9.1 Loss Mechanisms in Gallium Phosphide The bandgap of gallium phosphide is 2.26 eV, or 549 nm [33]. One therefore expects minimal linear losses at a wavelength of 1030 nm, or photon energies of 1.21 eV, as this is signi cantly below the bandgap. We consider rst, therefore, the possibility of two photon absorption contributing to the optical loss at these wavelengths. In the presence of such an e ect, the intensity of a beam propagating at position z in the medium decreases as dI(z) dz = [ + TPAI(z)] I(z); (3.64) where is the common linear absorption coe cient, and TPA is the two photon absorption coe cient. In order to determine if the optical loss scaled in the manner expected, the transmission through the crystal was measured in a single-pass con guration using the ampli ed mode-locked oscillator out- put described in section 2.4. The power into the crystal was varied using a crossed polariser con guration to avoid possible changes in pulse duration 873.9. Linear Optical Loss of Gallium Phosphide 0.00 0.02 0.04 0.06 0.08 0.10 0 100 200 300 400 500 600 Additional Parasitic Loss Optim al Ava ilab le Enh an cem ent Figure 3.23: E ect of parasitic loss on the optimal enhancement factor that can be obtained from a passive enhancement cavity. The re ectivity of the input coupler is varied such that the cavity is impedance-matched for a given parasitic loss. The re ectivity of the remaining ve mirrors is 0.9997. It is useful to note that, in the absence of any parasitic loss, the impedance- matched condition would require an input coupler of transmissivity 0.15%, and this con guration would have a nesse of greater than 2000. Shown in the dotted line is the e ect of a 3% parasitic loss on the enhancement factor, as predicted from the thermal expansion data of gure 3.22. 883.9. Linear Optical Loss of Gallium Phosphide 0 0.1 0.2 0.3 0.4 0.5 0.6 0.7 0.8 0.9 10 0.1 0.2 0.3 0.4 0.5 0.6 0.7 0.8 0.9 1 Incident Power (W) Trans mitte dP owe r(W ) Figure 3.24: Average power transmitted through the gallium phosphide crys- tal as a function of incident average power. A clear linear trend is evident, with a slope suggesting a 6% linear loss, or â€™ 20m 1. that may occur if using the power ampli er pump diode current to vary the optical power. Further, the input eld was horizontally polarised so as to minimise re ection losses from the crystal, which was placed at Brewsterâ€™s angle. That the observed loss does not scale in the manner predicted by equation 3.64 can be seen from gure 3.24. The â€™ 6% linear loss shown is greater than that suggested by tting of the thermal expansion data but, in light of the assumptions made regarding thermal conductivity away from the interaction region, this is not surprising. A perhaps more satisfying conclusion is drawn from a comparison of this loss with the nesse data of section 3.7, where we considered a cavity that, without the nonlinear crystal, had a measured nesse of â€™500, such that one would expect an enhancement of â€™300 times. However, experimentally, the observed enhancement with the crystal in the cavity was only seven times, consistent with equation 3.12 for an additional parasitic loss of 7%. In light of the linear dependence of the observed optical absorption of the nonlinear crystal on input optical power, it is therefore natural to ne- 893.9. Linear Optical Loss of Gallium Phosphide glect optical losses from two photon absorption. Indeed, we note that the negligible contribution of two photon absorption processes is consistent with other works. See, for example, [53], and comments made in [54]. 3.9.2 Characterisation of the Spectral Absorption of Gallium Phosphide The observed below-bandgap optical absorption in gallium phosphide was further studied by measuring the linear transmission as a function of wave- length. This was achieved by illuminating the sample at normal incidence with narrow (0.9 nm) bandwidth light from a standard grating monochro- mator. The source used was a 100 W halogen lamp, and the detector was a silicon photodiode, found to have adequate response even at long wave- lengths. The measured intensity transmission, It was converted to a measure of the absorption coe cient, as de ned in the well known Beer-Lambert law for a propagation length x, It Iin = e x: (3.65) The expected re ection coe cient was determined using data from [33], and the transmission normalised to reference spectra. The resulting data is shown in gure 3.25. Figure 3.25 clearly shows the abrupt change in absorption correspond- ing to the bandgap of gallium phosphide of 2.26 eV (549 nm). However, of particular interest is that the below bandgap absorption coe cient does not fall immediately to zero, but rather decays slowly in a manner super- cially similar to that observed in other indirect bandgap semiconductors at room temperatures, [55] [56]. In these cases, this below bandgap absorp- tion was attributed to the phonon-assisted production of electron-hole pairs. Indeed, a study of the temperature dependence of the intrinsic absorption edge of gallium phosphide [57] attributed this below band-gap absorption at room temperatures to processes involving the absorbtion of longitudinal and transverse-optical and acoustic phonons. In the data considered here, 903.9. Linear Optical Loss of Gallium Phosphide 400 500 600 700 800 900 1000 110010 âˆ’1 100 10 1 102 Wavelength (nm) 3.1 2.48 2.07 1.77 1.55 1.38 1.24 Photon Energy (eV) 1.13 Bandgap edge Long wavelength absorption 'tail' Figure 3.25: Spectral absorption of gallium phosphide, showing the abrupt bandgap edge at â€™ 550 nm, and the additional long â€˜tailâ€™ for long wave- lengths out to 1100 nm, where the response of the silicon detector limits the measurement range. 913.10. Conclusions of Enhancement Cavity Work this tail gives rise to an absorption coe cient of â€™ 40cm 1 at 1030 nm for which, for the 2.1 mm propagation length in the enhancement cavity, one would expect â€™8% single pass loss. Scattering from lattice defects may also be a contributing factor to this large absorption at wavelengths below bandgaps. Indeed, we note that the crystals had a speci ed etch pit density (EPD, commonly used as an indi- cator of the density of lattice defects) of <(0.5-1) 105 cm 2. Whilst this is typical of commonly available gallium phosphide substrates, it is a relatively large value when compared to other semiconductors (for example, gallium arsenide crystals are commonly available with an EPD of â€™ 1 104 cm 1 [58]). Several authors have reported that higher densities of lattice defects give rise to increased broadening of the bandgap edge and, in particular, increased below bandgap absorption [59]. Whilst it is beyond the scope of the current work to study such e ects in detail, it is, perhaps, possible that such e ects may be mitigated should manufacturing processes improve to the point that gallium phosphide crystals with fewer lattice defects become available in the future. 3.10 Conclusions on the use of a Passive Enhancement Cavity for High E ciency Generation of Terahertz Radiation It is clear that the enhancements predicted by the modelling work of section 3.3 cannot be achieved unless the observed optical absorption of gallium phosphide can be overcome. As discussed above, it is possible that small reductions in the linear loss could be achieved by increasing the quality of the gallium phosphide substrates, such that scattering due to lattice defects are decreased. However, the data in gure 3.25 would suggest that the dominant loss mechanisms are phonon-assisted processes within the gallium phosphide lattice itself. This being the case, signi cant reductions in the linear optical loss can therefore only be realised in two ways. Firstly, by reducing the optical path length in the nonlinear crystal, such that the total 923.10. Conclusions of Enhancement Cavity Work absorption is reduced, or changing the crystal material entirely. In considering the possibility of alternative nonlinear materials, it is worth noting that Theuer et al. [60] were able to increase the e ciency of terahertz generation using a similar cavity con guration. In contrast to the work presented here, the authors of [60] generated terahertz radiation from Cherenkov-type optical recti cation within a lithium niobate crystal. The crystal was placed within a passive enhancement cavity seeded with the 800 nm output of a titanium-doped sapphire glass oscillator. These au- thors achieved an 8-7 times enhancement in the optical power within the cavity, which in turn produced a 5 times enhancement in the terahertz eld observed over that measured without the aid of the cavity. The enhance- ment of the cavity was again found to be limited by the â€™3% linear loss of the crystal, rising to â€™8% for higher intracavity optical powers, owing to increased two-photon absorption in lithium niobate. The lower linear losses of lithium niobate therefore make this method perhaps more appealing for enhancement cavity-based terahertz generation. However, the need to drive the crystal at 800 nm, due to phase-matching restrictions, necessitates seed- ing the cavity with titanium-doped sapphire oscillators. Ytterbium-doped bre laser systems are inherently more power-scalable with high repetition rates, whilst maintaining the frequency comb structure. Further, the con- ical output geometry of the terahertz eld in the Cherenkov-type optical recti cation leads to complications in shaping the resulting terahertz beam. Other authors have reported successes using certain organic crystals for ter- ahertz generation via optical recti cation [61], [62] at optical wavelengths of 800 nm and 1550 nm respectively. Whilst these crystals can o er increased performance in terms of the e ective nonlinear optical coe cient, to date no e orts have been made to reduce the linear optical loss at their respective pump wavelengths. As a result, the absorption coe cient for these crystals is typically greater than that measured for gallium phosphide at 1030 nm in section 3.9. Indeed, Brunner et al. quote an absorption coe cient of not more than 3 cm 1 for 2-cyclooctylamino-5-nitropyridine at 800 nm [61]. Continued research in this area may, however yield a crystal with low linear optical losses such that intracavity generation becomes more attractive. 933.10. Conclusions of Enhancement Cavity Work 0.0 0.1 0.2 0.3 0.40 200 400 600 800 Crystal Length (mm Max Possi ble En han cem ent ) Figure 3.26: Maximum possible enhancement factor provided by an impedance-matched enhancement cavity containing a nonlinear crystal hav- ing absorption coe cient, = 0:5 cm 1 and varying length. The mirror re ectivity, R = 0:99975. The cavity is assumed to be perfectly tuned to resonance for all incident wavelengths, and the input Gaussian beam is as- sumed to perfectly-matched to that of the cavity, thereby providing an upper limit on the enhancement that can be obtained for given crystal length. To illustrate the possibility of reducing the optical path length in the nonlinear crystal within the cavity, we rewrite equation 3.12 to include the crystal loss as Imax Iin = 1 Ric (1 p RRice l)2 (3.66) where l is the optical path length in the nonlinear crystal. In this case, the impedance-matching condition, equation 3.13 becomes Ric = Re l (3.67) We plot in gure 3.26 the maximum possible enhancement predicted by equation 3.66 for varying crystal lengths, assuming a crystal absorption co- 943.10. Conclusions of Enhancement Cavity Work e cient = 0:5cm 1. It is seen that, in order for the cavity to provide a relatively modest enhancement factor of 100, the maximum crystal thickness is less than 200 m . Equation 3.27 suggests that, for given optical power in- cident on the crystal, the terahertz power produced is a factor of 100 down on that expected from the 2 mm crystal used here. In light of emerging technology in the eld of high power femtosecond combs [63], [64], it would therefore seem that driving a longer crystal with such a high-power comb in a single-pass con guration o ers greater potential for the generation of high-power terahertz frequency combs. 95Chapter 4 A Single Comb Terahertz Time-domain Spectrometer with Detection Using Conventional Electro-optic Sampling 4.1 Introduction As discussed previously, there is signi cant interest in the use of terahertz radiation to characterise various molecules due to the fact that many rota- tional and vibrational molecular transitions occur in this wavelength range. The common technique to do this is known as terahertz time domain spec- troscopy, (THz-TDS). In this technique, the terahertz eld is used to inter- rogate the sample, whilst a portion of the optical beam used to generate the terahertz radiation is used to probe the transmitted terahertz pulse, most commonly using an optically gated antenna [65], or via electro-optic sampling (EOS) within a nonlinear crystal [34]. The delay between the two pulses is varied using a mechanical delay of one pulse, and the resulting trace of the terahertz electric eld is recorded as a function of the delay. Frequency analysis is performed by taking a Fourier transform in a manner identical to conventional Fourier transform spectroscopy (see for example, [65]). As conventional THz-TDS uses a sample of the driving optical eld to sample the terahertz eld, we refer to this as single-comb THz-TDS (SC THz-TDS) 964.2. Experimental Arrangement for clarity in later discussions where we use a separate laser oscillator to perform the sampling step. We have built such a SC THz-TDS system, along the same design as [34], using free-space EOS to measure the terahertz eld. In light of the results of Chapter 3, we use a gallium phosphide waveguide emitter of the type demonstrated in [66], giving an increased interaction length of the driving optical eld and the generated terahertz radiation in the nonlinear crystal, so as to increase the power of the terahertz eld for the available optical pump power. The following chapter discusses our experimental arrangement and results. 4.2 Experimental Arrangement The experimental apparatus was arranged as in gure 4.1, following a design similar to [21]. In this conventional THz-TDS arrangement, both the â€˜pumpâ€™ pulse used to generate the terahertz radiation in the nonlinear crystal, and the sampling probe pulse are derived from the same laser output. In our case, the â€™ 3W output from the ampli ed bre laser system described in section 2.4, and illustrated in gure 2.10a). The optical output pulses were compressed to 240 fs using the grating compressor arrangement described in section 2.4.2. We use a 10% beam sampling optic to provide up to â€™ 300 mW in the probe beam used to sample the terahertz eld. The high- power pump eld is then sent through a mechanical delay stage to vary the delay between the pump pulse and the sampling probe pulse used to detect the terahertz eld. This con guration gave up to 2.5W of optical power incident on the nonlinear crystal used for terahertz generation. At the laser output, the polarisation state of the light is optimised to that giving maximum di raction e ciency for the grating compressor. A subsequent half-wave plate was used in both the pump arm and the probe arm to give the optimum linear polarisation orientation for both terahertz generation (as discussed in section 2.2) and EOS within the detection crystal. Recall equation 2.29, where we saw that, under conditions of perfect phase-matching, the terahertz power generated through the optical recti - 974.2. Experimental Arrangement cation process increases with interaction length in the nonlinear crystal. We therefore seek to maximise this interaction length by increasing the crys- tal thickness (note that, in the single-pass optical recti cation scheme used here, we are not limited by the resulting increased dispersion and linear ab- sorption of the optical eld). In order to increase the interaction length of the optical recti cation process, it is necessary to ensure that the driving optical beam remains collimated throughout the interaction volume. To un- derstand this, we write the scaling of the radiated terahertz power, PTHz, as PTHz = difAITHz; (4.1) where A is the THz-emitting area within the crystal, and dif is a dimen- sionless coe cient that accounts for di raction due to the nite source area. The intensity of the driving optical eld scales as Iopt = Popt A / Popt r2 : (4.2) where r is the radius of the optical beam. For large optical spot sizes, r > THz, dif â€™ 1, and essentially all of the terahertz eld is able to propagate into the far- eld, resulting in the power of the terahertz eld scaling as PTHz / P 2opt r2 : (4.3) If r < THz, however, di ractive e ects reduce dif , and only part of the terahertz eld will radiate into the far- eld. Bethe et al. [67] showed that, for a general eld of wavelength , dif / r4 for r . In the case of the terahertz eld of wavelength THz then, PTHz / P 2 optr 2: (4.4) Equations 4.3, and 4.4 show that, for xed optical pump power, the scaling of the emitted terahertz power scales as 1 r2 for large optical spot sizes, and as r2 for small optical spot sizes. Xu et al. [68] showed experimentally that the maximum terahertz power is produced for an optical spot size roughly equal 984.2. Experimental Arrangement in radius to the wavelength of the terahertz eld. For a terahertz frequency of 1 THz, one obtains an optical beam radius of 300 m . In bulk crystals, such a spot size leads to strong divergence of the terahertz eld, reducing the spatial overlap of the optical and terahertz elds. In order to overcome this, we use a gallium phosphide waveguide emitter of the type developed in [66]. Due to the large refractive index contrast, the waveguide emitter con nes the terahertz eld such that the spatial overlap between the two beams is maintained along the 6 mm length of the waveguide. The faces of the crystal were 0.4 0.6 mm. The optical spot diameter was 190 m , as per [66], and the Rayleigh range was determined to be in excess of 8 mm using the technique described in section 3.5.2, such that the waveguide structure guides only the terahertz beam. The output optical and terahertz beams then exit the crystal co-linearly. A schematic of this arrangement is shown in gure 4.2. The diverging terahertz eld is collimated by a 1" focal length, aluminium coated o -axis parabolic mirror, and refocussed into the nonlin- ear detection crystal using a second, identical parabolic mirror. A piece of high-density PTFE was placed between the two mirrors to block the residual optical pump beam, whilst transmitting the terahertz radiation [69]. Two di erent detection crystals were used for this experiment. A 1 mm thick piece of < 110 > zinc telluride, and the 2 mm thick piece of gallium phosphide described in section 3.3. The probe pulse was coupled into the crystal using a nitro-cellulose pellicle beam splitter, such that the optical probe pulse was co-linear with the focused terahertz beam, and normally incident upon the nonlinear detection crystal. The 45% re ectivity of this pellicle splitter produced up to 200 mW of average power in the optical probe pulse incident on the nonlinear detection crystal. It was found that, despite the inferior phase-matching at the optical wavelength of 1030 nm, the zinc telluride crystal gave the better sensitivity when compared to the gallium phosphide crystal, on account of the larger nonlinear optical coe cient. In both the single comb case pictured in gure 4.1 and in the dual-comb case to be discussed later, two di erent detection schemes were utilised, both being based in EOS. The rst of these employing a balanced detector and conventional ellipsometry, and the second, using a single channel detector. 994.2. Experimental Arrangement FC1 GaP ZnT e QW P THz WP PTF E HW P HW PVariable delay Balanced detector PeB S Figure 4.1: Experimental arrangement for the single-comb terahertz time- domain spectrometer used to characterise the terahertz output prior to demonstrating the dual-comb spectrometer of section 5. The apparatus uses a single optical frequency comb (FC1) from which 10%of the optical power is picked o using a beam splitting optic (BSO) to form the probe beam. The remaining optical power is used to drive the nonlinear crystal to generate the terahertz radiation. In this arrangement, a variable delay line in the pump arm allows the temporal overlap of the terahertz and op- tical probe pulses in the electro-optic detection crystal to be varied. The probe pulse is coupled into the detection crystal using a pellicle beam split- ter (PeBS). Waveplates are used to optimise the polarisation states of the various beams so as to utilise the largest nonlinear coe cients, and a piece of polytetra uoroethylene (PTFE) is used to block the residual optical beam after the generating crystal. 1004.3. Detection of the Terahertz Frequency Comb via Electro-optic Sampling Weakly focussed input optical beam GaP waveguide Diverging THz output Figure 4.2: Beam propagation in the gallium phosphide waveguide structure. The optical beam is weakly focussed such that the Rayleigh range greatly exceeds the length of the crystal. In contrast, the terahertz beam diverges strongly by virtue of its longer wavelength. This is counteracted in the crystal due to the high refractive index contrast. The beam at the output face of the crystal diverges strongly, and is collimated using a 90 o -axis parabolic mirror. A discussion of the operation and merits of each of these detection schemes follows. 4.3 Detection of the Terahertz Frequency Comb via Electro-optic Sampling 4.3.1 The Linear Electro-optic E ect In order to detect the terahertz eld, we use electro-optic sampling [70]. In this technique, the terahertz eld is used to write a polarisation rotation onto an optical probe eld via the linear electro-optic e ect in a nonlinear detection crystal. For this work, we use two con gurations to perform this task. The rst of these splits the probe beam into orthogonal linear po- larisation states and measures a change in the relative intensities of each state due to overlap of the probe and terahertz elds in the detection crys- 1014.3. Detection of the Terahertz Frequency Comb via Electro-optic Sampling tal. The second scheme uses a crossed polariser and single detector placed behind the detection crystal. The polariser is set such that a null signal is observed for zero overlap of the terahertz and probe elds. When the elds overlap, the polarisation of the probe is altered such that some small signal is observed. For both detection schemes lock-in detection is employed in the single comb spectrometer case so as to increase the signal to noise ra- tio. There follows a detailed analysis of the electro-optic detection process, followed by a discussion of the merits of each of the two detection schemes. The terahertz and sampling optical probe elds are incident on the de- tection crystal in a collinear geometry. Wave-retarding plates are used to rotate the polarisation of the optical eld transmitted by the crystal in such a way so as to give a null signal in the absence of the terahertz eld. The e ect of the terahertz eld is to induce birefringence in the crystal according to the linear electro-optic e ect such that a non-zero signal is detected in each of the detection schemes. In the absence of the terahertz eld, the refractive index ellipsoid of the sampling crystal, that is, the surface over which the energy density, we = 1 2 ~E ~D is constant, is de ned by the relation u21 "11 + u22 "22 + u23 "33 = 1: (4.5) Here, we de ne the dimensionless vector, ~u according to the electric dis- placement vector, ~D by ~u = ~D p 2"0we ; (4.6) and "ij is the dielectric permeability tensor, which, when expressed in the principal basis of the crystal, is diagonal, hence "ij = 0 for i 6= j. To explain the change in this index ellipsoid in the presence of the terahertz eld, we start by considering "ij , and assume that it can be written in the form of a power series in the strengths of the components, Ek, of the applied eld as: 1 "ij = 1 "(0)ij + X k rijkEk + X kl sijklEkEl + : : : (4.7) where rijk is the linear electrooptic tensor, and sijkl describes the quadratic 1024.3. Detection of the Terahertz Frequency Comb via Electro-optic Sampling electrooptic e ect. Typically, sijkl rijk and we consider only the linear ef- fect here. Since "ij is real and symmetric, it is common to use the contracted index notation rhk where h = 8 >>>>>>>>>>>>< >>>>>>>>>>>>: 1 for ij = 11; 2 for ij = 22; 3 for ij = 33; 4 for ij = 23 or 32; 5 for ij = 13 or 31; 6 for ij = 12 or 21: (4.8) Relating the dielectric permeability tensor to the refractive index via 1 n2h = 1 "ij ; (4.9) we see that the deformation of the refractive index ellipsoid, that is, the lowest order changes to the optical constants 1 n2 i are described by the relation ( 1 n2 )i = 3X j=1 rijEj : (4.10) For zincblende crystals of point group 43m such as gallium phosphide and zinc telluride, the electrooptic coe cients rij vanish except r41 = r52 = r63 which, for zinc telluride has a value 4.04pm/V [71], and for gallium phos- phide 0.97ppm/V [72]. The equality of the non-zero coe cients makes the crystal uniaxial, and hence, making the common de nitions of the ordinary and extraordinary refractive indices, n2o = "11 = "22 and n 2 e = "33, the index ellipsoid is described as u21 n2o + u22 n2o + u23 n2e = 1: (4.11) Under the in uence of an applied electric eld, ~Ea = EaX i^+EaY j^ +EaZ k^, 1034.3. Detection of the Terahertz Frequency Comb via Electro-optic Sampling this ellipsoid is modi ed according to equation 4.10 such that u21 n2o + u22 n2o + u23 n2e + 2r41(Ea1u2u3 + Ea2u3u1 + Ea3u1u2) = 1: (4.12) Both zinc telluride and gallium phosphide are cubic zincblende lattices and, owing to the high degree of crystal symmetry, are optically isotropic (show zero birefringence) for zero applied electric elds. In this case, we can there- fore write ne = no. Using the coordinate system de nitions of gure 4.3 for the [110] cut crystals used here, we consider the terahertz eld to be incident along the x^0, or [-1 -1 0] direction, linearly polarised at an angle to the y^0 direction. Alternatively, in the coordinate basis of the crystal lattice, ~Ea = Eap 2 0 B @ cos cos p 2 sin 1 C A : (4.13) Equation 4.12 can then be written in matrix form as ~u ^( ~Ea) ~u; (4.14) where ^( ~Ea) is the eld-dependent impermeability tensor ^( ~Ea) = 1 n2o 0 B @ 1 0 0 0 1 0 0 0 1 1 C A+ r41 Eap 2 0 B @ 0 p 2 sin cos p 2 sin 0 cos cos cos 0 1 C A : (4.15) In order to nd the principal axis of the crystal under the in uence of this electric eld, and the relevant refractive indices, we nd the eigenvalues of the impermeability tensor, ^( ~Ea): 1;2 = 1 n2o r41Ea 2 (sin p 1 + 3 cos2 ); 3 = 1 n20 + r41Ea sin ; (4.16) 1044.3. Detection of the Terahertz Frequency Comb via Electro-optic Sampling Figure 4.3: Crystal geometry for the electro-optic detection scheme. Both the terahertz and optical probe eld are assumed to be incident along the x0 = [ 1 10] direction, with the terahertz eld polarised at an angle with respect to the y0 = [ 110] direction. 1054.3. Detection of the Terahertz Frequency Comb via Electro-optic Sampling and the normalised eigenvectors U1 = 1 2 s 1 + sin p 1 + 3 cos2 0 B B B @ 1 1 2 p 2 cos p 1 + 3 cos2 sin 1 C C C A U2 = 1 2 s 1 sin p 1 + 3 cos2 0 B B B @ 1 1 2 p 2 cos p 1 + 3 cos2 sin 1 C C C A U3 = 1 p 2 0 B @ 1 1 0 1 C A : (4.17) Under the e ect of the applied electric eld, the crystal principal axes are parallel to these eigenvectors, and the refractive indices are given by ni = 1 i ; (4.18) which yields, to good approximation, n1 =no + n3or41Ea 4 (sin + p 1 + 3 cos2 ) n2 =no + n3or41Ea 4 (sin p 1 + 3 cos2 ) n3 =no n3or41Ea 2 sin : (4.19) From equation 4.17, the vector U3 is perpendicular to the (110) crystal plane, or parallel to the direction along which the electric eld is incident. The refractive index ellipsoid of the crystal can therefore be projected into the (110) plane, as shown in gure 4.4. The orientation of this ellipsoid is described by the angle that the vector U1 makes with the [-110] crystal 1064.3. Detection of the Terahertz Frequency Comb via Electro-optic Sampling Figure 4.4: Deformation of the two-dimensional refractive index ellipsoid in the (110) crystal plane, upon which the applied dc eld, Ea is incident with polarisation angle with respect to the [-110] crystal axis. The end result is a deformation of the index ellipsoid such that the fast and slow axis are along U1 and U2 respectively. axis, which can be shown to be cos 2 = sin p 1 + 3 cos2 : (4.20) The frequency of the optical probe pulse is far greater than that of the terahertz eld. On the timescale of the probe eld, the terahertz eld there- fore represents just such a dc electric eld as Ea. If both the terahertz and optical probe elds are incident along the [110] direction, the deformation of the index ellipsoid is such that the components of the optical probe along the principal axes ~U1and~U2 receive a relative phase shift of ( ) = !d c (n1 n2) = !d 2c n3or41jETHzj p 1 + 3 cos2 ; (4.21) 1074.3. Detection of the Terahertz Frequency Comb via Electro-optic Sampling where d is the thickness of the crystal, ! is the angular frequency of the optical probe eld, and jETHzj is the magnitude of the terahertz electric eld. The polarisation of the optical probe beam is therefore altered in a manner that depends both on the polarisation and the magnitude of the terahertz eld. This is exactly analogous to the linear Pockels e ect. The crystals used here are cut with one edge parallel to the y^0 =[ 110] direction, as shown in gure 4.3. Both the terahertz and optical probe beams are incident along the x^0 =[110] direction. In order to determine the e ect of this relative phase shift on the orthoganal polarisation states of an arbitrary, but linearly polarised probe beam, we resort to the method of Jones calculus [73]. The optical probe eld has complex amplitude ELO with horizontal and vertical components described in matrix form by ~Eh = ELO 1 0 ! ; ~Ev = ELO 0 1 ! : (4.22) Upon propagating through the crystal, these components of the sampling optical probe eld undergo a phase change described in the crystal principal axes system by the matrix XT( ) = exp( in1( )!d=c) 0 0 exp( in2( )!d=c) ! ; (4.23) which we can rewrite in terms of the relative phase change, ( ) as XT( ) = exp( i ) exp( i ( )=2) 0 0 exp(+i ( )=2 ! : (4.24) Putting together these matrix components, the e ect of the nonlinear crystal on a horizontally polarised probe beam is therefore ELO R( ( )) XT( ) R( ( )) 1 0 ! : (4.25) where R( ( )) is the matrix corresponding to the rotation of the coordinate 1084.3. Detection of the Terahertz Frequency Comb via Electro-optic Sampling system by an angle ( ) from the lab to that of the crystal principal axis. 4.3.2 Comparison of Detection Schemes Up until this point, the analysis of the polarisation rotation of the optical probe beam has been independent of the detection method employed. We now consider the e ect of analysing the beam using a single channel detec- tor and crossed polariser con guration, versus the dual channel, balanced detector method illustrated in gure 4.1. In the rst case, we use a half- wave plate to rotate the polarisation of the transmitted probe beam to be perpendicular to that transmitted by the polariser. The eld of equation 4.25 after transmission through the polariser is then given by ELO 0 0 0 1 ! R( ( )) XT( ) R( ( )) 1 0 ! : (4.26) Substituting in equation 4.24 and taking the square of the transmitted ver- tical component gives the intensity measured at a photodetector positioned behind this polariser: S1Ch( ) = E 2 LO sin 2(2 ( )) sin2 ( ) 2 : (4.27) Alternatively, the nominally horizontally polarised (in the absence of the crystal birefringence induced by the terahertz eld) probe beam is passed through a quarter-wave plate at 45 to the horizontal, giving circular po- larisation. This is then analysed using a polarising beam splitter to obtain orthogonal polarisation states, and a pair of photodiodes wired in a balanced con guration. The eld transmitted by the crystal and quarter-wave plate arrangement is now obtained by multiplying equation 4.25 by R( 4 ) QWP R( 4 ) (4.28) where QWP is the commonly known Jones matrix for a quarter-wave plate 1094.3. Detection of the Terahertz Frequency Comb via Electro-optic Sampling exp( i =4) 0 0 exp(i =4) ! : (4.29) One obtains, for the eld incident on the polarising beam splitter, R( 4 ) QWP R( 4 ) ELO R( ( )) XT( ) R( ( )) 1 0 ! : (4.30) In the absence of any applied terahertz eld ( ) = 0, as predicted by equation 4.21, and XT( ) is equal to the 2 2 identity matrix. In this case, equation 4.30 predicts equal magnitudes in both orthogonal polarisation states. Analysing the eld of equation 4.30 using a polarisation beam splitter to detect each polarisation state on a balanced detector therefore produces a null signal. If the terahertz and optical elds overlap in the crystal, however, the transmitted optical polarisation is elliptically polarised. In particular, in the case where = 0 the relative phase delay between the polarisation states, ( ) is a maximum, and = 4 . In this case, equation 4.30 gives, for the components of the detected eld Edet = ELOp 2 sin( =2) cos( =2) i(sin( =2) + cos( =2)) ! : (4.31) The output of the balanced detector then gives the di erence in the inten- sities of each component Sbal = jEdet;hj 2 jEdet;vj 2 = jELOj2 2 cos2( 2 ) + sin2( 2 ) 2 cos( 2 ) sin( 2 ) cos2( 2 ) sin2( 2 ) 2 cos( 2 ) sin( 2 ) = jELOj2 2 4 cos( 2 ) sin( 2 ) = jELOj 2 sin( ) (4.32) 1104.3. Detection of the Terahertz Frequency Comb via Electro-optic Sampling In the typical case of weak terahertz eld 1, which means Sbal =â€™ jELOj 2 / jETHzjjELOj 2 (4.33) and we see the nal result, that the di erence signal is proportional to the magnitude of the terahertz electric eld. That is indeed much less than one can be seen by considering the values for r41 given earlier, and noting that = 1 would require a terahertz eld â€™ 108 V/m for a 1 mm crystal. Under the same conditions of = 0 and ( ) 1, equation 4.27 gives S1Ch = jELOj 2 sin2( 2 ) â€™ E2LO 4 2: (4.34) In both detection schemes, the power in the terahertz eld, PTHz / jETHzj2 is modulated using a mechanical chopper at frequency fchop, pro- ducing a corresponding modulation in the relative phase delay, , between the optical probe eld components along each of the crystal principal axes. A lock-in ampli er is then used to record only signal modulations at fre- quency fchop, arising due to the dependence of the photocurrent on in both cases. This further increases the signal to noise of the overall detection system. For our case, fchop â€™ 1 kHz. Comparing equations 4.33 and 4.34 shows that S1Ch Sbal â€™ 4 ; (4.35) and that, therefore, the balanced detection scheme gives greatly increased signal levels over those of the crossed polariser scheme. Indeed, it has been shown [74] that, for typical terahertz powers of several microwatts, the bal- anced detection scheme can provide signal levels three orders of magnitude greater than those from the crossed polariser scheme. The proportionality in both cases to the power in the probe eld should also be noted, as it allows the signal strength to be increased above the level of any instrument noise that may be present in the chosen detection system. Further, as the level of shot noise associated with this signal will 1114.4. Results of Single Comb Detection Scheme scale as the square root of the power in the probe, in the shot-noise-limited case the signal to noise ratio of the measurement system can be increased with increases in optical probe power, as is the case for direct detection of an optical eld. It is clear, therefore, that either a pair of photodiodes in a balanced con- guration, or a single photodiode placed behind a crossed polariser allows the measurement of the induced polarisation rotation of the optical probe pulse due to the induced birefringence of the crystal in the presence of the terahertz eld. As it is this terahertz eld responsible for this change in bire- fringence, sweeping of the delay between the two pulses results in a direct measurement of the electric eld, in contrast to conventional optical mea- surement techniques that are sensitive to only the intensity. As is common practice in terahertz time domain spectroscopy, the sensitivity of the appa- ratus is further increased through the use of lock in detection, whereby the optical pump beam is modulated at â€™1.5 kHz using a mechanical chopper, and the signal component at this modulation frequency is detected using a standard lock-in ampli er. 4.4 Results of Single Comb Detection Scheme Figure 4.5 shows typical electro-optic sampling (EOS) traces of the terahertz electric eld, obtained using the zinc telluride detection crystal and both the single channel ( gure 4.5a) and balanced detection ( gure 4.5b) schemes. In addition, the Fourier transform of the time-domain data traces are shown, indicating the spectral content of the terahertz eld. In both cases, the time constant of the lock-in ampli er used was 30 ms. A measurement was recorded at changes in stage position of 20 m in each case, and the ampli er allowed to settle for â€™200 ms for each reading. Considering rst gure 4.5a, one notes from the spectral form of the time domain trace a central frequency of â€™0.75 THz and a spectral band- width of â€™0.5 THz. The signal to noise ratio in the frequency domain is approximately 20 dB. Also of note is the secondary pulse, seen with a de- lay of approximately 6 mm from the main pulse, that can be attributed to 1124.4. Results of Single Comb Detection Scheme âˆ’6 âˆ’4 âˆ’2 0 2 4 âˆ’0.5 0 0.5 Optical Path Delay of Terahertz Pulse (mm) Am pl itu de 0 0.5 1 1.5 2 2.5 3 3.5 40 0.2 0.4 0.6 0.8 1 Frequency (THz) Am pl itu de (a) Electro-optic sampling trace with single channel detector âˆ’6 âˆ’4 âˆ’2 0 2 4 âˆ’0.5 0 0.5 Optical Path Delay of Terahertz Pulse (mm) Am pl itu de 0 0.5 1 1.5 2 2.5 3 3.5 40 0.2 0.4 0.6 0.8 1 Frequency (THz) Am pl itu de (b) Electro-optic sampling trace acquired using balanced detection scheme. Figure 4.5: Time-domain electro-optic sampling traces of the generated ter- ahertz eld, acquired with both a single channel detector and crossed po- lariser, and by balanced detection of orthogonal polarisation states, as ex- plained in section 4.3. The spectral content of the signal is computed via Fourier transform. 1134.4. Results of Single Comb Detection Scheme etaloning e ects of the terahertz pulse in the 1 mm zinc telluride detection crystal. That this is the case can be veri ed by calculating the optical path length corresponding to one roundtrip within the crystal, having a refractive index at the terahertz wavelength of n(0:75 THz) =3.2. Fresnel re ections at the crystal-air boundary due to the high refractive index contrast there- fore cause a portion of the terahertz pulse to propagate through twice the thickness of the detection crystal, which will interfere with the optical probe pulse when the terahertz pulse is advanced with respect to the position of maximum signal by an optical path length 2 n(0:75 THz) l â€™ 6.4 mm, where l is the thickness of the detection crystal. Likewise, the Fresnel re ec- tions for such a refractive index contrast suggest that the pulse undergoing an additional two re ections at the crystal-air boundary is â€™ 15 of the power of that passing straight through the crystal, in good agreement with the relative pulse amplitudes shown in gure 4.5a. That it is the terahertz pulse undergoing the additional re ections within the crystal, and not the optical pulse can be seen by the secondary pulse appearing for shorter delays of the terahertz pulse with respect to the optical probe. The Fourier transform of the time-domain EOS trace shows interference fringes as a result of this etaloning e ect, with the fringes having spacing equal to the free spectral range of this â€˜cavityâ€™, or c 2nTHzl â€™ 50 GHz. Figure 4.5b shows an EOS trace obtained under similar experimental conditions to gure 4.5a, but using the balanced detection scheme described in section 4.3. The expected increase in signal level is not observed. This is attributed to di culties with aligning the beam into the two detector chan- nels and, in particular, with maintaining the coupling of each polarisation state into the photodiodes as the mechanical delay stage is swept. Indeed, the drift in dc signal level with mechanical delay is seen in the time domain trace, and this is re ected as a non-zero level of frequency components near dc in the Fourier transform. 1144.5. Conclusions 4.4.1 Signal Scaling with Optical Probe Power Equations 4.27 and 4.34 show that, in both the single channel and balanced detection schemes, the expected signal magnitude scales with the power in the optical probe eld. In later work, we replace the sampling optical probe eld with that from a separate mode-locked bre laser. As this second laser lacks the high-power ampli er stage of that used here, we anticipate a reduction in the available optical power in the sampling eld. We therefore seek to verify the scaling of the observed signal with optical probe power. To do this, the power in the probe beam was varied using standard neutral density optical lters, placed in the probe beam after the sampling plate used to pick o the probe pulse. The resulting time-domain traces can be seen in gure 4.6. That the photocurrent scales linearly with the intensity of the probe eld, as predicted by equation 4.27, can be seen by plotting the magnitude of the peak of the non-normalised detected photocurrent signal with respect to the average power in the probe pulse, as in gure 4.7. It should be noted that this neglects any variation in the phase o set between the carrier wave of the terahertz eld and the terahertz envelope. As seen from the shifting of the peaks of the carrier wave from the maxima of the envelope in gure 4.6, there is such a phase o set, likely due to the mismatch of the terahertz phase and group velocities in air. However, it is clear from gure 4.6 that this o set does not vary between scans, consistent with phase-group velocity mismatch over a xed path length. Indeed, the only conditions that would give rise to a change in this o set would be a change in the total optical path length of the terahertz pulse. As this o set remains constant, it is reasonable to compare the peak of the electro-optic sampling signal between scans. 4.5 Conclusions Our SC THz-TDS arrangement demonstrates signal to noise ratios ofâ€™20 dB for our available probe powers. In order to compare this to other work, we need to obtain a numerical estimate of the terahertz power of our system. 1154.5. Conclusions 5 10 15 20 25 30 Delay (mm) 200mW 125mW 69mW 14mW 5.3mW 2.1mW 0.5mW Sig na l A mp litu de Figure 4.6: Resulting time-domain EOS traces for varying average powers of the optical probe pulse. The traces have been normalised to the peak signal of each trace and shifted in delay by 2 mm with respect to the preceding trace for clarity. 1164.5. Conclusions 10âˆ’1 100 101 102 103 LO Power (mW) Relativ ePe ak Photoc urren t Figure 4.7: Scaling of the peak photocurrent observed from the single chan- nel electro-optic sampling apparatus as a function of average power of the optical pulse. A dual logarithmic plot is used to overcome the di culties in producing small changes in high average powers of the sampling pulse. Such measurements are notoriously di cult, hence the prevalence of the electro-optic sampling technique used here, which produces a measurement of the terahertz electric eld, but includes a response of the electro-optic ma- terial which is typically di cult to characterise su ciently to allow an exact calibration [75], [76]. The most reliable way to produce an absolute mea- surement of the terahertz power is to use a helium-cooled bolometer, such as that used by [66]. Such instruments are slow and expensive to operate due to the need for helium cooling. Whilst solid state pyroelectric devices have been demonstrated [77], and indeed are now commercially available, these devices remain uncalibrated at frequencies lower than â€™5 THz. Further, the speed of these pyroelectric devices remains limited to around 5 Hz. Rather than measure the terahertz power directly, we therefore extrap- olate from the results of Chang et al. [66] who used an identical gallium phosphide wave guide emitter for terahertz generation via optical recti ca- tion. The terahertz power obtained in this case was 120 W for a 14 W 1174.5. Conclusions optical pump. One notes that the central wavelength, observed terahertz spectrum, and driving optical wavelength are all comparable to our work, the only di erence being additional ampli cation stages allowing higher op- tical pump powers. From equation 2.14, which states that the magnitude of the nonlinear polarisation at the di erence frequency scales with the square of the driving electric eld, one derives PTHz / P 2opt, from which one esti- mates a total terahertz power of 2.5 W for the experimental arrangement demonstrated in this thesis. We now compare the performance of our SC THz-TDS system to the work of Ho man et al. [78]. Whilst these authors use a di erent scheme for generation of the terahertz eld, the result is comparable terahertz pow- ers to that calculated above for our system. These authors use free-space electro-optic sampling in gallium phosphide to obtain signal to noise ratios of â€™20 dB, which is comparable to that observed in gure 4.5a. In common with other THz-TDS experiments [21], [79], the need to sweep the mechanical delay line in our experiment results in di culties maintaining the overlap of the optical and terahertz beams in the sam- pling crystal, and in long data acquisition times of four minutes for the data shown in gure 4.5. The following chapters discuss a method of overcoming these di culties by eliminating the mechanical delay line. 118Chapter 5 Dual-comb Spectroscopy in the Terahertz and Far-IR Region 5.1 Introduction The use of mechanical delay lines in terahertz time domain spectrometry (THz-TDS) is fraught with drawbacks. Varying the optical path gives rise to changes in optical alignment and focussing which, in turn, will produce signal artefacts that are di cult to isolate from behaviour of the sample itself. Further, and perhaps more importantly, the speed with which the delay can be swept is limited, such that data acquisition is extremely time consuming. Note that the frequency resolution of such a spectrometer system depends on being able to vary the optical path length over large distances which, for the relatively slow velocities with which it is possible to vary the path delay, increases the required data acquisition time. For example, Mittleman and co-workers reported on a gas-sensing THz spectrometer that required tens of milliseconds in order to complete a single scan with adequate signal to noise [2]. The following chapter proposes a method of overcoming these drawbacks, whereby two mode-locked lasers, di ering slightly in the pulse repetition rate, are used to provide the two required optical elds in a technique knowns as asynchronous electro-optic sampling (A-EOS) [80, 81]. In this method one mode-locked laser is ampli ed and used for generation of the terahertz eld via either optical recti cation [82] in a suitable nonlinear crystal, or by 1195.1. Introduction driving a photoconductive antenna [80]. A second mode-locked laser is used to probe the terahertz eld with a slight di erence in pulse repetition rate, such that the pulse overlap between the terahertz and sampling eld in the detection crystal is modulated at a frequency given by the repetition rate detuning, frep. In this way, all moving parts are eliminated, and the speed with which data can be acquired greatly increased [83]. Related techniques using detuned frequency combs have already been demonstrated at mid-to-near infrared wavelengths. For example, Schlei- iser et al. [84] demonstrated a Fourier transform spectrometer covering the spectral region 9-12 m (25-33 THz) for real-time remote gas analysis. Fur- ther, Reinhardt et al. [85] have demonstrated such a system operating at near-IR wavelengths, in the spectral range 819-830 nm (362-368 THz) for spectroscopy of atomic transitions. More recently, the technique has been applied to the terahertz range. In particular, Bartels et al. [86], [83] have shown that asynchronous optical sampling of a terahertz frequency comb allows rapid acquisition of the spectral lines in water vapour. This sys- tem uses an electro-optic antenna to generate the terahertz pulse from a mode-locked titanium-sapphire oscillator operating at 800 nm, and a second sampling pulse from a similar oscillator to interrogate the terahertz eld in a nonlinear zinc telluride crystal. The two oscillators have nominal pulse repetition rates of 1 GHz, but are detuned by frep=10 kHz by a system of rf electronic feedback to an intracavity piezo-electric actuator. Barbieri et al. [87] recently showed that an optical femtosecond fre- quency comb (FFC) could be used to characterize a quantum cascade laser (QCL) operating at 2.5 THz. With a time domain analysis and subject to certain limitations on rf bandwidth, they showed that A-EOS can be used to map the THz frequency comb (THz FC) of a QCL to an rf photocurrent, demonstrating the comb nature of the QCL output. In this section we complete a fully frequency domain analysis of A-EOS. Using this technique (which is a new form of dual-comb spectroscopy em- ploying electro-optic sampling) our results indicate it is possible to pursue wideband precision spectroscopy in the far-IR and THz spectral regions by directly converting the far-IR/THz frequency combs to the rf domain. To 1205.1. Introduction this end, we show that it is possible to signi cantly improve the frequency resolution of the technique developed in [80] such that the individual comb elements of the THz FC are recorded. Such a technique would allow the Hz-level frequency resolution [23] previously discussed (see section 1.1.3) to be of such interest at these frequencies. Figure 5.1 outlines the key steps in the detection process. In the rst step, the A-EOS process creates a comb of sidebands adjacent to each ele- ment of the optical local oscillator comb used to interrogate the THz FC. As shown in gure 5.1, each set of sidebands contains the full spectral in- formation of the original THz FC and therefore represents a scaled copy of the THz FC. A subsequent direct heterodyne detection process on a fast photodetector beats each set of sidebands with the nearest LO comb ele- ment to produce the rf frequency spectrum shown in gure 5.1. We show, with a full frequency domain analysis and subject to appropriate limitations on the repetition rate detuning, that each of the resulting rf comb elements is produced by a single element of the THz FC. Further, the comb of rf beat frequencies is symmetrical about every half pulse repetition rate of the sampling optical comb. One can therefore recover the entire spectral in- formation of the original THz FC by recording, for example, the frequency content between dc and half the pulse repetition rate, as illustrated in gure 5.1. Spectroscopy of a sample can be performed by repeating this measure- ment with and without the sample placed in the THz FC beamline. We show here, with a fully frequency domain treatment, that optical recti cation of an optical FFC results in a THz FC and, further, that asyn- chronous electro-optic sampling of this THz FC with a second, detuned local oscillator (LO) FFC writes a comb of rf phase sidebands onto the sampling optical pulse. Subsequent heterodyne beating of this modulated optical comb gives rise to a comb of rf frequencies, with the crucial characteristic that each rf comb element is produced by beating processes involving one and only one element of the original terahertz comb. In this manner, a sin- gle, unique copy of the original terahertz eld is created at readily detectable rf frequencies. The nature of each frequency comb at each stage of the experimental 1215.1. Introduction Figure 5.1: Overview of the multi-frequency heterodyne detection scheme for the dual-comb terahertz spectrometer. Asynchronous electro-optic sam- pling (A-EOS) results in a scaled copy of the original THz frequency comb (THz FC) being created adjacent in frequency to each element of the local oscillator (LO) optical comb. Further mixing of these sidebands with the LO comb during the direct heterodyne detection process produces a comb of rf frequencies, periodic in the pulse repetition rate of the LO comb and symmetric about half the pulse repetition rate. Careful analysis (see text) shows that each rf comb element arises from a single THz FC comb element, such that the full spectral content of the original THz FC is contained within a frequency range less than half the LO pulse repetition rate, where it may be recovered electronically. 1225.2. Terahertz to RF Comb Mapping arrangement is shown in gure 5.2. A THz FC is generated by di erence frequency generation (DFG) from an optical FFC with a repetition rate frep + frep. In the case where the THz FC is generated using a single optical comb the former necessarily has a zero o set frequency, as explained in section 2.1. This distinction is made as recent work [88] has demonstrated that DFG between two optical FFCs can extend the bandwidth of the dif- ference frequency comb up to the mid-infrared. In this case, the following theory still applies, but one must be careful to include the resulting o set frequency of the di erence frequency comb. The generated THz FC is used to probe a sample that imparts phase and amplitude changes onto this comb. It is this terahertz spectroscopic data that we seek to transfer into the rf region of the spectrum, where it may be easily detected and analysed using conventional electronics. This is achieved by using a second optical FFC, the LO FFC. The THz FC has a repetition rate that is detuned from that of the LO FFC, frep, by an amount frep. Upon mixing with the terahertz comb in a second nonlinear crystal, referred to henceforth as the electro-optic sampling (EOS) crystal, a polarisation modulation is produced by the EOS process. Due to the di ering repetition rates of the two combs, this results in a set of polarisation modulation side- bands being written onto each comb element of the LO FFC. As we show in the following section, direct detection of the LO FFC produces a comb of rf frequency modulations, each frequency element of which correspond uniquely to a particular terahertz comb element, allowing recovery of both the phase and amplitude spectroscopic data. 5.2 Mapping of a Terahertz Frequency Comb to the Radio Frequency Spectral Region via Asynchronous Electro-optic Sampling In the frequency domain EOS can be viewed as sum frequency generation (SFG) and DFG [89] between the LO FFC (serving as the probe) and the 1235.2. Terahertz to RF Comb Mapping WP FFC GaP THz LO FFC Sample Sampled THz comb Modulated output comb EOSE Figure 5.2: Schematic of a terahertz spectrometer where the pump fem- tosecond frequency comb (FFC) and local oscillator (LO FFC) frequency combs di er in pulse repetition rate by an amount frep. The terahertz frequency comb (THz-FC) is generated by di erence frequency generation in a nonlinear crystal (GaP). After passing through the sample to be stud- ied, the terahertz comb is electro-optically sampled using the detuned LO frequency comb in a second nonlinear crystal, the electrooptic sampling el- ement (EOSE). Following a polariser, direct detection of the sampled LO frequency comb completes an element-by-element mapping from the tera- hertz frequency range to the rf domain. 1245.2. Terahertz to RF Comb Mapping THz FC2. In the following section, we show that the SFG and DFG processes of asynchronous electro-optic sampling create sidebands adjacent to each element of the LO FFC, within each set of which is contained the full spectral and amplitude information of the original THz FC, as in gure 5.3a) and c). We refer to these sidebands as A-EOS sidebands. Subsequently, we will show that the beating process arising from direct detection of the A-EOS sidebands with the original LO FFC on a fast photodetector creates an rf photocurrent, itself having many comb elements, each corresponding to a unique element of the original THz FC, as illustrated in Fig 5.3b). We begin our analysis by considering the polarization rotation imparted via SFG and DFG onto the LO FFC through the asynchronous electro-optic sampling process. The elds incident on a polarisation beam splitter placed behind the nonlinear detection crystal consist of the original LO FFC, as well as those elds produced through SFG and DFG with the THz FC3. We write these latter two elds as ~ESFGEOS (t) and ~E DFG EOS (t) respectively. The total incident eld at time t can then be written ~Etot = ~ELO(t) + ~E SFG EOS (t) + ~E DFG EOS (t): (5.1) Let the transmission axes of the polarising beam splitter be denoted x^ and y^. The elds along these axes are then ~Etot(t) = Etot;x(t)x^+ Etot;y(t)y^; (5.2) where Etot;x(t) = ELO;x(t) + E SFG EOS;x(t) + E DFG EOS;x(t) Etot;y(t) = ELO;y(t) + E SFG EOS;y(t) + E DFG EOS;y(t): (5.3) 2Whilst it is intuitive to think of A EOS as the mixing of di ering frequency elements, the same can be seen to be true of conventional EOS employing a swept mechanical delay by realising that the e ect of the mechanical delay is to Doppler-shift the frequency elements of the portion of the optical FFC used as the LO by an amount 2frep v c , where v is the sweep velocity of the mechanical delay [25]. 3Of course, the THz FC itself propagates through to the detector also, but as our detector is not sensitive to such long wavelengths we need not include it here. 1255.2. Terahertz to RF Comb Mapping We follow the procedure of [89] and rewrite these polarisation components as Etot;x(t) = ELO;x(t) (1 + iâ€™x(t)) Etot;y(t) = ELO;y(t) (1 + iâ€™y(t)) (5.4) where the phase modulations, â€™x(t) and â€™y(t) are given by â€™x(t) = ESFGEOS;x(t) + E DFG EOS;x(t) iELO;x(t) â€™y(t) = ESFGEOS;y(t) + E DFG EOS;y(t) iELO;y(t) : (5.5) For reasonable eld amplitudes â€™x; â€™y 1 thus equation 5.4 can be well approximated by Etot;x(t) = ELO;x(t) exp(iâ€™x(t)) Etot;y(t) = ELO;y(t) exp(iâ€™y(t)): (5.6) The e ect of the SFG and DFG elds is therefore to create an amplitude modulation in the orthogonal polarisation states of the measured LO eld. This amplitude modulation can be recorded either directly using a crossed- polariser geometry, or by using ellipsometry involving balanced detection of orthogonal polarizations, analogous to the schemes demonstrated in section 4.3. The simultaneous mixing of multiple comb elements via both SFG and DFG necessitate careful consideration in order to verify the unique mapping of the THz FC elements to a corresponding heterodyne beat in the rf domain of the resulting signal. We will show that, in the case where the LO and THz elds are of the form of broadband, detuned FCs, the resulting amplitude modulation can be described by phase modulation sidebands of the original LO FFC. 1265.2. Terahertz to RF Comb Mapping The mode-locked oscillator used as the LO FFC can be described by, ELO(z; t) = X M ALO(z; !M ) exp( i!M t) (5.7) where ALO(z; !M ) is the complex spectral amplitude of the frequency com- ponent at frequency !M and location z respectively. We include within the complex spectral amplitude the spectral phase of the LO FFC. The discrete comb elements are indexed by M , !M = 2 (Mfrep + f0) (5.8) where frep is the laser pulse repetition rate, f0 is the carrier-envelope o set frequency, and M runs over the bandwidth of the laser. The THz FC, gener- ated via DFG with an optical FFC that has an element spacing (repetition rate) of (frep + frep) can be described by, ETHz(z; t) = X N ATHz(z; N ) exp( i N t) (5.9) Again, the complex spectral amplitude ATHz(z; N ) includes within it the spectral phase of the N th THz comb element. As this THz FC is assumed to be generated by DFG of optical comb elements having the same o set frequency, there is zero o set frequency of the THz FC comb elements. The comb elements are indexed by N and, further N = 2 N(frep + frep): (5.10) The nonlinear polarization produced by a single THz FC element at fre- quency N due to the SFG process is given by, P N (z) !M+ N (z) = (2)(!M + N ;!M ; N )ALO(z; !M )ATHz(z; N ) (5.11) and, for the DFG process, P N (z) !P N (z) = (2)(!P N ;!P ; N )ALO(z; !P )A THz(z; N ) (5.12) 1275.2. Terahertz to RF Comb Mapping where the index P is introduced to index the LO FFC during DFG. Through- out this discussion, we use a frequency superscript to indicate a quantity that varies in time with the indicated frequency, whilst a subscript argument in- dicates the frequency of the THz FC element producing the particular LO sideband. P!M+ N N (z) therefore represents a nonlinear polarization at fre- quency (!M + N ) produced by the SFG process between the THz FC line at frequency N and the optical LO FFC element at frequency !M , whilst P!P N N (z) represents the nonlinear polarization due to the DFG process between the LO FFC element at frequency !P and the THz FC line at frequency N . We now follow the principals of section 2.1 and assume that the eld radiated as a result of the nonlinear polarisation of equations 5.11 and 5.12 satis es the slowly varying amplitude approximation. That is, the amplitude of the radiated eld varies slowly on the scale of the wavelength. The radi- ated eld amplitude, AEOS , of the sampled LO FFC at frequency !M + N at position z in the nonlinear detection crystal due to SFG therefore satis es a wave equation of the form of equation 2.19, @ @z + (!M + N ) AEOS(z; !M + N ) = 2 i c2 (!M + N )2 k0(!M+N ) P!M+ N N (z) exp(i k0+z [ ( N ) + (!M )] z): (5.13) Similarly for DFG, @ @z + (!P N ) AEOS(z; !P N ) = 2 i c2 (!P N )2 k0(!P N ) P!P N N (z) exp(i k0 z [ ( N ) + (!P )] z); (5.14) where k0+ = k 0(!M )+k0( N ) k0(!M + N ) and k0 = k 0(!P ) k0( N ) k0(!P N ) correspond to the wavevector mismatches in the SFG and DFG cases respectively. For consistency with [89], we use k = k0+ i such that k0 represents the real part of the wavevector, and the absorption coe cient. 1285.2. Terahertz to RF Comb Mapping Substituting in the previously derived nonlinear polarizations of Eqs. 5.11 and 5.12, and integrating over the length, l of the nonlinear crystal, we obtain AEOS(l; !M + N ) = 2 i c2 (!M + N )2 k0(!M+N ) (2)(!M + N ;!M ; N ) exp(i k+l) 1 i k+ exp( (!M + N )l) ALO(!M )ATHz( N ); (5.15) for the SFG term, and AEOS(l; !P N ) = 2 i c2 (!P N )2 k0(!P N ) (2)(!P N ;!P ; N ) exp(i k l) 1 i k exp( (!P N )l) ALO(!P )A THz( N ); (5.16) for the DFG term. We make the undepleted pump approximation and as- sume ALO(z; !) and ATHz(z; ) to be constant in z. Substituting the fre- quency equations 5.8 and 5.10, and summing over the LO FFC elements gives the total eld due to the THz FC element at frequency N = N(frep+ frep) ESFGEOS; N (l; t) = X M 2 i c2 (!M + N )2 k0(!M+N ) (2)(!M + N ;!M ; N ) exp((i k+l)) 1 i k+ exp( (!M + N )l) ALO(!M )ATHz( N ) exp( 2i ((M +N)frep +N frep) t)] ; (5.17) 1295.2. Terahertz to RF Comb Mapping and EDFGEOS; N (l; t) = X P 2 i c2 (!P N )2 k0(!P N ) (2)(!P N ;!P ; N ) exp((i k l)) 1 i k exp( (!P N )l) ALO(!P )A THz( N ) exp( 2 i ((P N)frep N frep) t)] : (5.18) Focussing solely on the time dependence of the above expressions, it can be seen that each THz FC line of frequency N produces a sideband on every LO FFC element (at frequency !M for the SFG case, and !P in the DFG case), with the spacing of that sideband from the LO FFC element from which it is derived being equal to the THz frequency N = N(frep+ frep). As this spacing is not a multiple of frep (the spacing of the LO FFC elements themselves) each THz sideband is displaced from an LO FFC element other than that used in its generation by an amount N frep. Further, we note that this is true for both the SFG and DFG cases, with the THz sideband being greater or lesser in frequency than the nearest LO FFC element in the SFG or DFG cases respectively. By considering the SFG and DFG processes within the same LO FFC spectral interval, we note that that the M +N + 1th LO FFC element in the SFG case corresponds to the P N th LO FFC element in the DFG case. This enables us to express equations 1305.2. Terahertz to RF Comb Mapping 5.17 and 5.18 as a single sum, EEOS; N (l; t) = 2 i c2 X M (!M + N )2 k0(!M+N ) (2)(!M + N ;!M ; N ) exp(i k+l) 1 i k+ ALO(!M )ATHz( N ) exp( 2 i ((M +N)frep +N frep) t) + (!M+2N+1 N )2 k0(!M+N+1) (2)(!M+2N+1 N ;!M+2N+1; N ) exp(i k l) 1 i k ALO(!M+2N+1)A THz( N ) exp( 2 i ((M +N + 1)frep N frep) t) (5.19) It is important to note this sum represents the sidebands resulting from a single THz FC element, speci cally the N th element. When we consider the THz FC in its entirety, we nd the nal result that a sideband corre- sponding to each element of the THz FC is created both above and below every single LO FFC element, as shown in gure 5.3, and we refer to these sidebands as A-EOS sidebands. In general, these sidebands will have a dif- ferent polarization to that of the LO FFC. In order to observe a heterodyne beat between these eld components, we use either an optical polarizer to take parallel projections of the original LO FFC and A-EOS sidebands, or a quarter waveplate and balanced detection of orthogonal polarization states, as in the case of ellipsometry discussed in [89]. Having fully described the nature of the A-EOS sidebands written onto the LO FFC by the THz FC, and henceforth referring to this modulated comb as the sampled LO FFC, we now consider the direct detection process of this modulated comb, using a polariser to take parallel projections of the modulation sidebands and the original LO elements. With a judicious choice of frep the direct detection process produces rf modulation frequencies in the four main bands as shown in gure 5.3 b). First we start with the A-EOS SFG sidebands, connected graphically 1315.2. Terahertz to RF Comb Mapping M,N M-1,N+1M+1,N-1 M+N LO line M+1+ N LO line Ï… fRF M,N M+1,N-1M-1,N+1 M LO line P-1-S or M+N LO line Ï… P-S or M+1 +N LO line SFG sidebands c) DFG in A-EOS b) RF spectrum a) SFG in A-EOS Band A Band B Band C Band D P LO line Figure 5.3: a) Optical frequency spectrum of the LO FFC after SFG with the THz FC. In the SFG process, the N th THz FC element writes a sideband onto the M th LO FFC element, producing a modulation at a frequency near the (M +N)th LO FFC element. b) RF spectrum of directly-detected photocurrent due to the modulated LO FFC obtained through asynchronous optical sampling. Four bands of heterodyne beats are formed. Of interest are the middle two bands (which are conjugate pairs) as each rf beat within these bands is created by a unique element of the THz FC. c) Optical spectrum of the LO FFC after di erence frequency generation (DFG). Similar to SFG, the process of DFG writes a group of sidebands onto the LO FFC and, after direct detection of the sampled LO, there is the same unique mapping of a particular THz FC element to a unique rf heterodyne beat as shown in b). 1325.2. Terahertz to RF Comb Mapping between panels a) and b). Bands A and D result from the intra-A-EOS sideband beating and as they are very low in power they are not considered further. Of more interest is the range of beat frequencies due to beating of the M th group of A-EOS SFG sidebands with the (M + N)th LO FFC element (Band B). A third group of beat frequencies (Band C) is due to the M th group of A-EOS SFG sidebands mixing with the (M + N) + 1th LO element as shown in gure 5.3b). This group of beats is a mirror image of Band B, re ected about frep=2. Continuing this analysis shows each group of sidebands beats with its nearest and next nearest LO elements and contributes an identical beat structure to the rf spectrum shown in gure 5.3 b). The DFG sidebands will also contribute to the photodetectorâ€™s rf signal generating a similar set of bands of rf frequencies within the range dc to frep in the photodetection signal. In the central two bands, the N th A-EOS DFG sidebands will appear as an rf beat at exactly the same frequency as the N th A-EOS SFG sidebands. This overlap is shown pictorially in gure 5.3 b). Similar to the SFG case, each group of A-EOS DFG sidebands beats with its nearest and next nearest LO elements. Herein lies the key point: for A- EOS to be employed as a detection method for multi-frequency heterodyne THz spectroscopy each rf beat must result from one and only one THz FC element. Indeed this is the case, as the N th mode of THz FC is the only element responsible for generating the rf beat at N frep from both SFG and DFG sidebands. Limits on frep necessary to produce this unique mapping are discussed below. The photocurrent due to the simultaneously incident A-EOS sidebands and LO elds is proportional to the square of the electric eld described by equation 5.1 [90], S(l; t) / j ~ELO(t) + ~E SFG EOS (l; t) + ~E DFG EOS (l; t)j 2: (5.20) Neglecting the dc photocurrent terms corresponding to each of the LO and A-EOS elds incident on the detector individually, and subject to the band- width limitations discussed, one obtains the time-varying term due to the 1335.2. Terahertz to RF Comb Mapping mixing of the LO eld with the A-EOS sideband. S N (l; t) / E SFG EOS; N (l; t)E LO(l; t) + E DFG EOS; N (l; t)ELO(l; t): (5.21) Substituting in equations 5.7, 5.8, 5.17 and 5.18 yields, S N (l; t) / 2 i c2 X M (!M + N )2 k0(!M+N ) (2)(!M + N ;!M ; N ) exp(i k+l) 1 i k+ ALO(!M )ATHz( N )A LO(!M+N ) exp ( 2 i [((M +N) (M +N)) frep +N frep] t) + (!M+2N+1 N )2 k0(!M+N+1) (2)(!M+2N+1 N ;!M+2N+1; N ) exp(i k l) 1 i k A LO(!M+2N+1)ATHz( N )ALO(!M+N+1) exp ( 2 i [((M +N + 1) (M +N + 1)) frep +N frep] t) : (5.22) Upon simplifying the time dependence in the exponential terms, and taking relevant terms outside the sum, the true characteristics of this heterodyne comb become clear: S N /ATHz( N ) exp( 2 iN frept) 2 i c2 X M (!M + N )2 k0(!M+N ) (2)(!M + N ;!M ; N ) exp(i k+l) 1 i k+ ALO(!M )ALO(!M+N ) + (!M+2N+1 N )2 k0(!M+N+1) (2)(!M+2N+1 N ;!M+2N+1; N ) exp(i k l) 1 i k ALO(!M+2N+1)A LO(!M+N+1) : (5.23) 1345.3. E ect of Spectral Phase on Heterodyne Beat Signal The term outside the sum can be rewritten ATHz( N ) exp ( 2 iN frept) = ATHz( N ) exp i N frep frep t : (5.24) Comparison with equation 5.9 therefore shows that the nal result of equa- tion 5.23 is analogous to a map of the original THz FC, but scaled in fre- quency by a factor frepfrep and weighted by the amplitude and phase of the LO FFC, as well as the phase-matching and nonlinear susceptibility terms due to the optical mixing processes. Recalling that the dc photocurrent terms corresponding to the photocurrents measured if any of the elds ELO, ESFGEOS or E DFG EOS are detected independently were neglected in deriving equa- tion 5.21, the nal signal can be seen to have the form of, in the time do- main, the sum of these dc photocurrents, upon which is written an amplitude modulation at rf frequencies described by equation 5.23. With appropriate rf ltering, only those modulations at frequencies within the range of Band B, as indicated in gure 5.3 will be observed. Assuming the strong dc portion of the signal to be removable with con- ventional electronic ltering techniques, we now turn our attention to an analysis of the factors a ecting the magnitude of the rf modulation. The sum over the optical LO FFC elements will vary with the spectral phase of the input elds, and with the wavevector mismatch in the nonlinear material used for the EOS crystal. If the SFG and DFG signal components gener- ated by a single THz FC element at frequency N have opposite phases, for example, one can expect signi cant degradation of the heterodyne beat signal at frequency N frep. The e ect of the LO FFC spectral phase, and the wavevector mismatch on the magnitude of the heterodyne beat signal forms the basis of the following section. 5.3 E ect of Spectral Phase on Heterodyne Beat Signal Consider again equation 5.23. There is, contained within the complex spec- tral amplitude ALO of the local oscillator a spectral phase associated with 1355.3. E ect of Spectral Phase on Heterodyne Beat Signal each and every optical comb element. Whilst a given rf comb element is produced by the beating of many optical sidebands with their respective nearest neighbour, it is seen from equation 5.23 that every single sideband contributing to a given rf comb element is produced by a single terahertz comb element, allowing the full structure of the THz FC to be retrieved us- ing commonly available rf electronics. The amplitude of any given rf comb element will, however, depend on the phase distribution of the many optical sidebands involved in its generation, and of the LO FFC elements with which they are mixed. This can be understood by considering the spectral phase terms within the sum of equation 5.23. If the spectral phase of the optical comb is at, such that all the terms within the sum have the same phase, the amplitude of the signal will be a maximum. In contrast, any spectral chirp to the optical LO comb will reduce the amplitude of the heterodyne beat, due to the phase mismatch between contributions to the single rf frequency. Here we consider the e ect of various contributions to the spectral chirp of the LO comb on the heterodyne beat signal. In general, the output pulse from a bre oscillator such as that used to provide the LO comb in this work will exhibit a non-uniform spectral phase distribution due to the combined e ects of material and waveguide dispersion, as described in section 2.4.2, as well as the nonlinear phase shift resulting from the intensity dependence of the material refractive index. Recall expression 2.36 describing the propagation of an optical pulse in a dispersive medium: E(z; t) = ei!ct Z ~E(0; )ei( t (!c+ )z)d ; (5.25) and the Taylor expansion of the propagation constant (!c + ) thus, (!c + ) = (!c) + d d! !c + 1 2 d2 d!2 !c 2 + 1 6 d3 d!3 !c 3 + : : : ; (5.26) where !c and are the optical carrier frequency and ! = !c + , as de ned in section 3.3.3. The electric eld of the local oscillator comb, having prop- agated a distance z in the silica bre components composing the oscillator 1365.3. E ect of Spectral Phase on Heterodyne Beat Signal and subsequent ampli er stages, can therefore be rewritten E(z; t) = ei!ct Z ~E(0; ) exp(+i t ikz i 2 2 2z i 6 3 3z)d ; (5.27) where i = di d!i and 2 and 3 are the GVD and TOD parameters respec- tively. 2z is the total second order group delay dispersion (GDD). The combined e ect of material and waveguide dispersion is therefore to write onto the electric eld a spectral phase consisting of terms proportional to increasing powers of the optical frequency . As a speci c example for illustration, we consider here the e ect of a non-uniform spectral phase of the LO FFC on the heterodyne beat signal resulting from the N th THz FC element and assume perfect phase-matching, the subject of which forms the basis of the following section. We choose an LO FFC centered at 1030 nm and a THz FC element at 0.85 THz (N = 8500 for fr = 100 MHz) based on recent successes producing these THz frequencies using ytterbium-doped ber lasers [65]. We plot in gure 5.4 the e ect of spectral phase terms up to quadratic and cubic dependence on the optical frequency, corresponding to group ve- locity and third order dispersion are included. The e ect of fourth order dispersion was also calculated but found to have no signi cant e ect on the calculated signal amplitude. Figure 5.4 shows that the signal falls o rela- tively quickly as the pulse is temporally broadened due to GVD. Indeed, the point at which the signal falls to half its expected peak, for a total group delay of â€™ 1:5 104 fs2, corresponds to an optical pulse duration of 170 fs2, â€™ 10% longer than the transform limit of 160 fs. It is therefore necessary to compensate for the e ects of GVD in the local oscillator eld so as to maintain the heterodyne beat signal predicted by equation 5.23. Among others, Treacy et al [37] showed that the second order spectral phase shift due to GVD could be compensated for using a pair of di raction gratings, as described in section 2.4.2. A second grating compressor, similar to the design of that used to compress the optical pulse used to generate the ter- ahertz radiation, was therefore used to compensate for the e ects of GVD 1375.3. E ect of Spectral Phase on Heterodyne Beat Signal in the local oscillator pulse. Utilizing such an arrangement, it is possible to minimise the signal reduction as a result of GVD in the LO pulse, and it is anticipated that, providing careful attention is paid to compensation of GVD, TOD will have the greater e ect on signal degradation. In contrast, the e ect of a spectral phase pro le that is cubic in optical frequency, as for third order dispersion (TOD) is observed to have a more modest e ect. The signal remains within 50 % of the expected maximum (obtained for the initially transform limited pulse duration of 160 fs) for total third order delays of up to 7 106 fs3, or pulse durations up to 500 fs, in the presence of TOD alone. Typical values of the third order disper- sion coe cient, 3 for single mode bers at 1030 nm are â€™0.05 106 fs3/m, and hence the LO pulse would have to propagate in an unreasonably large (â€™100 m) amount of such bre for the heterodyne beat signal to be signi - cantly degraded due to the e ects of TOD alone. That TOD has little e ect on the calculated signal amplitude may be expected due to the asymmet- rical spectral phase pro le that results from TOD, giving a atter spectral phase pro le at the central optical frequency than is observed for the GVD case. It is worth noting that techniques have been demonstrated that al- low independent compensation of GVD and TOD [39], but the e ciency of these methods is such that the power in the LO pulse is reduced, and overall signal level will decrease. To date, such methods have therefore not been employed in the experimental work discussed here. In addition to the contribution to the spectral phase due to material and waveguide dispersion, there will be a nonlinear phase shift, owing to the intensity dependence of the material refractive index and the short optical pulses used. As a result of such an optical Kerr e ect, it is appropriate to write the refractive index in fused silica bre as n = n0 + n2I (5.28) where I is the intensity of the beam, n0 is the material refractive index in the limit of in nitesimal low intensities, and n2 is the optical Kerr coe cient in units of m2/W. As a result, the temporal form of the pulse is distorted 1385.4. E ect of Phase-matching on the Heterodyne Beat Signal according to E(z; t) = E0e i((n+n2 c" 2 jE0j2)kz !t) : (5.29) Wise et al. [63] showed that, for typical bre lasers of the design used here, a nonlinear phase shift of the order of leads to signi cant spectral dis- tortion of the optical pulse, in the form of increased energy in the spectral wings. The resulting non-uniform spectral phase pro le signi cantly limits oneâ€™s ability to re-compress the pulse using a grating compressor. That we do not observe such spectral distortion serves to verify that an upper bound of the peak nonlinear phase shift of is appropriate for this discussion. In order to calculate the e ect of such a nonlinear phase shift on the signal magnitude predicted by equation 5.23, a phase shift proportional to the instantaneous intensity of the Sech2 pulse envelope is applied in the time domain, before Fourier transforming back to the Fourier domain to calculate the new spectral phase of each frequency element. Upon substituting the resulting spectral phase pro le into equation 5.23 we see that the hetero- dyne beat signal remains within 50% of the predicted peak value for total nonlinear phase shifts of â€™ . Such a nonlinear phase shift would result in optical pulses many times longer in duration than the predicted transform limit, [91] and may be avoided using standard techniques of chirped pulse ampli cation [92]. We therefore do not expect a signi cant reduction in the heterodyne beat signal resulting from the nonlinear phase shift of the optical pulse. 5.4 E ect of Phase-matching on the Heterodyne Beat Signal In addition to a variation in signal strength with the spectral phase of the LO comb, equation 5.23 shows that there is a dependence of the signal strength on the wavevector mismatch of the terahertz comb line and the optical comb line with which it mixes to produce the A-EOS sideband. In order to study the e ect of the phase-matching term of equation 5.23, we 1395.4. E ect of Phase-matching on the Heterodyne Beat Signal 0 2 4 6 8 10 x 104 0 0.2 0.4 0.6 0.8 1 Normalised Ph otoc urr en t GDD (fs2) 0 2 4 6 8 10 Peak Nonlinear Phase Shift (Rads) 0 2 4 6 8 10 x 106TOD (fs3) GDD NLP TOD Figure 5.4: E ect of various spectral phase contributions on the expected signal magnitude predicted by equation 5.23, for the local oscillator de- scribed in section 2.4. the input spectral bandwidth is 6.9 nm at a central wavelength of 1029 nm, giving a transform limited sech2 pulse duration of 160 fs. 1405.4. E ect of Phase-matching on the Heterodyne Beat Signal assume the spectral phase of the LO FFC to be at, and compute the dependence on the wavevector mismatch, k+ and k of the total SFG and DFG contributions respectively. Figure 5.5 shows the spectral phase and amplitudes of the polarization sidebands written onto the LO FFC, as a function of the optical wavelength. The nonlinear crystal used for detection is ZnTe, with complex refractive index data taken from [93]. For frequencies greater (shorter wavelengths) than the central frequency of the LO FFC, the A-EOS sidebands are produced mainly via the SFG process, due to the larger amplitude of the LO comb elements involved. Likewise, at frequencies lower (longer wavelengths) than the central fre- quency of the LO FFC, the A-EOS sidebands are dominated by the DFG process. For this reason, the spectral phase of the A-EOS sidebands follows that calculated via the SFG term of equation 5.23 at shorter wavelengths, and that of the DFG term at longer wavelengths. For a ZnTe crystal of length 0.4 mm, it is seen that the phases of the LO sidebands vary by . 2 across the optical bandwidth of the LO FFC. The rf photocurrents produced by the beating of each LO sideband with its nearest-neighbor LO FFC element therefore also display minimal variation in phase, giving no degradation of either the SFG or DFG contributions to the rf signal . Further, note that the average phases of the SFG and DFG sidebands di er by less than 2 such that the contributions to the heterodyne beat from the SFG and DFG sidebands add in phase. Under these conditions, the phase-matching is therefore optimal. In contrast to gure 5.5, extending the nonlinear crystal to 0.8 mm, as shown in gure 5.6, produces an increased di erence in the average spectral phase of the SFG and DFG A-EOS sidebands. This is due to the increased phase contribution of the wavevector mismatch term in equation 5.23, giving a greater phase di erence between the SFG and DFG components. The LO sidebands therefore transition in phase from that of the SFG terms at short wavelengths, to that of the DFG terms at longer wavelengths. When the contribution of the entire LO spectrum to the rf heterodyne beat is calculated, the opposing phases of these sidebands gives partial cancellation of the rf photocurrent at N fr fr . 1415.4. E ect of Phase-matching on the Heterodyne Beat Signal total SFG DFG Contribution to Heterodyne Beat LO Spectral Profile Wavelength (nm) No rm alis ed Fi eld Am pli tud e Ph as e/Ï€ 0 0.5 1 1015 1020 1025 1030 1035 1040 1045 1 0 -1 Figure 5.5: Variation of phase (top) and amplitude (bottom) of A-EOS sidebands as a function of optical wavelength for a ZnTe detection crystal of 0.4 mm length. The spectral phase is shown for the sidebands produced through the SFG and DFG processes, as well as the sum of both, weighted by the amplitude contribution predicted by equation 5.23. total SFG DFG Contribution to Heterodyne Beat LO Spectral Profile Wavelength (nm) No rm alis ed Fi eld Am pli tud e 1015 1020 1025 1030 1035 1040 1045 Ph as e/Ï€ 1 0 -1 0 0.5 1 Figure 5.6: Variation of phase (top) and amplitude (bottom) of A-EOS sidebands as a function of optical wavelength for a ZnTe detection crystal of 0.8 mm length. Compared to that of gure 5.5, the di erence in phase between the SFG and DFG A-EOS sidebands at similar wavelengths leads to partial cancellation of the heterodyne beat signal. The amplitude of the contribution to the heterodyne beat is normalised with respect to that in gure 5.5. 1425.4. E ect of Phase-matching on the Heterodyne Beat Signal 1015 1020 1025 1030 1035 1040 1045 Wavelength (nm) No rm alis ed Fi eld Am pli tud e 1 0 -1 total SFG DFG Contribution to Heterodyne Beat LO Spectral Profile 0 Ph as e/Ï€ 0.5 1 Figure 5.7: Variation of phase (top) and amplitude (bottom) of A-EOS sidebands as a function of optical wavelength for a ZnTe detection crystal of 1.6 mm length. With the longer crystal, the A-EOS sidebands themselves are reduced in amplitude due to the wavevector mismatch between the optical LO and THz comb elements. Again, the amplitude of the contribution to the heterodyne beat is normalised with respect to that in gure 5.5. Further increasing the crystal length varies the expected signal in a peri- odic manner similar to that observed in other nonlinear optical processes, as shown in gure 5.8. We see, for example, that a crystal length of 0.4 mm is the optimum choice for the experimental conditions shown, whilst a crystal length of 0.8 mm produces SFG and DFG terms that are nearly opposite in phase, giving near total cancellation of the heterodyne beat signal, as illustrated in gure 5.6. It should be noted that, for the SFG and DFG pro- cesses individually, one calculates a coherence length of 1.6 mm in ZnTe for these frequency conditions, due to dephasing of the optical and THz elds, as shown in gure 5.7. It is the dephasing of the SFG and DFG components with respect to each other that reduces the e ective coherence length for the heterodyne beating process responsible for the mapping of the THz FC to the rf frequency range. 1435.5. Bandwidth and Choice of Detuning 0 0.5 1 1.5 2 2.5 30 0.2 0.4 0.6 0.8 1 Crystal Length (mm) Re la tiv e Si gn al L ev el Figure 5.8: Calculated heterodyne beat signal for a THz frequency of 0.85 THz, as a function of nonlinear crystal length for a ZnTe detection crystal, and an LO FFC of 7 nm bandwidth, centred at 1030 nm. 5.5 Bandwidth and Choice of Detuning As mentioned previously, in order that the bands of rf heterodyne beats do not overlap, it is necessary that N frep frep 2 . This places restrictions on the useful frequency bandwidth of the terahertz spectrometer, and raises consideration for laser design and data acquisition. For a given system design, the maximum terahertz frequency that can be measured is given by N 2 f2rep 2 frep : (5.30) Figure 5.9 shows the maximum terahertz frequency one can hope to resolve for realistic values of pulse repetition and detuning frequencies. The choice of repetition rate detuning ( frep) sets an upper limit on the resolution bandwidth required to resolve distinct spectral elements in the rf frequency range, which in turn sets a lower limit on the required acquisition time, tacq as tacq â€™ 1RBW . Again, gure 5.9 shows that tacq can vary from 0.1 to 250ms for reasonable performance parameters. A larger detuning allows for shorter acquisition times but limits the maximum frequency that can be measured, 1445.5. Bandwidth and Choice of Detuning Pulse Repetition Rate (MHz) Repetitio nR ate Detun ing (Hz ) 1 10 100 1000 100001 10 100 1000 10000 100000 10 1 0.1 0.01 0.001 0.0001 Approximat eAcquisiti on Tim e(s ) Figure 5.9: Map of the expected bandwidth performance one can reason- ably expect from the terahertz heterodyne spectrometer system for given constraints in oscillator design and acquisition methods. The maximum terahertz frequency one can hope to measure before the heterodyne beats overlap at frep2 is labelled in units of terahertz on the contour lines. Note that, for a laser pulse repetition rate of 100 MHz, and a reasonable detuning of 1 kHz, one can expect a terahertz bandwidth of >10 THz, with data acquisition times â€™ 10 ms. similar to the constraints of a conventional Fourier transform spectrum anal- yser. Conversely, smaller detunings allow greater terahertz bandwidths to be resolved, but data must be acquired over longer time intervals, and in- creased constraints are placed on the stability of the relative pulse repetition rates of the optical combs. From gure 5.9, it is seen that very respectable performance can be obtained for realistic design characteristics. For exam- ple, for a laser pulse repetition rate of 100 MHz, and a reasonable detuning of 1 kHz, one can expect a terahertz bandwidth of 10 THz, with data acquisition times â€™ 10ms. Further increases in the terahertz bandwidth can be obtained by going to higher repetition rate oscillators, as was noted in [83]. 1455.6. Signal to Noise Scaling 5.6 Signal to Noise Scaling One nal further aspect of this detection scheme is worthy of consideration. We note from equation 5.23 that the measured photocurrent is proportional to the square of the amplitude of the LO eld, in contrast to conventional heterodyne detection where the photocurrent scales linearly with the LO eld. This di erence arises due to the LO FFC being involved twice in the beating process. Once to generate the phase modulation sidebands onto the LO FFC, and once in a beating process with these modulation sidebands to produce the rf heterodyne beat, as shown in gure 5.3. The power associated with this signal photocurrent then scales as PS / hi 2 Si / PTHzP 2 LO; (5.31) where PTHz / A2THz, PLO / A 2 LO, and iS is the signal photocurrent. The shot noise current, hiN i associated with iS is described by hiN i = p 2eiS f; (5.32) where f is the bandwidth of the measurement, and e is the electronic charge. The signal to noise power ratio is then de ned as S N = hi2Si hi2N i / p PTHzPLO: (5.33) Whilst the increase in shot-noise-limited signal to noise ratio increases with the square root of the power in the terahertz eld, as in the case for both heterodyne detection of a weak optical eld, and direct detection of an optical eld, the increase in signal to noise ratio with power in the LO eld is perhaps less expected. In conventional heterodyne detection, where the signal photocurrent scales linearly with the LO oscillator electric eld, it is common to increase the LO power until one reaches the shot-noise limit, where the shot-noise power dominates above all other noise sources. At this point, further increases in LO power produce no bene t as the noise power amplitude increases with the signal amplitude. However, in the heterodyne 1465.6. Signal to Noise Scaling detection system here, the additional beating process means that, whilst the shot noise power level scales with the local oscillator power, the signal level will scale with the square of the local oscillator power, providing a potentially large increase in the signal to noise power ratio. 147Chapter 6 Experimental Demonstration of Dual-comb Terahertz Spectroscopy 6.1 Coupling of Detuned Local Oscillator Pulse In Chapter 4 we demonstrated a conventional, single-comb terahertz time- domain spectrometer (SC THz-TDS). We established the spectral content of the terahertz eld being generated in the gallium phosphide waveguide. The e ectiveness of the electro-optic sampling geometry is seen from the detection noise level of â€™ 20dB in the scans of gure 4.5. We now move on to a discussion of the experimental e orts to detect this eld using the dual-comb detection scheme outlined in Chapter 5. The optical probe beam of Chapter 4 was replaced with the pulsetrain of a second mode-locked ytterbium-doped- bre laser, henceforth the local oscillator comb (LO FFC). Theâ€™ 200 mW output of the laser in gure 2.10b) was compressed to 250 fs pulse duration using the technique described in section 2.4.2. The polarisation of this beam was then adjusted to match that of the sampling beam found to give the maximum signal magnitude in the conventional, single-comb, time domain spectroscopy case using a half- wave plate. The LO FFC was coupled into the experimental arrangement as shown in gure 6.1 using a switchable mirror, such that the existing beam alignment was unperturbed. In this way, the LO FFC was coaxial with the sampling pulse used in the single-comb THz-TDS experiment of section 4. Alignment of the LO FFC pulse with the previous sampling pulse 1486.2. Controlling The Laser Repetition Rates was veri ed using mechanical apertures, and a CCD camera placed at the location of the nonlinear detection crystal. The LO FFC beam was focussed to a spot size of 50 m with a Rayleigh range in excess of 5 mm, veri ed using the technique described in section 3.5.2, such that the optical spot size remained smaller than that of the terahertz spot size throughout the interaction length of the crystal. In this way, the entire optical pulse is in uenced by the change in refractive index due to the terahertz pulse. Due primarily to losses in the grating compressor and in the re ectivity of the pellicle beam splitter, the LO power incident on the nonlinear detection crystal was limited to 30 mW. 6.2 Controlling The Laser Repetition Rates 6.2.1 RF Locking Scheme In order to detect the heterodyne comb predicted in section 5, it is necessary that each rf comb line have a width far less than the spacing of the rf comb lines, so as to ensure the heterodyne comb itself is not obscured by the widths of the individual comb lines. As the rf comb is produced by mixing of the optical LO FFC and terahertz combs, it is necessary to stabilise the relative frequencies of the LO FFC and the optical comb used to produce the THz FC. In order to do this, the di erence in pulse repetition rate of the two laser oscillators of gure 2.10 is locked to a reference oscillator using the system of rf electronics shown in gure 6.2. It is important to note the fact that the heterodyne beat between the LO FFC and the THz FFC is produced by the mixing of many optical and terahertz comb lines, centred at â€™300 THz and 1 THz respectively. The respective combs are therefore centred at approximately the 3 106 and 1 104th harmonics of the â€™ 100 MHz repetition rate. As the rf heterodyne beat results from mixing of THz FC and LO FFC elements at â€™ 1 THz and 300 THz respectively, it is drifts in frequency between these comb elements that we wish to minimise in designing the locking scheme. Whilst locking of dual-combs at optical frequencies has been demonstrated [94], the tech- 1496.2. Controlling The Laser Repetition Rates FC1 GaP ZnT e QW P THz WP PTF E HW P HW P Balanced detector PeB S FC2 Figure 6.1: Experimental arrangement of the dual-comb terahertz spectrom- eter. In contrast to gure 4.1 the apparatus replaces the probe pulse with that from a second optical frequency comb (FC2), coupled into the experi- ment using a switchable mirror (SM) so as not to perturb the existing beam alignment. This second comb is detuned in repetition rate such that the delay between pulses from FC1 and FC2 varies in time. In this way the need to sweep the mechanical delay line is eliminated (the driving optical pulse still passes through the retro-re ector used to create the delay for reasons of alignment). As in gure 4.1, the sampling pulse is coupled into the detection crystal using a pellicle beam splitter (PeBS). Waveplates are used to optimise the polarisation states of the various beams so as to utilise the largest nonlinear coe cients, and a piece of PTFE is used to block the residual optical beam after the generating crystal. 1506.2. Controlling The Laser Repetition Rates Figure 6.2: Schematic of the rf locking scheme used to control the pulse repetition rates of the two lasers. Optical paths (in bre) in red, electrical paths in black. A 10% portion of each laser output is picked o using a bre splitter. A bre coupled fast (â€™10 GHz) photodiode is used to detect the pulse train from each laser. Each resulting photocurrent signal is rst ltered (BPF) at the nth harmonic of the nominal repetition rate of 104 MHz and ampli ed prior to combining the two signals in a double balanced mixer. The resulting mixer output is subsequently mixed with a reference oscillator (Osc) at n . Harmonics of the mixer output are removed using a third tuneable BPF before being fed to a variable loop lter (VLF). The VLF is essentially a two-pole device enabling tuning of the low frequency gain as well as the poles of each lter stage. The output of the lter is ampli ed and used to control the intracavity piezoelectric actuator in the laser used as the local oscillator, or probe laser. A signal (RF ref) at frep is provided for reference and triggering purposes. Similarly, a monitor of the residual noise on the error signal used to control the intracavity PZT is provided prior to the BPF and VLF stages for the purpose of system characterisation. 1516.2. Controlling The Laser Repetition Rates nique requires additional reference cavities and single mode lasers, greatly increasing the experimental complexity. Instead, it is common practice to lock the pulse repetition rates of lasers of this type by detecting harmonics of the repetition rate at more readily accessible rf frequencies. In the work presented here, we lock at the 37th harmonic of the nominal pulse repetition rate, at a frequency of â€™3.7 GHz. Whilst we do not ex- pect comparable performance to the optical locking techniques cited earlier, locking at a higher order harmonic of the repetition rate serves to reduce the broadening of the optical comb lines of concern due to residual jitter at the lock frequency. To understand this, consider the case where one locks the di erence in fundamental of the laser pulse repetition rates, frep, such that the relative broadening of the repetition rate di erence, the detuning, is . In this case, the Nth harmonic is broadened by an amount N . In contrast, if the lock is performed at some higher harmonic of the repetition rate, at nfrep, say, then the di erence frequency of the beat between the nth harmonics is broadened by , and the optical frequency at Nfrep is broad- ened by an amount Nn . It is therefore desirable to lock at the highest harmonic, n, of the repetition rate possible. The previously cited work of [94] bene ts from locking at optical frequencies, where n 3 106. For the purposes of this thesis, the two laser combs were locked at the more readily accessible rf frequency of the 37th harmonic of the laser pulse repetition rate at 3.7 GHz. That is to say that n =37 in gure 6.2. E orts to optimise the locking scheme, so as to reduce the expected broadening of the rf hetero- dyne beat as well as determine the extent to which the performance of our locking scheme is expected to broaden the heterodyne comb, are examined in section 6.3. 6.2.2 Component Testing and Characterisation For the purposes of the following discussion, we refer to the rst mixing stage, and any preceding components as â€˜high frequencyâ€™ rf, operating at nfrep â€™ 3.7 GHz, and the following components as â€˜low frequencyâ€™ rf, op- erating at n kHz frequencies. In order to reduce broadening of the 1526.2. Controlling The Laser Repetition Rates Figure 6.3: Arrangement for measuring of the noise oor of the available low frequency mixers. The frequency, f of an oscillator is varied such that the two signals combine at the mixer in quadrature for the (non-variable) delay, giving zero dc output. The mixer output is then observed on a spectrum analyser. heterodyne beat frequencies due to lock instabilities, it is desirable that the noise oor of each component contributes as little as possible to the repe- tition rate detuning of the two combs. To this end, the noise oor of three available low frequency mixers was measured using the setup shown in gure 6.3, with the results shown in gure 6.4. It is seen that the Anzac models display both a lower noise oor, and reduced susceptibility to line harmon- ics, especially at higher frequencies. Where possible then, these mixers were favoured over the Maccom model also tested. A similar study was performed for the available high frequency mixers. Due to the di culty in providing a reference signal at the necessary 3.7 GHz, the locking electronics were assembled and tested as shown in gure 6.2 with the various mixers in place. The residual noise on the control signal used to drive the piezo-electric actuator in the laser cavity was then measured prior to the band-pass ltering stage, as indicated in gure 6.2. The residual noise as a function of frequency for the various mixers is shown in gure 6.5. Note that this con guration actually compares the 37th harmonic of the repetition rate di erence to the oscillator frequency, which is set to the 37th harmonic of the desired repetition rate di erence. From gure 6.5, it is seen that the servo loop is e ective at reducing noise on the control 1536.2. Controlling The Laser Repetition Rates 100 101 102 103 104 105âˆ’150 âˆ’130 âˆ’110 âˆ’90 âˆ’70 âˆ’50 Frequency (Hz) Pow er Sp ect ral De nsit y(d Bm /Hz) Anzac D525 Maccom M2BC Anzac MDC 123 Figure 6.4: Measured noise oors for the three available low frequency mix- ers. Strong line harmonics remain, despite signi cant e orts to reduce these. 1546.2. Controlling The Laser Repetition Rates 100 101 102 103 104âˆ’140 âˆ’120 âˆ’100 âˆ’80 âˆ’60 âˆ’40 âˆ’20 Frequency (Hz) Unlocked Noise Level Anaren 70340 Watkins Johnson M1G Watkins Johnson M1H Anaren with Modified Power Supply Pow er Sp ect ral De nsit y(d Bm /Hz) Figure 6.5: Spectrum of the residual noise on the error signal for the range of high frequency mixers available, together with a reference trace showing the noise level in the absence of any precise locking scheme. For this reference trace, the dc bias to the oscillator cavity PZT was manually controlled so as to maintain the di erence in laser repetition rates in â€˜closeâ€™ lock to the oscillator frequency, but no servo gain was employed. Data is also shown for the Anaren mixer following work to isolate ground loops and provide cleaner power supplies for the servo electronics and rf ampli ers. In general, the servo loop is seen to reduce the level of noise on the control signal for frequencies out to around 3 kHz, where the phase response of the servo loop is such that further increases in gain write on additional noise to the control signal, and the residual noise rises above that in the unlocked case. 1556.3. Cross Correlation of Laser Pulses to Verify Timing Jitter signal out to frequencies around 3 kHz, where the phase response of the loop is such that increased loop gain writes additional noise onto the control signal. This results in the characteristic "servo bump" [95] at 3 kHz. The Anaren mixer shows the lowest residual noise oor. Further, the control signal displays residual line harmonics at 60 Hz and 180 Hz, regardless of the mixer used. Considerable e ort was therefore put into modifying the power supply circuitry for both the servo electronics, and for the rf ampli ers used in the servo loop. In particular, all required supply voltages were derived from linear voltage regulators, and ground loops removed wherever possible. Figure 6.5 shows the results of this work for the Anaren mixer, with the reduction in 60 Hz noise clearly visible. 6.3 Cross Correlation of Laser Pulses to Verify Timing Jitter Having taken steps to optimise the locking used to control the pulse rep- etition rate o set of the two laser oscillators, we now seek to verify the performance of this locking scheme by performing a measurement derived from the optical output of the lasers, independent of the control circuitry under study. This was done using a method similar to that documented in [86]. A cross-correlation measurement of the two laser pulses was acquired using non-collinear sum frequency generation in a -Barium Borate (BBO) crystal as per the experimental arrangement shown in gure 6.7. The rep- etition rate di erence of the two lasers is locked at frep. This causes the delay between the pulses, (t), to vary in time as shown in gure 6.6. If the two lasers have pulse repetition rates frep;1 and frep;2 = frep;1 + frep, and the delay between pulses is zero at time t = 0, then the delay can be written 1566.3. Cross Correlation of Laser Pulses to Verify Timing Jitter Figure 6.6: Pulse trains from two lasers, with pulse repetition rates frep;1 and frep;2 = frep;1+ frep. The delay (t) between pulses varies as (t) = t frep frep where t = 0 corresponds to the time at which the pulses overlap perfectly. as (t) = tfrep;1 1 frep;1 1 frep;2 = t frep frep;1 + frep â€™ t frep frep (6.1) Any uctuations in the pulse repetition rate di erence of the two lasers will results in uctuations on the pulse delay, . In order to measure the uctuations in , we provide an rf trigger at the average repetition rate di erence by virtue of the port labelled â€˜RF refâ€™ in gure 6.2 in order to trigger a 500 MHz digital storage oscilloscope. Due to the residual jitter in , averaging over many traces appears to broaden the measured pulse. We therefore average over a su ciently long time period such that the averaged signal does not appear to broaden any further, at which point the duration 1576.3. Cross Correlation of Laser Pulses to Verify Timing Jitter Figure 6.7: Experimental arrangement for the cross-correlation measure- ment. The two optical beams are focussed into a -BBO crystal using an 80 mm focal length lens. The generated noncolinear sum frequency radia- tion is focused through an aperture onto a silicon avalanche photodiode (Si APD). In order to reduce the residual optical power reaching the photodiode an interference lter was used, angle tuned to the wavelength of the second harmonic light at 515 nm. Phase-matching in the BBO crystal is type I angle-tuned. of the pulse represents the overall time resolution of this setup. A typical trace where the repetition rate detuning is locked to 400 Hz is shown in gure 6.8. The data shown is an average of 128 measurements over an acquisition time of â€™ 10 ms, limited by the averaging function of the sampling oscilloscope. The observed signal has the form of a 100 MHz pulse train (governed by the nominal repetition rates of the lasers) modu- lated by an envelope with pulse full width half-maximum duration â€™100 ns and modulation frequency determined by the detuning, frep = 400 Hz. From the time axis, one calculates the delay between the two optical pulses according to (t) = t frep frep , from which one obtains a width of the aver- aged cross-correlation signal of â€™400 fs. The width of this envelope did not increase with increased averaging time, thus the residual jitter in the time base provided by this locking scheme is 400 fs. Such an envelope decon- volves to an actual sech2 pulse width of 260 fs, which is comparable to the original optical pulse width, indicating minimal residual timing jitter due 1586.3. Cross Correlation of Laser Pulses to Verify Timing Jitter âˆ’600 âˆ’400 âˆ’200 0 200 400 600 800 1000 1200 1400 0 0.2 0.4 0.6 0.8 1 Delay Time (fs) Signa l âˆ’130 0 130 260 390 âˆ’130 0 130 260 390 âˆ’130 Real Time (ns) Figure 6.8: Blue: Cross-correlation trace of the femtosecond optical combs in -BBO with the repetition rate detuning set to 400 Hz. Time axes are shown both for the real time (lower) and scaled by a factor frep frep to give the delay time. Black: Overlay of the sech2 envelope corresponding to the cross-correlation signal expected from pulse widths of 260 fs. to the phase noise of the electronic components used. Further veri cation of the lock performance was performed by reducing the repetition rate de- tuning to 15 Hz, and analysing the resulting frequency modulations in the cross-correlation signal using an rf spectrum analyser. The results are shown in gure 6.9. It is seen that the modulation of the cross-correlation signal remains resolvable for even such small detunings. Indeed, the broadening observed is comparable to that expected due to the limited 2 Hz resolution bandwidth of the instrument. The conclusion is therefore drawn that the broadening of the heterodyne beat between the two lasers is not so severe so as to prevent detection of the individual comb lines, and the lock is deemed to be working satisfactorily. 1596.3. Cross Correlation of Laser Pulses to Verify Timing Jitter âˆ’200 âˆ’150 âˆ’100 âˆ’50 0 50 100 150 200 250âˆ’150 âˆ’130 âˆ’110 âˆ’90 âˆ’70 âˆ’50 Frequency Offset From Repetition Rate of 104.35MHz (Hz) (dBm ) Power Figure 6.9: RF spectral output for the cross-correlation measurement de- picted in gure 6.7. The detuning, frep has been reduced to 15 Hz, whilst the resolution bandwidth of the instrument is set to 2 Hz. The increased sidebands at â€™ 180 Hz due to instabilities in the output of the diode used to pump the power ampli er stage remain visible, as discussed in section 2.4.3. 1606.4. Schemes for Detection of the Heterodyne Beat 6.4 Schemes for Detection of the Heterodyne Beat 6.4.1 High Gain Balanced Detector With High Extinction of Frequencies Greater than 80 MHz The rst attempt to detect the comb of heterodyne beats between the de- tuned terahertz comb and the optical local oscillator was performed using a custom designed balanced detector, giving a gain at 100 kHz of 8 V/mA, and a high frequency roll-o of â€™40 dB/8ve at 80 MHz in order to attenuate the large repetition rate signal at 104 MHz. In addition, a high pass lter of 50 dB/decade at 1 kHz was used to eliminate any dc component arising from slight errors in setting the balancing of the polarisation channels. A schematic of the detector circuit is given in gure A.1 in appendix A. The response of the detector to a balanced input is shown in gure 6.10. For a terahertz comb and LO comb with repetition rate detuning frep one expects the terahertz frequency at N(frep + frep) to be mapped to an rf photocurrent signal of frequency N frep, as derived in section 5. For a terahertz comb centred at 0.75 THz, as in gure 4.5a, N = fTHzfrep â€™ 7500, so, for a typical detuning of 1 kHz, one expects the particular heterodyne beat due to the terahertz frequency at line centre to be at 7.5 MHz. It should, however, be noted that each comb element of the terahertz comb creates its own heterodyne beat, and that therefore there is, as explained in section 5, a comb of rf frequencies separated by frep. This comb extends from dc up to the bandwidth of the terahertz pulse, scaled by a factor frepfrep . The balanced detector, in conjunction with additional external ltering if necessary, therefore provides large gain of the frequencies at which we expect to observe the comb of heterodyne beats, whilst simultaneously reducing the rf power at 100 MHz, allowing the gain to be further increased without saturating the power spectrum. The optical power giving shot-noise limited performance of this detector can be determined from the noise spectra of gure 6.11, which shows the unampli ed detector response function for lower frequencies than that of 1616.4. Schemes for Detection of the Heterodyne Beat 0 20 40 60 80 100 120âˆ’50 âˆ’40 âˆ’30 âˆ’20 âˆ’10 0 10 Frequency (MHz) Powe rLeve l(dBm ) With additional 32MHz LPF Detector Output Figure 6.10: Output of the custom-designed balanced detector for a typical balanced input with additional 32dB external ampli cation. The band-edge is seen to attenuate the 104 MHz repetition rate signal such that the in-band noise- oor is only 25 dB below the 104 MHz peak. If necessary, additional ltering can be employed to further attenuate the repetition rate signal, as in the case shown where an additional 32 MHz low pass lter (LPF) has been used. gure 6.10. One observes a pass-band noise level of -140 dBm. Compare this to the expected shot noise on the dc photocurrent, iDC produced from the 0.3 W of optical power incident on a single channel. Using an on-board gain of 7.3 V/mA, the dc voltage drop across a 50 load was observed in this case to be 2V, implying a photocurrent of 0.27mA. The shot-noise power associated with such a photocurrent is given by hiN i 2R = 2eiDC ; (6.2) where iN is the noise current and R is the load. For this case, one obtains 1626.4. Schemes for Detection of the Heterodyne Beat a shot-noise power of 144dBmHz , in good agreement with the noise oor of gure 6.11. Further, increasing the optical power incident on the detector produced a corresponding increase in the observed noise oor for frequen- cies within the pass-band. 0.3 W is a readily available optical power from the local oscillator, and therefore it is reasonable to expect shot-noise lim- ited performance from this detector. Despite the considerable gain in the 100 101 102 103 104 105âˆ’180 âˆ’170 âˆ’160 âˆ’150 âˆ’140 âˆ’130 âˆ’120 âˆ’110 Frequency (Hz) Pow er Sp ect ral De nsi ty (dB m/ Hz) Figure 6.11: Noise spectrum for the high-speed balanced detector used for detection of the heterodyne beat signal. The incident optical power on a single channel in this case was 0.3 W. The observed noise- oor in the pass-band of -140 dBm corresponds well to the expected shot noise on the observed dc photocurrent of 0.274mA. Note that, also visible, is the roll-o of the high pass lter at 1 kHz. frequency range expected to contain the comb of heterodyne beat signals, e orts to resolve the predicted rf heterodyne comb at low frequencies us- ing the bandwidth-limited balanced detector were unsuccessful. Whilst the measurement was indeed shot-noise limited, equation 5.33 suggests that, for 1636.4. Schemes for Detection of the Heterodyne Beat this detection scheme, increases in the signal to noise ratio can be expected by increasing the power in the LO eld beyond that of the shot-noise-limited case. Unfortunately, further increases in LO power were not possible without adding additional high power ampli cation stages to the LO laser output, for which resources were not available. In light of this, an alternative de- tection scheme employing phase-sensitive detection of a single heterodyne comb element was employed. This forms the topic of the following section. 6.4.2 RF Heterodyne Detection Using VCO Reference With the bandwidth limited balanced detector proving unsuccessful, an at- tempt was made to increase the signal to noise ratio of the expected rf heterodyne comb by employing a form of rf phase-sensitive detection. To do this, a voltage-controlled reference oscillator was locked to the Nth har- monic of the repetition rate detuning using a servo loop in addition to that used to control the repetition rate detuning itself, as shown in gure 6.12. This oscillator signal was then used as the LO input to a double-balanced mixer, with the photodiode signal providing the rf input. In this way, a particular comb element of the rf heterodyne comb described by equation 5.23 is demodulated down to dc frequency. We choose this particular comb element to be near the centre of the predicted rf heterodyne comb so as to maximise the expected signal level. Recall the rf signal corresponding to a THz FC element at frequency N varies in time as S N (t) / ATHz( N ) exp( 2 iN frept); (6.3) where ATHz( N ) is the complex spectral amplitude of the THz FC element at frequency N . Using the experimental arrangement of gure 6.12, we provide a reference signal at frequency N frep = N frep . By mixing this ref- erence signal with the measured photocurrent in a double balanced rf mixer (DBM) the heterodyne beat corresponding to a terahertz comb element N at frequency N is mapped to dc frequency. Similarly, the heterodyne comb elements adjacent in frequency, corresponding to terahertz comb elements 1646.4. Schemes for Detection of the Heterodyne Beat at N+R = (N +R)frep; (6.4) where R is the order of the comb element with respect to the N th element at frequency N , are mapped to rf frequencies R frep frep . In order to observe the strongest heterodyne beats, we seek to set N = Nfrep to the centre of the terahertz spectrum shown in gure 4.5. We then expect to observe a dc signal corresponding to the heterodyne beat at N frep, and a set of rf comb elements spaced by the repetition rate detuning, frep arising from the terahertz comb lines at N+R. A voltage controlled oscillator (VCO) was used to provide the necessary reference frequency. Component availability stipulated that the output frequency of the VCO, fV CO was nominally 80 MHz, with a tuning range of 10 MHz. A frequency pre-scaler (UPB1509GV) was therefore used to divide this by four to obtain a reference at â€™18 MHz. A further pre-scaler was used to divide the frequency of a portion of this output by a factor of 256, such that a second double-balanced mixer could be used to generate a VCO control signal from comparison with the signal at 37 frep, picked o from the servo loop used to control the repetition rate detuning. This required the repetition rate detuning, frep, be set to 1.9 kHz. With this setup, the heterodyne beat mapped to dc corresponds to a terahertz frequency of N = 18MHz frep frep = 0:9THz: (6.5) From gure 4.5a, it is seen that this is less than 50% down on the peak of the terahertz emission spectrum produced from the gallium phosphide waveguide. Limited component availability (and, crucially, the requirement that the pre-scalers used can only divide by a power of two) means that it was not possible to get closer to the terahertz band-centre. In order to control the VCO frequency, an active lter was built con- sisting of an integrator and low-pass lter stage, a schematic of which is shown in gure 6.13. The poles of the lter were varied to give optimum performance, as determined by gure 6.14. 1656.4. Schemes for Detection of the Heterodyne Beat + _ Figure 6.12: Schematic of the rf circuitry used to provide a reference signal at the expected rf heterodyne beat frequency. The output frequency of a voltage-controlled oscillator (VCO) is divided down by a factor of 4 to provide the required reference signal at 18 MHz. A small portion of this 18 MHz signal was then further divided by a factor of 256 so as to enable comparison to a portion of the repetition rate detuning signal at n frep (â€™ 70:3 kHz for a detuning of 1.9 kHz). The control signal of the VCO, VCC in, is derived from the IF output of the mixer, using a servo loop, similar to that used to control frep, but using a purpose-built active lter (PID) in place of the loop lter used in the detuning locking case. A manually adjustable bias voltage, Vbias is provided for tuning of the VCO frequency, and is added to the control signal using a conventional adder circuit. The VCO output is low pass ltered to remove harmonics and ampli ed to provide the necessary rf power to be used as the reference in a mixing stage with the optical signal. Measurements were made using both a single fast photodiode con gured as per the single-channel detection scheme in 4.3, or from the bandwidth- limited balanced photodiode described in section 6.4.1, con gured as per the balanced detection scheme of section 4.3. 1666.4. Schemes for Detection of the Heterodyne Beat + _ + _ + _ Figure 6.13: Circuit schematic for the active lter used to lock the voltage control oscillator (VCO) to the expected frequency of the heterodyne beat signal predicted by the analysis of Chapter 5. The component values shown were decided upon following analysis of the lock performance, as shown in gure 6.14. As shown, the lter consists of an initial integrator stage, with a pole at approximately 20 kHz, and a second low-pass lter with a pole at 315 Hz. A potentiometer is provided at the nal output for varying the overall response. Figure 6.14 shows the residual noise on the VCO control signal as a func- tion of frequency. Prior to considering the performance of the feedback loop in locking the VCO frequency to that of the laser repetition rate detuning, the ability of the servo loop to lock the VCO frequency to a xed refer- ence oscillator (Hewlet-Packard model no. 8662A) is characterised by a at noise oor out to the resonance of the integrator stage at â€™ 20 kHz. Whilst it was possible to change the con guration of the lter in gure 6.13 such that the calculated pole of the integrator stage was shifted to â€™ 80 kHz, such a broad bandwidth resulted in an unstable lock of the VCO to the reference oscillator, indicating a 180 phase roll-o in the response of other loop components. It should be noted here that identical VCOs are locked to reference frequencies with loop bandwidths approaching 100 kHz in other experiments in our lab. The limitations in bandwidth therefore are not at- tributed to the VCO itself, but other components in the loop highlighted in green in gure 6.12. In the absence of alternative loop components, the 1676.4. Schemes for Detection of the Heterodyne Beat 100 101 102 103 104 105âˆ’130 âˆ’120 âˆ’110 âˆ’100 âˆ’90 âˆ’80 âˆ’70 âˆ’60 Frequency (Hz) Pow er Sp ect ral De nsit y(d Bm /Hz) Noise on Detuning Lock Noise on VCO Lock Lock to Reference Figure 6.14: Measured residual noise level on the VCO control signal for locking of the VCO to the detuning (in red) and a reference oscillator (in black). The measured noise oor on the laser detuning lock (in blue) is shown for comparison. 1686.5. Results and Conclusions from Dual-comb Spectroscopy Measurement pole of the integrator stage was reduced in frequency so as to reduce the lter gain beyond 20 kHz, rather than thoroughly determine the root cause of this limited locking bandwidth. Figure 6.14 shows that, when tracking the residual drift in the repetition rate detuning, frep, the loop is able to reduce the residual noise on the VCO control signal to that observed when locking to a reference oscillator for frequencies up to â€™3 kHz, but that there is an increase in the residual noise on the VCO control signal at â€™6 kHz, where the resonance of the detuning control loop, as previously observed in gure 6.5, writes additional noise onto the VCO control signal. However, below this resonance, the residual noise of the VCO lock is well below that of the locking signal used to control the repetition rate detuning, indicating that the VCO servo loop is able to tune the VCO frequency su ciently fast to â€˜trackâ€™ residual drifts in the detuning out to a bandwidth of â€™ 6 kHz. Attempts to detect the expected rf heterodyne beat with the the de- scribed VCO reference signal were made using both the bandwidth-limited balanced detector described previously, and a fast single-channel detector in conjunction with a half wave-plate and crossed polariser, as described in section 4.3. In the latter case, the optical power after the nonlinear de- tection crystal was attenuated to â€™1 W so as to avoid saturating the on board ampli er circuitry. Neither of these methods proved successful. We therefore dedicate the remainder of this chapter to a discussion of where the current experiment might be improved, and a comparison to other works where appropriate. 6.5 Results and Conclusions from Dual-comb Spectroscopy Measurement No comb elements were visible in the measured rf spectrum, despite addi- tional ampli cation of the photodiode signal, such that the shot-noise limited noise oor of the output of the nal mixer stage was â€™-10 dBm, well above the total instrument noise level of â€™-130 dBm. The following discussion 1696.5. Results and Conclusions from Dual-comb Spectroscopy Measurement outlines some of the experimental limitations, and, in particular, compares the current work to that of other authors who have been able to observe a beating of the two detuned combs, albeit with di ering measurement and data acquisition schemes. From equation 5.23 one sees that the magnitude of the contribution to the rf photocurrent observed at the heterodyne beat frequency due to a single A EOS sideband scales linearly with the amplitude of the terahertz eld, ATHz / p PTHz, where PTHz is the total power of the terahertz eld. Recall in section 4.5, we compared our experimental arrangement to that of [66], giving an estimate of the terahertz power in our work of 2.5 W. Given the 0.5 THz terahertz spectral bandwidth shown in gure 4.5 and the comb element spacing of frep â€™ 104 MHz, one calculates a power per terahertz comb element of 0.3 nW for our apparatus. Using this estimate of the terahertz power per comb element, we compare the signal to noise performance obtained by two works in particular. That of Bartels et al. (particularly [86], but see also [96]) and that of Barbieri et al. [97]. In [86], a total terahertz power of 40 W is claimed from an electro-optic antenna, giving a power per terahertz comb line of â€™30 nW. These authors use a similar detuned comb arrangement, acquiring many time-domain data-traces, each of approximately 100 s duration. An average over â€™ 107 of these traces is then taken and Fourier transformed to allow the spectral content of the terahertz output to be be analysed. This acquisition time corresponds to a range in pulse delay of 1 ns, as predicted by equation 6.1. This limited delay time gives rise to the frequency resolution of the experiment of 1 GHz, which meant that the authors were unable to resolve the individual comb lines of the terahertz spectrum, being themselves spaced by 1 GHz. It is notable that such signi cant averaging was required, despite the higher power per terahertz comb line, due to residual instabilities in the locking scheme used to govern the repetition rate detuning of the two laser oscillators. From the measurements presented in section 6.3, we see that the residual timing jitter in our locking scheme is at least as good as that in [86] (the longer duration of our optical pulses prevents us from measuring any 1706.5. Results and Conclusions from Dual-comb Spectroscopy Measurement lower levels of timing jitter). However, we note that the larger repetition rate of [86] lessens the demand on the locking scheme, as the spacing of the rf heterodyne beats is increased. In the time domain, one can interpret this as an increase in sampling rate of the modulation due to the detuning of the two combs, which means that an average over a given number of traces is acquired over a shorter total acquisition time. For a total of 107 averages, Bartels et al. obtain a signal to noise ratio at line-centre of â€™50 dB. If one were to attempt a similar measurement with the 0.3 nW per terahertz comb line and 30 mW LO of our experiment, one would expect this to be reduced to 0 dB, highlighting the need for high-power combs with this technique. Barbieri et al [97] demonstrated a mode-locked quantum cascade laser (QCL), producing 17 mW of total terahertz power at 13 GHz repetition rate over a narrow spectral bandwidth of 32 GHz. In this case, the tera- hertz spectrum contains only a few comb lines, giving a considerable average power per comb line of â€™7 mW. In contrast to Bartels et al. these authors acquired data in the frequency domain using a swept rf spectrum analyser with a resolution bandwidth of 100 kHz. In the work of Barbieri et al. the bandwidth of the lock used to control the repetition rate of the QCL was only a few hundred Hz. Such large resolution bandwidths and comparatively poor locking performance does not, in this case, a ect the ability to resolve the individual terahertz comb lines due to the spacing of the latter by the large repetition rate. Barbieri et al. observe a signal to noise ratio of 20 dB for their experiment. Performing a similar comparison as that done for [86] above, one predicts a signal to noise ratio at terahertz line-centre of -55 dB for our power levels, due largely to the reduced terahertz power per comb line. In order to increase the performance of the experimental arrangement presented in this thesis, there exists four options. Firstly, to reduce the residual relative timing jitter between the two optical pulse trains. The most likely method for achieving this would be to stabilise the relative pulse repetition rates at optical frequencies rather than at GHz frequencies, as discussed in section 6.2.1, and demonstrated in [94]. This requires con- siderable additional experimental complexity in the form of a high nesse 1716.5. Results and Conclusions from Dual-comb Spectroscopy Measurement optical reference cavity and CW laser sources, and was beyond the scope and resources of this project. The second possible option is to increase the pulse repetition rate of the bre laser oscillators used, such that the comb line spacing is increased. Unfortunately, achieving pulse repetition rates greater than â€™200 MHz in bre lasers involves signi cant engineering challenges due to the reliance of the nonlinear polarisation evolution mod- elocking mechanism on the induced birefringence of the bre components [98]. The amount of polarisation rotation accumulated in a cavity roundtrip therefore decreases for higher repetition rate cavities, leading to instabilities in modelocking [99]. The third option for increasing the signal to noise ratio of the system presented here is to increase the power in the terahertz comb. The obvious way to achieve this is to increase the optical power with which the nonlinear generating crystal is pumped, requiring additional ampli ca- tion stages in the bre laser system, resources for which were not available. Finally, electro-optic antennas have demonstrated increased terahertz con- version e ciency at these powers. Indeed, Bartels et al. obtain their 40 W of terahertz power from only a 500 mW optical pump. The trade o is saturation of the conversion e ciency for higher optical driving elds, and reduced terahertz spectral bandwidth owing to the time constants of the carrier dynamics within the crystal. As a nal area for improvement of the work presented in this thesis, we note that the magnitude of each term contributing to the rf photocurrent in equation 5.23 scales with the amplitude of two optical LO comb elements, ALO(!M ) and ALO(!M+N ) for the SFG contribution and ALO(!M+2N+1) and A LO(!M+N+1) in the case of the DFG contribution. It was shown in section 5.6 that this gives rise to a scaling of the shot-noise-limited signal to noise ratio as / PLO, in contrast to conventional heterodyne detection where one observes no bene t in terms of signal to noise ratio beyond the shot-noise-limit. It is worth noting that both of the works discussed above used greater LO powers, 400 mW in the case of [86] and 50 mW in the case of [97], than the 30 mW available for this work. A nal useful calculation is to estimate the signal to noise ratio expected in the dual-comb detection scheme from the measured performance of the 1726.5. Results and Conclusions from Dual-comb Spectroscopy Measurement single-comb detection scheme. The peak signal in gure 4.5a represents a voltage swing of 500 V from the background-corrected mean value of zero. We assume the mechanical chopper to modulate the driving optical eld with the form of a square wave, and that therefore the terahertz eld also exhibits a square-wave modulation at the chopper frequency of fchop â€™ 1.5 kHz. Considering a square wave as the sum of its Fourier components, f(t) = 4 X n=1;3;5::: 1 n sin (n!t) ; (6.6) the 500 V root-mean-square (rms) signal level measured at the fundamental chopping frequency corresponds to a peak-to-peak level of the total square wave signal of 900 V. Further, the single-channel detector used in the acquisition of gure 4.5a had a power gain of 37.5 dB, and a transimpedance gain of 50 V/A, implying a photocurrent generated in the diode itself of approximately 340 nA. In the single-comb case, all of the rf heterodyne beat frequencies are mapped to an rf frequency of zero, and are therefore degenerate, by virtue of the zero value of frep. The resulting change in signal level is therefore due to the heterodyne beat frequencies from each of the â€™5000 THz rf comb elements summing at dc frequency. In the dual- comb case, where the non-zero detuning of the pulse repetition rates serves to lift the degeneracy and map each of the heterodyne beats to a unique rf frequency, each comb element is reduced in amplitude with respect to the single-comb case by, to a rst order approximation, a factor given by the number of THz frequency comb elements. Using the above numbers, the photocurrent produced by each rf heterodyne beat is calculated to be 69 nA. Accounting for the transimpedance gain of 8000 V/A of the balanced detector shown in Appendix A, and feeding this signal through an rf power ampli er with 32 dB of power gain, this gives a signal level of 550 nV, or a power into a 50 load of 9.5 pW, or -80 dBm. Signi cantly lower than the shot noise-limited noise oor of this balanced detector con guration, shown to be â€™-20 dBm in gure 6.10. Due to the linear scaling of the signal strength with PLO versus the scaling as p PTHz, the necessary 60 dB 1736.5. Results and Conclusions from Dual-comb Spectroscopy Measurement increase in signal strength is likely to be more readily obtained through increases in the local oscillator power. Indeed, achieving such an increase in signal strength through increases in terahertz comb powers alone would require an increase in PTHz of 12 orders of magnitude. Due to the large increases in signal required, it is highly likely that improvements will have to be made in all of the areas outlined above. That is, increases in both the local oscillator power and terahertz comb power, and a reduction in the relative timing jitter between the two laser pulse trains. Consider now e orts to detect the expected heterodyne comb by pro- viding an rf reference to an additional mixing stage, as described in sec- tion 6.4.2, where it was observed that the ability of the servo loop to track the laser repetition rate detuning was limited to a frequency bandwidth of â€™6 kHz, in contrast to loop bandwidths achieved elsewhere in our lab using the same design of voltage-controlled oscillator (VCO) and di erent loop architectures. Given su cient time and resources, the components limiting this bandwidth could be identi ed and replaced and/or redesigned, and it is anticipated that the locking bandwidth of the VCO control loop could be in- creased to â€™50 kHz. Note however that this detection scheme inherently has the drawback of introducing additional instrumentation noise in the form of the noise oors of the components used, and that therefore, direct detection of the heterodyne beat is likely to o er the preferred solution. A precise estimate of the bene t of these improvements, or indeed, an estimate of the necessary improvements required in order to attain a partic- ular signal to noise ratio, is a di cult task owing to the complexity of the physical processes involved. Indeed, e orts to calculate the absolute tera- hertz power from the optical recti cation process alone are rarely seen in scienti c literature. Calculating the magnitude of the rf photocurrent from the asynchronous electro-optic sampling process is even more problematic due to the lack of numerical estimates of the nonlinear susceptibility at ter- ahertz frequencies. In fact, it is possible that the technique presented here would allow such measurements in the future, with high spectral resolution. 174Chapter 7 Final Conclusions In this work, we have shown that signi cant technical challenges remain if a passive optical resonator is to be e ective in increasing the e ciency of the nonlinear conversion process involved in terahertz generation via optical recti cation. Namely, crystal materials must be developed that o er single pass conversion e ciencies comparable to those already available, but with- out the optical absorption observed for gallium phosphide. In the absence of such materials, high power terahertz frequency combs would therefore seem better served by conventional single-pass generation geometries. Such techniques may take better take advantage of the long coherence lengths in readily available nonlinear materials at wavelengths that are conveniently produced by now well-established, high power bre lasers. We have shown analytically that asynchronous electro-optic sampling can provide a high-resolution terahertz frequency comb spectrometer. Fur- ther, we have derived the equations governing the resultant heterodyne comb using a fully frequency-domain analysis. To date, our experimental e orts to detect the expected heterodyne comb have been unsuccessful. This is in spite of the di erent detection geometries, large signal gain at the fre- quencies of interest, e orts to reduce noise on commercial electronics used, and considerable e ort to stabilise the relative repetition rates of the two laser oscillators. The fact that, despite these measures, the heterodyne beat signal was still not observed only serves to highlight the need for high power frequency combs in this technique. However, in the course of this work other authors have detected similar heterodyne combs using higher power quan- tum cascade laser (QCL) terahertz sources. With the higher comb powers available with ever advancing bre laser technology, this technique will likely warrant future interest for the purposes of high resolution terahertz spec- 175Chapter 7. Final Conclusions troscopy, where the high cost and complex cooling of QCL terahertz sources cannot be justi ed. 176Bibliography [1] Y Hu, XH Wang, LT Guo, and CL. Zhang. Terahertz time-domain spectroscopic study of carbon monoxide. PubMed, 26(6):1008{1011, 2006. [2] D.M. Mittleman, R.H. Jacobsen, R. Neelamani, R.G. Baraniuk, and M.C. Nuss. Gas sensing using terahertz time-domain spec- troscopy. Applied Physics B: Lasers and Optics, 67:379{390, 1998. 10.1007/s003400050520. [3] Edward P J Parrott, J Axel Zeitler, Tomislav Fris, Michael Pepper, William Jones, Graeme M Day, and Lynn F Gladden. Testing the Sensitivity of Terahertz Spectroscopy to Changes in Molecular and Supramolecular Structure : A Study of Structurally Similar Cocrys- tals & DESIGN 2009. Crystal Growth & Design, 2009. [4] L. S. Rothman, I. E. Gordon, A. Barbe, Benner, P. F. Bernath, M. Birk, V. Boudon, L. R. Brown, A. Campargue, and J. P. Champion. The HI- TRAN 2008 molecular spectroscopic database. Journal of Quantitative Spectroscopy and Radiative Transfer, 110(9-10):533{572, June 2009. [5] P.H. Siegel. Terahertz technology. Microwave Theory and Techniques, IEEE Transactions on, 50(3):910 {928, mar 2002. [6] P.F. Taday, I.V. Bradley, D.D. Arnone, and M. Pepper. Using tera- hertz pulse spectroscopy to study the crystalline structure of a drug: A case study of the polymorphs of ranitidine hydrochloride. Journal of Pharmaceutical Sciences, 92(4):831{838, 2003. 177Bibliography [7] Je rey Barber, Daniel E Hooks, David J Funk, Richard D Averitt, Antoinette J Taylor, and Dmitri Babikov. Temperature-dependent far- infrared spectra of single crystals of high explosives using terahertz time-domain spectroscopy. The Journal of Physical Chemistry. A, 109(15):3501{5, April 2005. [8] Feng Huang, Brian Schulkin, Hakan Altan, John F. Federici, Dale Gary, Robert Barat, David Zimdars, Minghan Chen, and D. B. Tanner. Terahertz study of 1,3,5-trinitro-s-triazine by time-domain and Fourier transform infrared spectroscopy. Applied Physics Letters, 85(23):5535, 2004. [9] Matthias Ho mann, Bernd M. Fischer, and Peter U. Jepsen. Terahertz spectroscopy of rna and dna and spot array imaging. In Optical Tera- hertz Science and Technology, page WB2. Optical Society of America, 2005. [10] Paul A. George, Wallace Hui, Farhan Rana, Benjamin G. Hawkins, A. Ezekiel Smith, and Brian J. Kirby. Micro uidic devices for terahertz spectroscopy of biomolecules. Opt. Express, 16(3):1577{1582, Feb 2008. [11] P. Baron, J. Mendrok, Y. Kasai, S. Ochiai, T. Seta, K. Sagi, K. Suzuki, H. Sagawa, and Joachim Urban. AMATERASU: Model for atmospheric terahertz radiation analysis and simulation. Journal of the National Institute of Information and Communications Technology, 55:1, s. 109- 121, 2008. [12] Wai Lam Chan, Jason Deibel, and Daniel M Mittleman. Imaging with terahertz radiation. Reports on Progress in Physics, 70(8):1325, 2007. [13] Herbert M Pickett and Dean B Peterson. Far-ir fabry-perot spectrom- eter for OH measurements. Remote Sensing of Atmosphere Chemistry, 1715:451{456 610, 1991. [14] Yihong Yang, Alisha Shutler, and D. Grischkowsky. Measurement of the transmission of the atmosphere from 0.2 to 2 THz. Opt. Express, 19(9):8830{8838, Apr 2011. 178Bibliography [15] Benjamin S. Williams. Terahertz quantum-cascade lasers. Nature Pho- tonics, 1:517{525, September 2007. [16] B.S. Williams, S. Kumar, Q. Hu, and J.L. Reno. High-power terahertz quantum-cascade lasers. Electronics Letters, 42(2):89 { 91, jan. 2006. [17] Mikhail A. Belkin, Federico Capasso, Feng Xie, Alexey Belyanin, Milan Fischer, Andreas Wittmann, and J ero^me Faist. Room temperature terahertz quantum cascade laser source based on intracavity di erence- frequency generation. 92(20):201101, 2008. [18] D. H. Auston, K. P. Cheung, and P. R. Smith. Picosecond photocon- ducting hertzian dipoles. 45(3):284{286, 1984. [19] A. Bartels, A. Thoma, C. Janke, T. Dekorsy, A. Dreyhaupt, S. Winnerl, and M. Helm. High-resolution THz spectrometer with kHz scan rates. Opt. Express, 14(1):430{437, Jan 2006. [20] Georgi L. Dakovski, Brian Kubera, and Jie Shan. Localized terahertz generation via optical recti cation in ZnTe. Journal of the Optical Society of America B, 22:1667, 2005. [21] Ajay Nahata, Aniruddha S Weling, and Tony F Heinz. A wideband coherent terahertz spectroscopy system using optical recti cation and electro-optic sampling. October, 69(October):2321{2323, 1996. [22] Steven T Cundi . Phase stabilization of ultrashort optical pulses. Jour- nal of Physics D: Applied Physics, 35(8):R43, 2002. [23] Scott a Diddams, Leo Hollberg, and Vela Mbele. Molecular ngerprint- ing with the resolved modes of a femtosecond laser frequency comb. Nature, 445(7128):627{30, February 2007. [24] S Schiller. Spectrometry with frequency combs. Optics letters, 27(9):766{8, May 2002. [25] Julien Mandon, Guy Guelachvili, and Nathalie Picqu e. E cient Two- Comb Fourier Spectroscopy. pages 1{9, 2008. 179Bibliography [26] A C Vutha, W C Campbell, Y V Gurevich, N R Hutzler, M Parsons, D Patterson, E Petrik, B Spaun, J M Doyle, G Gabrielse, and D De- Mille. Search for the electric dipole moment of the electron with tho- rium monoxide. Journal of Physics B: Atomic, Molecular and Optical Physics, 43(7):074007, 2010. [27] J. J. Hudson, B. E. Sauer, M. R. Tarbutt, and E. A. Hinds. Measure- ment of the Electron Electric Dipole Moment Using YbF Molecules. Phys. Rev. Lett., 89:023003, Jun 2002. [28] R Jason Jones and Jun Ye. Femtosecond pulse ampli cation by coherent addition in a passive optical cavity. October, 27(20):1848{1850, 2002. [29] Christoph Gohle, Thomas Udem, Maximilian Herrmann, Jens Rauschenberger, Ronald Holzwarth, Hans a Schuessler, Ferenc Krausz, and Theodor W H ansch. A frequency comb in the extreme ultraviolet. Nature, 436(7048):234{7, July 2005. [30] R.W. Boyd. Nonlinear Optics. Electronics & Electrical. Academic Press, 2003. [31] M. Bass, P. A. Franken, J. F. Ward, and G. Weinreich. Optical recti - cation. Phys. Rev. Lett., 9:446{448, Dec 1962. [32] X Zhanga and D H Auston. Optoelectronic measurement with fem- tosecond optics surfaces and interfaces. pages 326{338, 1992. [33] [34] Ajay Nahata, Aniruddha S. Weling, and Tony F. Heinz. A wide- band coherent terahertz spectroscopy system using optical recti cation and electro-optic sampling. Applied Physics Letters, 69(16):2321{2323, 1996. [35] K. Kieu and F. W. Wise. All- ber normal-dispersion femtosecond laser. Opt. Express, 16(15):11453{11458, Jul 2008. 180Bibliography [36] M. Hofer, M. E. Fermann, F. Haberl, M. H. Ober, and A. J. Schmidt. Mode locking with cross-phase and self-phase modulation. Opt. Lett., 16(7):502{504, Apr 1991. [37] E. Treacy. Optical pulse compression with di raction gratings. Quan- tum Electronics, IEEE Journal of, 5(9):454 { 458, sep 1969. [38] R. L. Fork, C. H. Brito Cruz, P. C. Becker, and C. V. Shank. Compres- sion of optical pulses to six femtoseconds by using cubic phase compen- sation. Opt. Lett., 12(7):483{485, Jul 1987. [39] Miklos Stern, Jonathan P Heritage, and E W Chase. Grating compen- sation of third-order ber dispersion. 2(12), 1992. [40] Oscar Eduardo Martinez. 3000 Times Grating Compressor with Pos- itive Group Velocity Dispersion : Application to Fiber Compensation in 1.3-1.6 micron Region. Quantum, (1):59{64, 1987. [41] T. J. Hammond, Arthur K. Mills, and David J. Jones. Simple method to determine dispersion of high- nesse optical cavities. Optics Express, 17(11):8998, May 2009. [42] J. Z. Xu and X.-C. Zhang. Optical recti cation in an area with a diam- eter comparable to or smaller than the center wavelength of terahertz radiation. Opt. Lett., 27(12):1067{1069, Jun 2002. [43] P U Jepsen and S R Keiding. Radiation patterns from lens-coupled terahertz antennas. Optics letters, 20(8):807{9, April 1995. [44] Jesse Petersen. Terahertz generation via resonantly - enhanced optical recti cation of ultrafast laser pulses. Internal Memo. [45] R. W. P. Drever, J. L. Hall, F. V. Kowalski, J. Hough, G. M. Ford, a. J. Munley, and H. Ward. Laser phase and frequency stabilization using an optical resonator. Applied Physics B Photophysics and Laser Chemistry, 31(2):97{105, June 1983. 181Bibliography [46] B.E.A. Saleh and M.C. Teich. Fundamentals of photonics. Wiley series in pure and applied optics. Wiley-Interscience, 2007. [47] A.E. Siegman, M.W. Sasnett, and Jr. Johnston, T.F. Choice of clip lev- els for beam width measurements using knife-edge techniques. Quantum Electronics, IEEE Journal of, 27(4):1098 {1104, apr 1991. [48] J ero^me Poirson, Fabien Bretenaker, Marc Vallet, and Albert Le Floch. Analytical and experimental study of ringing e ects in a fabry{perot cavity. application to the measurement of high nesses. J. Opt. Soc. Am. B, 14(11):2811{2817, Nov 1997. [49] H. Bach and N. Neuroth. The Properties of Optical Glass. Schott Series on Glass and Glass Ceramics. Springer, 1995. [50] Micheal Thorpe, R Jones, K Moll, Jun Ye, and Ramin Lalezari. Precise measurements of optical cavity dispersion and mirror coating properties via femtosecond combs. Optics express, 13(3):882{8, March 2005. [51] Albert Schliesser, Christoph Gohle, Thomas Udem, and Theodor W H ansch. Complete characterization of a broadband high- nesse cavity using an optical frequency comb. Optics express, 14(13):5975{83, July 2006. [52] P. Deus, U. Voland, and H. A. Schneider. Thermal expansion of GaP within 20 to 300 K. physica status solidi (a), 80(1):K29{K32, 1983. [53] C.B. de Araujo and H. Lotem. New measurements of the two-photon absorption in GaP, CdS, and ZnSe relative to Raman cross sections. Physical Review B, 18(1):30, 1978. [54] Isao Tomita, Hiroyuki Suzuki, Hiroshi Ito, Hirokazu Takenouchi, Kat- suhiro Ajito, Rakchanok Rungsawang, and Yuko Ueno. Terahertz-wave generation from quasi-phase-matched GaP for 1.55 m pumping. Ap- plied Physics Letters, 88(7):071118, 2006. [55] WC Dash. Intrinsic optical absorption in single-crystal germanium and silicon at 77 K and 300 K. Physical Review, 99, 1955. 182Bibliography [56] Pallab Bhattacharya. Semiconductor Optoelectronic Devices. Prentice Hall, 2, illustr edition, 1997. [57] PJ Dean and DG Thomas. Intrinsic Absorption-Edge Spectrum of Gallium Phosphide. Physical Review, 150, 1966. [58] Japan Manufacturerâ€™s Society of Compound Semiconductor Materials (JAMS-CS). EPD measurements for low dislocation density GaAs and InP substrates. III-Vs Review, 12(6):32 { 37, 1999. [59] M Beaudoin, A J G Devries, S R Johnson, H Laman, and T Tiedje. Optical absorption edge of semi-insulating GaAs and InP at high tem- peratures. Semiconductors, 70(June):3540{3542, 1997. [60] M. Theuer, D. Molter, K. Maki, C. Otani, J. a. Lhuillier, and R. Beigang. Terahertz generation in an actively controlled femtosec- ond enhancement cavity. Applied Physics Letters, 93(4):041119, 2008. [61] Fabian D. J. Brunner, Arno Schneider, and Peter Gunter. Velocity- matched terahertz generation by optical recti cation in an organic non- linear optical crystal using a Ti:sapphire laser. Applied Physics Letters, 94(6):061119, 2009. [62] Jun Takayanagi, So Kanamori, Koji Suizu, Masatsugu Yamashita, Toshihiko Ouchi, Shintaro Kasai, Hideyuki Ohtake, Hiromasa Uchida, Norihiko Nishizawa, and Kodo Kawase. Generation and detection of broadband coherent terahertz radiation using 17-fs ultrashort pulse ber laser. Optics express, 16(17):12859{65, August 2008. [63] F O Ilday, H Lim, J R Buckley, and F W Wise. Practical all- ber source of high-power, 120-fs pulses at 1 micrometre. Optics letters, 28(15):1362{4, August 2003. [64] C K Nielsen. Using photonic bandgap bers for dispersion control and pulse compression. Optics Express, 14(13):4368{4373, 2006. 183Bibliography [65] Guoqing Chang, Charles J. Divin, Chi-Hung Liu, Steven L. Williamson, Almantas Galvanauskas, and Theodore B. Norris. Power scalable com- pact THz system based on an ultrafast Yb-doped ber ampli er. Opt. Express, 14(17):7909{7913, 2006. [66] Guoqing Chang, Charles J Divin, Jun Yang, Malakeh a Musheinish, Steven L Williamson, Almantas Galvanauskas, and Theodore B Nor- ris. GaP waveguide emitters for high power broadband THz generation pumped by Yb-doped ber lasers. Optics express, 15(25):16308{15, December 2007. [67] H. Bethe. Theory of Di raction by Small Holes. Physical Review, 66(7- 8):163{182, October 1944. [68] J Z Xu and X-C Zhang. Optical recti cation in an area with a diam- eter comparable to or smaller than the center wavelength of terahertz radiation. Optics letters, 27(12):1067{9, July 2002. [69] Paul D. Cunningham, Nestor N. Valdes, Felipe a. Vallejo, L. Michael Hayden, Brent Polishak, Xing-Hua Zhou, Jingdong Luo, Alex K.-Y. Jen, Jarrod C. Williams, and Robert J. Twieg. Broadband terahertz characterization of the refractive index and absorption of some impor- tant polymeric and organic electro-optic materials. Journal of Applied Physics, 109(4):043505, 2011. [70] Q Wu and X Zhang. Free-space electro-optic sampling of terahertz beams. 67(September):3523{3525, 1995. [71] Akhlesh Lakhtakia and Tom G Mackay. Electrical control of the linear optical properties of particulate composite materials. Proceedings of the Royal Society A: Mathematical, Physical and Engineering Science, 463(2078):583{592, 2007. [72] D. F. Nelson and E. H. Turner. Electro-optic and Piezoelectric Coe - cients and Refractive Index of Gallium Phosphide. Journal of Applied Physics, 39(7):3337{3343, 1968. 184Bibliography [73] R. CLARK JONES. A new calculus for the treatment of optical sys- tems. J. Opt. Soc. Am., 31(7):500{503, Jul 1941. [74] M Brunken and H Genz. Electro-Optic Sampling at the TESLA Test Accelerator : Experimental Setup and First Results. TESLA Report, 2003-11:1{24, 2003. [75] H. J. Bakker, G. C. Cho, H. Kurz, Q. Wu, and X.-C. Zhang. Distortion of terahertz pulses in electro-optic sampling. Journal of the Optical Society of America B, 15(6):1795, June 1998. [76] Q Wu and X Zhang. Free-space electro-optics sampling of mid-infrared pulses. Society, 71(September):1285{1286, 1997. [77] P Kaufmann. Uncooled Detectors of Continuum Terahertz Radiation. Spectrum, 10(1):288{294, 2011. [78] Matthias C Ho mann, Ka-Lo Yeh, J anos Hebling, and Keith a Nelson. E cient terahertz generation by optical recti cation at 1035 nm. Optics express, 15(18):11706{13, October 2007. [79] Yihong Yang, Alisha Shutler, and D Grischkowsky. Measurement of the transmission of the atmosphere from 0.2 to 2 THz. Optics express, 19(9):8830{8, May 2011. [80] A. Bartels, A. Thoma, C. Janke, T. Dekorsy, A. Dreyhaupt, S. Winnerl, and M. Helm. High-resolution THz spectrometer with kHz scan rates. Optics Express, 14(1):430{437, January 2006. [81] Fritz Keilmann, Christoph Gohle, and Ronald Holzwarth. Time-domain mid-infrared frequency-comb spectrometer. Opt. Lett., 29(13):1542{ 1544, Jul 2004. [82] Q. Wu and X.-C. Zhang. Free-space electro-optics sampling of mid- infrared pulses. Applied Physics Letters, 71(10):1285{1286, 1997. [83] a Bartels, R Cerna, C Kistner, a Thoma, F Hudert, C Janke, and T Dekorsy. Ultrafast time-domain spectroscopy based on high-speed 185Bibliography asynchronous optical sampling. The Review of scienti c instruments, 78(3):035107, March 2007. [84] Albert Schliesser, Markus Brehm, Fritz Keilmann, and Daniel van der Weide. Frequency-comb infrared spectrometer for rapid, remote chem- ical sensing. Opt. Express, 13(22):9029{9038, 2005. [85] S. Reinhardt, E. Peters, T. W. H ansch, and Th. Udem. Two-photon di- rect frequency comb spectroscopy with chirped pulses. Physical Review A, 81(3):1{5, March 2010. [86] A. Bartels, A. Thoma, C. Janke, T. Dekorsy, A. Dreyhaupt, S. Winnerl, and M. Helm. High-resolution THz spectrometer with kHz scan rates. Optics Express, 14(1):430{437, January 2006. [87] Stefano Barbieri. Coherent sampling of active mode-locked terahertz quantum cascade lasers and frequency synthesis. Nature Photonics, 5(5):306, 2011. [88] Alexander Sell, Rudiger Scheu, Alfred Leitenstorfer, and Rupert Hu- ber. Field-resolved detection of phase-locked infrared transients from a compact Er: ber system tunable between 55 and 107 THz. Applied Physics Letters, 93(25):251107, 2008. [89] G. Gallot and D. Grischkowsky. Electro-optic detection of terahertz radiation. J. Opt. Soc. Am. B, 16(8):1204{1212, 1999. [90] P.C.D. Hobbs. Building Electro-Optical Systems: Making It All Work. Wiley Series in Pure and Applied Optics. Wiley, 2009. [91] Fatih O. Ilday and Frank W. Wise. Nonlinearity management: a route to high-energy soliton ber lasers. Journal of the Optical Society of America B, 19(3):470, March 2002. [92] D Strickland and G Mourou. Compression of ampli ed chirped optical pulses. Optics Communications, 56(3):219{221, 1985. 186[93] E.D. Palik. Handbook of Optical Constants of Solids. Number v. 1 in Handbook of Optical Constants of Solids. Academic Press, 1985. [94] Ian Coddington, William Swann, and Nathan Newbury. Coherent Mul- tiheterodyne Spectroscopy Using Stabilized Optical Frequency Combs. Physical Review Letters, 100(1):11{14, January 2008. [95] R.C. Dorf and R.H. Bishop. Modern Control Systems. Pearson Prentice Hall, 2008. [96] G Klatt, R Gebs, C Janke, T Dekorsy, and a Bartels. Rapid-scanning terahertz precision spectrometer with more than 6 THz spectral cover- age. Optics express, 17(25):22847{54, December 2009. [97] Stefano Barbieri, Marco Ravaro, Pierre Gellie, and Giorgio Santarelli. Coherent sampling of active mode-locked terahertz quantum cascade lasers and frequency synthesis. Nature Photonics, 5(April), 2011. [98] M Hofer, M E Fermann, F Haberl, M H Ober, and a J Schmidt. Mode locking with cross-phase and self-phase modulation. Optics letters, 16(7):502{4, April 1991. [99] F Ilday, J Chen, and F K artner. Generation of sub-100-fs pulses at up to 200 MHz repetition rate from a passively mode-locked Yb-doped ber laser. Optics express, 13(7):2716{21, April 2005. 187Appendix A Technical Information Regarding High Gain, Low-pass Filtered, Balanced Detector for Detection of the Heterodyne Beat in the Dual-comb Spectrometer Setup Below is the circuit schematic for the high-gain, balanced photodetector used for detection of the heterodyne beat in the dual-comb terahertz spectrometer experiment. The detector provides strong attenuation of frequencies above 80MHz, as shown in gure 6.10. This circuit was designed and built by Pavel Trochtchanovitch at the University of British Columbia Department of Physics Electrical Shop speci cally for this application. 188Appendix A. Data for Bandwidth-limited Balanced Detector Used for Detection of Heterodyne Beat IN 1 GND 2OU T 3 TAB U4 LT 11 29 IST -5 IN 1 GND 5 OU T 3 SH DN 6 SE NS 4 IN 8 ILI M2 2 ILI M4 7 U7 LT 11 75 IN 8-5 1 2 3 P4 u1C2 8 u1C4 1 u1C2 9 u1C4 2 10 u 16 V C2 3 10 u 16 V C3 6 +5 V -5V 32 6 AD 80 48 U1 A 7 4 AD 80 48 U5 B 7 4 AD 80 48 U1 B 32 6 AD 80 48 U5 A 7 4 AD 80 48 U2 B 32 6 AD 80 48 U6 A 32 6 AD 80 48 U2 A 7 4 AD 80 48 U6 B 34 1 AD 86 65 U3 A 5 2 AD 86 65 U3 B u1C3 0 u1C4 3 10 u 16 V C2 4 10 u 16 V C3 7 4R 99 R2 4 4R 99 R2 8 +5 V -5V u1C3 1 u1C4 4 10 u 16 V C2 5 10 u 16 V C3 8 4R 99 R2 5 4R 99 R2 9 +5 V -5V u1C3 2 u1C4 5 10 u 16 V C2 6 10 u 16 V C3 9 4R 99 R2 6 4R 99 R3 0 +5 V -5V u1C3 3 u1C4 6 10 u 16 V C2 7 10 u 16 V C4 0 4R 99 R2 7 4R 99 R3 1 +5 V -5V u1C3 4 u1C4 7 +5 V -5V 2k2R2 3 49 R9 R2 2 P3 BN C_ 3 D2 G8 37 6 D1 G8 37 6 4R 99 R1 5 4R 99 R2 +5 V -5V u1C1 3 u1C6 10 00 uH L5 56 0u H L6 2k2R2 1 u1C8 68 nH L1 68 nH L2 4p5 -4 0p C7D NPC5 DN P C1 22 0 R1 TP 1 TP 2 22 0 R6 4p5 -4 0p C1 4 DN P C1 7 u1C9 47 0 R7 22 0p C2 1 2k2R3D NPC2 22 0R8 22 0R9 12 p C1 1 TP 3 68 nH L3 4p5 -4 0p C1 5 DN P C1 8 u1C1 0 47 0 R1 0 2k2R4D NPC3 22 0R1 1 22 0R1 2 12 p C1 2 TP 4 68 nH L4 4p5 -4 0p C1 6 DN P C1 9 2k2R5D NPC4 47 0R1 3 49 R9 R1 4 P1 BN C_ 3 G= +6 dB G= +6 dB G= +6 dB 22 p C2 0 TP 5 2k2 R2 0 6k8 R1 9 22 k R1 8 68 k R1 7 22 0k R1 6 1 3 4 5 62 7 8 P2HE AD ER 1X 8L +6 Vt o+ 20 V -6V to- 20 V DC Ou t AC Ou t ??? R3 2 ??? R3 3 10 u 16 V C2 2 10 u 16 V C3 5 Figure A.1: Circuit schematic of the custom-designed, bandwidth-limited balanced photodetector. 189
- Library Home /
- Search Collections /
- Open Collections /
- Browse Collections /
- UBC Theses and Dissertations /
- Advances in terahertz frequency combs
Open Collections
UBC Theses and Dissertations
Featured Collection
UBC Theses and Dissertations
Advances in terahertz frequency combs Stead, Robert A. 2012-10-17
pdf
Page Metadata
Item Metadata
Title | Advances in terahertz frequency combs |
Creator |
Stead, Robert A. |
Publisher | University of British Columbia |
Date Issued | 2012 |
Description | Following a review of the theory of terahertz generation via optical rectification in nonlinear crystals, a method for enhancing the nonlinear conversion efficiency of this process is proposed. A nonlinear crystal is placed at the intracavity focus of a passive optical resonator, which is seeded by an ytterbium-doped fibre laser. Models of this arrangement indicate that an enhancement of the optical field of several orders of magnitude is possible. As the difference frequency radiation produced through the optical rectification process results from a mixing of the spectral components of the optical field, one expects a corresponding increase in the terahertz field. We present a design of optical resonator that compensates for the large group velocity dispersion of the nonlinear crystal. Our experimental results indicate that below bandgap absorption in the crystal severely limits the resulting enhancement of the optical field, and hence the terahertz field one would expect from this nonlinear process. A scanning-delay terahertz time-domain spectrometer has been constructed, using a gallium phosphide guiding structure to increase the interaction length of the optical and terahertz fields, thereby increasing the terahertz power produced. Our experiment demonstrates 20 dB signal to noise ratio over the spectral range of 0.5-1 THz. We propose a method for increasing the spectral resolution, whilst simultaneously reducing the required data acquisition time of such a terahertz spectrometer, through the use of two femtosecond optical frequency combs. One of these fields drives the nonlinear optical rectification process, whilst the second serves as a sampling local oscillator field to probe the terahertz field via electro-optic sampling in a second nonlinear crystal. By precisely controlling the relative pulse repetition rates of the two oscillators, we show that the full spectral content of the terahertz field can be acquired at rf frequencies, and without the slow mechanical delay lines associated with conventional terahertz time-domain spectroscopy. Finally, we present experimental efforts towards the demonstration of this technique, and show that, to be effective, steps must be taken to increase the strength of the expected rf signal over that of the measurement noise floor. |
Genre |
Thesis/Dissertation |
Type |
Text |
Language | eng |
Date Available | 2012-10-17 |
Provider | Vancouver : University of British Columbia Library |
Rights | Attribution 3.0 Unported |
DOI | 10.14288/1.0073311 |
URI | http://hdl.handle.net/2429/43450 |
Degree |
Doctor of Philosophy - PhD |
Program |
Physics |
Affiliation |
Science, Faculty of Physics and Astronomy, Department of |
Degree Grantor | University of British Columbia |
Graduation Date | 2012-11 |
Campus |
UBCV |
Scholarly Level | Graduate |
Rights URI | http://creativecommons.org/licenses/by/3.0/ |
Aggregated Source Repository | DSpace |
Download
- Media
- 24-ubc_2012_fall_stead_robert.pdf [ 5.96MB ]
- Metadata
- JSON: 24-1.0073311.json
- JSON-LD: 24-1.0073311-ld.json
- RDF/XML (Pretty): 24-1.0073311-rdf.xml
- RDF/JSON: 24-1.0073311-rdf.json
- Turtle: 24-1.0073311-turtle.txt
- N-Triples: 24-1.0073311-rdf-ntriples.txt
- Original Record: 24-1.0073311-source.json
- Full Text
- 24-1.0073311-fulltext.txt
- Citation
- 24-1.0073311.ris
Full Text
Cite
Citation Scheme:
Usage Statistics
Share
Embed
Customize your widget with the following options, then copy and paste the code below into the HTML
of your page to embed this item in your website.
<div id="ubcOpenCollectionsWidgetDisplay">
<script id="ubcOpenCollectionsWidget"
src="{[{embed.src}]}"
data-item="{[{embed.item}]}"
data-collection="{[{embed.collection}]}"
data-metadata="{[{embed.showMetadata}]}"
data-width="{[{embed.width}]}"
async >
</script>
</div>
Our image viewer uses the IIIF 2.0 standard.
To load this item in other compatible viewers, use this url:
http://iiif.library.ubc.ca/presentation/dsp.24.1-0073311/manifest